To report on the short-term results of CO2-laser assisted stapedotomy combined with the àWengen titanium clip stapes prosthesis. A comparison with published series using other prostheses and/or different stapedotomy techniques is made.
Retrospective case series.
Patients with a history and audiologic data matching stapes fixation and computed tomographic imaging excluding other anomalies such as malleus fixation, dehiscent superior semicircular canal, and large vestibular aqueduct that may mimic stapes fixation-like hearing loss.
All patients underwent CO2 laser-assisted stapedotomy (Lumenis Co. Israel CO2 laser, Acuspot 712, SurgiTouch 870 scanner) and subsequent reconstruction by means of the àWengen titanium clip stapes prosthesis by Heinz Kurz Medizintechnik GmbH (Germany).
Comparison and statistical analysis of preoperative and postoperative audiologic data.
Sixty-two stapedotomies were performed (61 patients) using the CO2 laser and àWengen titanium clip stapes prosthesis. The mean postoperative air-bone gap 3 months postoperatively was 5.1 ± 0.5 dB (standard deviation [SD], 4.1 dB; 0.5, 1, 2, 4 kHz). Air-bone gap closure less than or equal to 10 dB was achieved in 54 cases (87%). Air-bone gap closure less than 20 dB was achieved in all cases. The average gain was 27.8 ± 1.5 dB (SD, 12 dB; 0.5, 1, 2, 4 kHz). The average bone-conduction threshold shift or "overclosure" on 2,000 Hz was 13.6 ± 1.3 dB (SD, 10 dB). There was no postoperative perceptive hearing loss exceeding 15 dB on any measured frequency. The Amsterdam Hearing Evaluation Plots have also been used to evaluate our data. These data were statistically analyzed and compare favorably to other published series.
The authors conclude that the combination of CO2 laser-assisted stapedotomy and the àWengen titanium clip stapes prosthesis is a combination likely to yield superior results in experienced hands.
*Department of Otorhinolaryngology and Head and Neck Surgery, Heilig Hart General Hospital Roeselare, Roeselare; †Department of Neurosciences, Faculty of Medicine, University of Antwerp; ‡Antwerp University Research centre for Equilibrium and Aerospace, Department of Otorhinolaryngology and Head and Neck Surgery, Antwerp University Hospital; and §Department of Biomedical Physics, University of Antwerp, Antwerp, Belgium
Address correspondence and reprint requests to Glen E. J. Forton, M.D., Ph.D., ENT Department, Heilig Hartziekenhuis Roeselare, Wilgenstraat 2, 8800 Roeselare, Belgium; E-mail: firstname.lastname@example.org
Presented at the annual meeting of the Dutch-Flemish Otological Society, Bruges (Belgium), March 2-3, 2007.