In a subanalysis comparing both groups (fixed vs. uniplanar screws) in only the Lenke 1 and Lenke 2 curve population, a similar result was found. (Table 3) The immediate postcorrection sagittal curvature measurements were greater using the uniplanar than the fixed screws, 32 versus 23 degrees (P=0.02). This difference was also seen at the final follow-up with greater kyphosis demonstrated in the uniplanar screw group at 44 degrees compared with 28 degrees in the fixed screw group. This difference was statistically significant (P=0.004). There was no difference seen across groups with respect to correction of the coronal thoracic deformity. The average first erect coronal Cobb measurements for the uniplanar group and fixed screw group were identical at 16 degrees (P=0.9). This remained the case at the final follow-up with an average coronal thoracic deformity for uniplanar group and fixed screw group of 16 degrees compared with 19 degrees (P=0.9).
The surgical treatment for AIS aims to achieve 3-dimensional correction of the deformity with a balanced spine, ultimately arresting the deformity progression.1,2,5 The conventional focus of AIS surgery has been the correction of the major curve in the coronal plane. The normal TK has been described in the range of 20 to 40 degrees.6 In patients with AIS, thoracic hypokyphosis is a common characteristic of the curve morphology. In addition, with the widespread use of pedicle fixation for AIS correction, it is apparent that the coronal correction is inversely proportional to the TK; thus, leading to a“thoracic flat back.”1,2,5–9 The suboptimal achievement of the TK following AIS surgery could contribute to the development of adjacent segment disease. It has been demonstrated that the preservation of TK is critical to maintain lumbar lordosis in AIS surgery.10 Suboptimal kyphosis restoration also causes loss of compensatory lumbar lordosis, which could contribute to positive sagittal balance and accelerated disk generation. Early symptomatic spinal degeneration can result in pain, disability, and suboptimal quality of life.1,2,5,8,10–13 Hence, the enthusiasm to improve the sagittal alignment has increased, partly due to the awareness about the flat back phenomenon that can occur following pedicle screw constructs. In general, the focus has moved beyond coronal correction toward global correction. With this background, the surgical restoration of TK in AIS patients has gained renewed attention. Thus, the reason for change in the type of screw was for kyphosis correction and better control over the sagittal profile with uniplanar screws. However, the long-term outcomes, of patients with all types of screw constructs, remain to be seen and may present unforeseen challenges.
Clements et al4 investigated the effect of implant density on the changes in sagittal contour preoperatively to postoperatively. They found higher implant density to be related to decreased kyphosis in the thoracic spine. Our study did not facilitate a way to analyze this fact. During the time these patients underwent surgery, our construct density was >90% which was at the highest end of the correlation reported in that study which compared construct densities that varied from below 20% up to 100%. The preponderance of constructs in that study was below 80%. Further, we lacked an adequate amount of variation in our construct density to examine this variable.
Lui and colleagues studied the effects of rod stiffness and implant density on coronal and sagittal plane correction in patients with main thoracic curve AIS. The study concluded that both rods with high and low stiffness combined with high or low screw density could provide effective correction in the coronal plane. With respect to the sagittal alignment, rods with high stiffness along with high screw density on the concave side could provide better outcomes with respect to TK restoration.2,5 Our study did not vary in terms of rod character; and therefore, a comparison on this factor was not possible. All of the rods used were 6.35-mm stainless steel.
A quantitative meta-analysis was conducted by Cao and colleagues on the studies reporting results of spinal surgery in AIS with instrumentation of pedicle screw or hybrid construct with the objective to evaluate the efficacy to restore TK. Hybrid constructs seem to be more powerful in restoring kyphosis than pedicle screws. Factors like rod stiffness, fashion of hybrid construct, in situ bending, and rod rotation maneuver could have an influence in restoration of the sagittal alignment.5,9 In contrast, a study by Suk et al14 indicated that segmental pedicle screw fixation was more effective than multiple hooks in restoring kyphosis in patients with hypokyphotic scoliosis and created kyphosis similar to that in patients without preoperative hypokyphosis.
Clement et al15 described the use of a technique involving simultaneous translation on 2 rods for AIS correction that achieved satisfactory restoration of TK. Similarly, Sudo et al16 described a simultaneous double-rod rotation technique for treating Lenke 1 AIS providing significant sagittal correction of the main thoracic curve while maintaining sagittal profiles and correcting coronal and axial deformities. In another study by Clement et al,17 they concluded that simultaneous translation on 2 rods provides a better correction of TK than the sequential approximation by cantilever reduction on patients with preoperative hypokyphosis. Sucato and colleagues compared the use of 3 surgical approaches in the treatment of AIS with the aim of restoring TK. They found anterior spinal fusion and instrumentation was the best method to restore TK when compared with posterior approaches using only hooks or a hybrid constructs.18 Imagama and colleagues described the surgical procedure and results for posterior correction and fusion with a hybrid approach using pedicle screws, hooks, and ultra-high molecular weight polyethylene tape with direct vertebral rotation for treatment of AIS. The study concluded that the technique is advantageous for kyphosis restoration compared with segmental screw fixation, and uses the minimum number of pedicle screws.19 Our present study did not involve the use of any hybrid fixation devices, special approaches, or correction techniques and did not evaluate these aspects.
Uniplanar pedicle screws have been a recent addition to the armamentarium used for AIS correction. Uniplanar screws allow for freedom of motion of the screw head in the sagittal plane. This feature can accommodate the rod curvature and eventually the sagittal spinal profile better than the fixed variant. In our study, the patient receiving uniplanar screws demonstrated an improvement in the TK, irrespective of their preoperative thoracic alignment being normo-kyphotic or hypokyphotic. Thus, the study suggests that uniplanar screws aid in maintaining TK.
Our study was limited by its small sample size as a single center study. Repeating the study across multiple sites prospectively would be ideal to determine if the choice of screws affects the postoperative sagittal contour. Because of our small sample size, subanalysis based on parameters such as Lenke curve type (only 13 thoracic curves in the fixed group and 7 in the uniplanar group) is limited and should be interpreted carefully. There was a statistical difference between patients treated with fixed versus uniplanar screws for the study as a whole and when Lenke 1 and 2 curve types were grouped together and examined as a data subset. This grouping was chosen because these curves are entirely based in the thoracic spine and tend to be the most hypokyphotic in our experience. Uniplanar screws seemed to offer an advantage in restoring a more normal kyphotic thoracic spine. We do note that by comparing in this manner, a small sample size results with only 13 cases in the fixed group compared with 7 in the uniplanar group. Identifying a larger sample size would aid in confirming this finding.
We selected examining the immediate postoperative (first erect) and final follow-up films as radiographic parameters. The final follow-up films demonstrated more TK than the immediate postoperative films. A similar change was not seen in the coronal Cobb values. We believe this is likely due to the transition to higher quality digital films for the 2-year follow-up imaging which allowed for better windowing and measurement of the cranial end vertebra; however, other possible explanations include measurement error or perhaps real changes in TK. We did not assess apical rotation in this study, either by postoperative computed tomographic scans or radiographic surrogates. Lastly, a selection bias and other unmeasured factors may have influenced the outcomes. A study design involving a larger sample size and a simultaneous randomization of the patients to the 2 groups would have greatly added to the power of this study.
We believe our study supports the potential benefit of uniplanar screws in the sagittal plane for idiopathic scoliosis. Clinically, this should be weighed against other factors such as implant cost, implant number, and surgeon comfort with the use of a particular implant. Further work in this area will facilitate scoliosis surgeons in selecting the proper implants for each type of deformity to provide the most natural spinal contour.
1. Newton PO, Yaszay B, Upasani VV, et al. Preservation of thoracic kyphosis
is critical to maintain lumbar lordosis in the surgical treatment of adolescent idiopathic scoliosis. Spine (Phila Pa 1976). 2010;35:1365–1370.
2. Ames CP, Blondel B, Scheer JK, et al. Cervical radiographical alignment: comprehensive assessment techniques and potential importance in cervical myelopathy. Spine (Phila Pa 1976). 2013;38(suppl 1):S149–S160.
3. Xiaobang H, Lieberman IH. Prevalence and factors affecting cervical deformity in adolescent idiopathic scoliosis patients. Paper #105. Scoliosis Research Society Annual Meeting. Minneapolis, MN; 2015.
4. Clements DH, Betz RR, Newton PO, et al. Correlation of scoliosis curve correction with the number and type of fixation anchors. Spine (Phila Pa 1976). 2009;34:2147–2150.
5. Liu H, Li Z, Li S, et al. Main thoracic curve adolescent idiopathic scoliosis: association of higher rod stiffness and concave-side pedicle screw density with improvement in sagittal thoracic kyphosis
restoration. J Neurosurg Spine. 2015;22:259–266.
6. Winter RB, Lonstein JE, Denis F. Sagittal spinal alignment: the true measurement, norms, and description of correction for thoracic kyphosis
. J Spinal Disord Tech. 2009;22:311–314.
7. Mac-Thiong JM, Roussouly P, Berthonnaud E, et al. Sagittal parameters of global spinal balance: normative values from a prospective cohort of seven hundred nine Caucasian asymptomatic adults. Spine (Phila Pa 1976). 2010;35:E1193–E1198.
8. Schmidt C, Liljenqvist U, Lerner T, et al. Sagittal balance of thoracic lordoscoliosis: anterior dual rod instrumentation versus posterior pedicle screw fixation. Eur Spine J. 2011;20:1118–1126.
9. Cao Y, Xiong W, Li F. Pedicle screw versus hybrid construct instrumentation in adolescent idiopathic scoliosis: meta-analysis of thoracic kyphosis
. Spine (Phila Pa 1976). 2014;39:E800–E810.
10. Khakinahad M, Ameri E, Ghandhari H, et al. Preservation of thoracic kyphosis
is critical to maintain lumbar lordosis in the surgical treatment of adolescent idiopathic scoliosis. Acta Med Iran. 2012;50:477–481.
11. Hwang SW, Samdani AF, Tantorski M, et al. Cervical sagittal plane decompensation after surgery for adolescent idiopathic scoliosis: an effect imparted by postoperative thoracic hypokyphosis. J Neurosurg Spine. 2011;15:491–496.
12. Ilharreborde B, Morel E, Mazda K, et al. Adjacent segment disease after instrumented fusion for idiopathic scoliosis: review of current trends and controversies. J Spinal Disord Tech. 2009;22:530–539.
13. Kim YJ, Bridwell KH, Lenke LG, et al. Proximal junctional kyphosis in adult spinal deformity after segmental posterior spinal instrumentation and fusion: minimum five-year follow-up. Spine (Phila Pa 1976). 2008;33:2179–2184.
14. Suk SI, Kim WJ, Kim JH, et al. Restoration of thoracic kyphosis
in the hypokyphotic spine: a comparison between multiple-hook and segmental pedicle screw fixation in adolescent idiopathic scoliosis. J Spinal Disord. 1999;12:489–495.
15. Clement JL, Chau E, Geoffray A, et al. Restoration of thoracic kyphosis
by simultaneous translation on two rods for adolescent idiopathic scoliosis. Eur Spine J. 2014;23(suppl 4):S438–S445.
16. Sudo H, Ito M, Abe Y, et al. Surgical treatment of Lenke 1 thoracic adolescent idiopathic scoliosis with maintenance of kyphosis using the simultaneous double-rod rotation technique. Spine (Phila Pa 1976). 2014;39:1163–1169.
17. Clement JL, Chau E, Kimkpe C, et al. Restoration of thoracic kyphosis
by posterior instrumentation in adolescent idiopathic scoliosis: comparative radiographic analysis of two methods of reduction. Spine (Phila Pa 1976). 2008;33:1579–1587.
18. Sucato DJ, Agrawal S, O’Brien MF, et al. Restoration of thoracic kyphosis
after operative treatment of adolescent idiopathic scoliosis: a multicenter comparison of three surgical approaches. Spine (Phila Pa 1976). 2008;33:2630–2636.
19. Imagama S, Ito Z, Wakao N. Posterior surgery for adolescent idiopathic scoliosis with pedicle screws and ultra-high molecular weight polyethylene tape: achieving the ideal kyphosis. Clin Spine Surg. 2016;29:E376–E383.
Keywords:Copyright © 2017 Wolters Kluwer Health, Inc. All rights reserved.
adolescent idiopathic scoliosis (AIS); posterior spinal fusion; sagittal thoracic alignment; thoracic kyphosis; uniplanar pedicle screw; fixed pedicle screw