Antibiotic cream (chloramphenicol 3%) and a sterile dressing were applied. Subjects were instructed not to touch the dressing or allow it to get wet and to return in 2 days for removal of the dressing, assessment of the site, and documentation of adverse events or pain. Subjects then returned for weekly follow-up until complete healing was achieved (complete epithelialization with no drainage from the circumcision site) as determined by visual inspection by the physician (Fig. 8).
In the main phase, procedure time for device placement was defined as the period from the start of rings placement on the penis to the patient standing up postplacement; procedure time for device removal was defined as the period from picking up the forceps to the beginning of wound dressing postremoval.
Photographs were taken of each subject's penis postplacement, preremoval, postremoval, and at every weekly follow-up visit. Using a standard clinical report form, information was gathered on the rate of success in achieving circumcision (glans fully exposed); time to complete healing; the incidence and severity of adverse events; and levels of pain (documented using a Visual Analog Scale [VAS] on which scores ranging from 0 to 10 represent no pain to the most severe pain) at several defined times (Table 3). Additional information was collected on nonphysicians' self-reporting of their sense of ease or difficulty in performing the procedure and time lost by subjects from work.
From a large number of volunteers, 81 men were deemed suitable for formal screening. Of these candidates, 58 were selected by the Principal Investigator; among those not selected, 2 were rejected because of phimosis (candidates with a relatively flexible foreskin were preferred, to facilitate placement of the device). All of the selected subjects showed up for the procedure, but 1 subject was later excluded when it was discovered that he was HIV positive.
The remaining 57 men ranged in age from 18 to 35 years (although the protocol allowed age up to 54, no one over age 35 enrolled); average age was 22.5 (SD: 4.3) years. Two subjects were subsequently excluded due to narrow prepuce opening, which had not been previously detected. Thus, the final study population comprised 55 men (5 soldiers, 50 civilians).
There were no cases of erroneous placement of the device in either phase of the study, and no device-related incidents (such as the device slipping out of position).
The first 5 subjects, comprising the population in the feasibility phase of the study, ranged in age from 21 to 31 years. Notwithstanding the previously described problems relating to waiting for self-detachment, all 5 subjects were successfully circumcised, with no adverse events reported over the 6-week study. Complete healing was documented by day 28 in all subjects.
The remaining 50 subjects comprised the population in the main phase; average age was 23.6 (SD: 4.3) years; median, 22.5; range, 18–35. All subjects were successfully circumcised. The majority of procedures were performed by physicians; 3 placement procedures and 1 removal procedure were performed by a certified nurse.
Active removal of the foreskin and the device was performed on day 4 in the first 5 subjects; however, to allow for more complete necrosis, removal was rescheduled to day 7 in the remaining 45 subjects. There were no differences in outcomes related to date of removal.
After the initial 20 subjects, we administered 1 g of paracetamol 30 minutes before device deployment and 30 minutes before device removal because some patients were experiencing discomfort postdeployment and/or during removal. Otherwise, mean VAS pain scores were predominantly low (Table 3). The preplacement use of paracetamol reduced postdeployment pain but preremoval use had no appreciable effect on the brief pain that subjects experienced during extraction of the inner ring.
After device removal, the only reported adverse event was diffuse edema in a patient with chronic urethritis from a belatedly reported sexually transmitted disease; the edema, presumably related to circumcision itself, resolved in 2 days with the oral anti-inflammatory ibuprofen. There were no cases of infection, bleeding, hematoma, delayed necrosis, or wound dehiscence.
Two subjects had light oozing after foreskin removal, which resolved with 10 seconds of applied pressure. There were 9 cases of painless, mild, localized edema of the frenulum, all of which resolved spontaneously (8 cases resolving in 3 days, 1 case in 2 weeks). The remnant of necrotic foreskin retained after device and foreskin removal dropped off spontaneously within 1–2 weeks (ie, between day 14 and day 21; median, day 14). In some cases, the physicians could detect an odor from the necrotic tissue, but no subjects made any complaint about smell.
Mean time for complete healing was 25.3 (SD: 5.7) days after device removal; median, 21 days. Complete healing was documented in 48 subjects (2 subjects missed their last follow-up visit before complete healing was documented). Complete healing was achieved by day 21 (2 weeks after device removal) in 26 subjects; at day 28 in another 16 subjects; at day 35 in another 4 subjects; and at day 42 in 2 subjects. Most subjects did not return after being told that complete healing had been achieved. Thus, among the 50 subjects in the main phase, 49 appeared for follow-up at week 3; 48 at week 4; 36 at week 5, and 17 at week 6. None of the subjects missed any days of military service or work relating to the circumcision.
Procedure times ranged from 3.0 to 5.1 (mean 4.3) minutes for device placement and from 1.9 to 5.8 (mean 3.8) minutes for removal. Personnel reported increasing ease in performing the procedures as they gained experience.
Prior studies with other products for adult male circumcision have reported advantages over conventional surgical circumcision, such as elimination of sutures and reduction in procedure time.12–16 However, we believe that all of these other methods require injection of local anesthesia for device placement and/or removal and that our study is the first report of adult male circumcision as a public health initiative performed without anesthesia.
Another adult male circumcision device is the Shang Ring (Wuhu SNNDA Medical Treatment Appliance Technology Co Ltd, Wuhu, China). A 2011 report describes a 6-week study of this device in Kenya. Similar to the present study, device placement was on day 1 and removal was on day 7.17 Of 40 procedures performed, all resulted in successful circumcision.
Both the Shang Ring and the PrePex device are single-use devices supplied in multiple sizes. The Shang Ring immediately crushes the foreskin and thus requires anesthesia (6–20 cc of 1% lidocaine without epinephrine, injected circumferentially around the penis) before placement; the PrePex device applies controlled radial elastic pressure and hence requires no anesthesia (though we found it helpful to provide 1 g of paracetamol to reduce discomfort postplacement). Use of the Shang Ring is a surgical procedure that requires a sterile setting because the live distal foreskin is removed immediately after placement of the device (in addition, 4 small incisions are made around the cut edge of the foreskin to minimize wound contraction and to limit pain from nocturnal erection15); use of the PrePex device is a nonsurgical procedure that can be performed in a standard consultation room because the distal foreskin is necrotic when removed, bloodlessly. Time for placement and removal of the Shang Ring averaged 4.8 and 3.9 minutes, respectively (total: 8.7 minutes), not counting time for anesthesia to be administered and to take effect; time for placement and removal of the PrePex device averaged 4.3 and 3.8 minutes (total: 8.1 minutes), and there was no need for anesthesia. There were 3 instances of partial detachment of the Shang Ring before removal; there were no mechanical incidents with the PrePex device. The Shang Ring was associated with 6 mild adverse events (3 cases of penile skin trauma, 2 of edema, and 1 of infection) in 40 subjects; the PrePex device was associated with 1 mild adverse event (diffuse edema in a man with chronic urethritis) in the 55 subjects in both phases of the study. In the Shang Ring study, 80% of the men returned to work within 2 days after device placement; in this study of the PrePex device, 100% of the men returned to work the day after device placement. Time to complete healing averaged 28.9 days with the Shang Ring and 25.3 days after removal of the PrePex device (ie, 32 days after placement). In a separate report on 328 Chinese men circumcised with the Shang Ring, major complications included infection (0.6%), bleeding (0.6%), wound dehiscence (0.6%), and edema (4.9%).15
The success of any public program encouraging adult circumcision depends on acceptability by candidates for the procedure. The large number of unsolicited volunteers who approached us before and during the study suggests adequate acceptability within the target population. Several subjects spontaneously expressed satisfaction with the procedure during the course of the study, and some said that they would recommend it to others.
Concerning the exclusion of men with phimosis, the present study was designed to assess a nonsurgical procedure for rapid scale-up programs in Sub-Saharan Africa; men with phimosis might be circumcised with the PrePex device after a small slit in the foreskin, as has been done with other devices, although that added step would transform the procedure from nonsurgical to surgical; future studies may explore this option. The true prevalence of phimosis is uncertain. In a study of 351 boys and men circumcised with the Shang Ring, 46 (13%) had phimosis,18 whereas among 81 screened candidates in the present study, 4 (5%) were excluded (2 during screening, 2 after admission) because of phimosis. According to a report from the British Association of Paediatric Urologists, 5% of boys ages 16–17 still have nonretractible foreskin.19
Healing and Adverse Events
The average time for complete healing after surgical circumcision in adult males is 4–6 weeks.10 In our study with the PrePex device, mean time to complete healing was 25.3 days after foreskin removal or 32 days after device placement, which is within the same range, and is similar to healing time with the Shang Ring.
Reported problems associated with surgical circumcision include postoperative bleeding, wound infection, anesthesia-related complications, and edema.20 With the PrePex device, localized edema is an expected occurrence during healing. Edema incidence rates of 50%12 and 14%13 have been reported with other ring devices. In our study, all cases of localized edema (9 of 50 subjects) occurred after device removal and resolved spontaneously. There were no adverse events while the device was in place. The only reported adverse event after removal was the case of diffuse edema in a subject with chronic urethritis.
During the initial phase of the study, self-detachment of the foreskin was explored but ruled out as unpredictable and impractical (although the experience suggested that there would be no acute danger if a man missed his scheduled visit for active removal). Accordingly, in the main phase, active removal of the foreskin and the device was a scheduled procedure, which took place on day 4 in the first 5 subjects and on day 7 in the remaining 45 subjects, to allow time for the foreskin to become more completely necrotic for bloodless removal with scissors. We noted no differences in outcomes related to removal on day 4 versus day 7.
Infection and Bleeding
There were no cases of infection or outright bleeding with the PrePex device, but 2 subjects had light, transient oozing after foreskin and device removal (stopped by local compression with gauze for 10 seconds). From various reports on the Shang Ring, the incidence of infection was 0.6%–2.5%, and the incidence of bleeding was 0.0%–0.6%.12,15,17,18 From a systematic review of published reports of nontherapeutic surgical circumcision in men, the 2 most common adverse events were infection (incidence 1.47%) and postoperative bleeding (1.32%).20
Because we noted that several of the first 20 subjects in the main phase experienced pain after placement and/or during removal of the device, we thereafter provided a 1-g dose of oral paracetamol 30 minutes before placement and before removal; no additional painkiller was required in any subject at any time. Subjects reported minimal pain at all other times. Paracetamol reduced postplacement discomfort but not the brief pain subjects experienced during removal of the inner ring. Circumcision using the Shang Ring requires injection of local anesthetic around the penis.15,17
We recommend that in future studies, VAS pain score should be checked earlier after device placement (in the present study, we checked 3–6 hours after placement). We now believe that ibuprofen administered immediately after placement may be a better means of managing discomfort at this step.
Among the 50 subjects in the main phase of our study, 10 reported minor discomfort with nocturnal erection while the PrePex device was in place, but this experience was not objectively measured. The need for a standardized means of assessing pain from nocturnal erection is suggested by contrasting the 20% incidence of pain on erection reported in the present study and the 86% incidence reported in one study of the Shang Ring.12
As the team gained experience, procedure time decreased. We anticipate that total procedure time would be less than 5 minutes when performed by experienced personnel. Total procedure time is marginally lower with the PrePex device than with the Shang Ring (8.1 versus 8.7 minutes), but anesthesia adds to the overall time required with the Shang Ring. In various reports, the time required for conventional surgical circumcision is 25–36 minutes.16,21,22
Topical Agents and Dressings
We used vaseline as a lubricant on the penis to ease insertion of the PrePex inner ring. In retrospect, we would recommend a water-based lubricant applied only on the Inner Ring itself.
After removal of the foreskin and the device, we covered the remnant of necrotic tissue with antibiotic cream and gauze dressing to reduce the risk of wound infection. In fact, there were no infections, and we suspect that there is no need for antibiotic cream; this question will be explored in future studies. An alternative would be a vaseline gauze dressing (as currently used in surgical circumcision) or a standard gauze dressing.
Staff and Facility Requirements
Circumcision using the PrePex device requires 2 trained staff members for device placement; both can be nurses, and at least 1 should have ample dexterity. Device removal requires 1 person with ample dexterity. Although most of the procedures were handled by physicians, we noted no differences in procedure time, pain, adverse events, or healing time related to whether procedures were performed by a nurse or a physician. Further study is warranted to document the feasibility of entrusting the procedure to nurses, but these preliminary results are encouraging and potentially important for large-scale male circumcision programs in Sub-Saharan Africa.
The main phase of our study was conducted in a standard consultation room. Aside from savings in costs and resource use from foregoing the sterile setting, men may feel less intimidated in a consultation room.
Binagwaho et al23 have demonstrated that adult male circumcision is cost-effective even in regions with a relatively low HIV prevalence, such as Rwanda. Although there was no formal assessment of costs in this study, the use of the PrePex device may reduce costs, as the procedures can be performed by nonphysician staff, without anesthesia or suturing, in an ordinary consultation room. In the main phase of our study, none of the subjects missed any days of work after placement or removal of the device, which is relevant in a national scale-up program.
Although we tried to select subjects with a more flexible foreskin, there was no validated method for assessing candidates on this basis. Also, we performed no methodical assessment of discomfort from erection while the PrePex device was in place; we recommend that future studies should obtain VAS pain scores during erection if subjects are cooperative. We did not formally analyze costs with the PrePex device versus other methods of circumcision, nor did we formally assess subjects' overall level of satisfaction with the procedure.
The PrePex device was safe and effective as a means of performing bloodless adult male circumcision that can be carried out by nonphysician staff without need for anesthesia, suturing, or sterile settings. Pending further research, such as a study in which the procedure is performed entirely by nonphysician personnel, these promising results could prove to be a significant advance in HIV prevention programs in sub-Saharan Africa.
This study was particularly meaningful to us as physicians and public policy makers. Although many countries are seeking drastic solutions to reduce HIV incidence, we now feel close to having a viable solution. The results of this study are promising, as male circumcision programs are an effective means of HIV prevention when combined with safe sex education and counseling, and the PrePex device has the potential to enable large-scale circumcision programs in resource-limited areas. We thank our International Male Circumcision Advisory Committee members Dr Natan Bar-Chama, Mt Sinai Medical Center; Dr Steven Kaplan, Weill-Cornell Medical Center, Dr Emmanuel Kayibanda, President of the Surgical Society Rwanda; Dr Renee Ridzon, the Bill and Melinda Gates Foundation; Dr Thomas Serena, Serena Group of Advanced Wound Healing; and Dr Ira Sharlip, University of California San Francisco, for their invaluable contributions to this study and focusing our attention on the vital aspects of assessing a new medical device. We thank Dr Tim Farley, World Health Organization, for his technical assistance with the study; and Dr Kim Dickson, World Health Organization, and Dr Renee Ridzon, who visited Rwanda to witness procedures and follow-ups. Their dedication to finding solutions to stop the spread of HIV by male circumcision and other means is exceptional. We thank Tzameret Fuerst, CEO of Circ MedTech Limited, and the company, for donating devices and for their commitment to improve the device for the wellbeing of patients and convenience of health care providers. We thank Steven Tiger, PA, and his employer, Complete Healthcare Communications, Inc (Chadds Ford, PA), for contributing their time and editorial support gratis during the preparation of this article. Finally, we thank the Rwandan men who trusted us. They are the true heroes of this study, which could have an important impact on global public health.
1. Gray RH, Kigozi G, Serwadda D, et al. Male circumcision
for HIV prevention
in men in Rakai, Uganda: a randomised trial. Lancet. 2007;369:657–666
2. Bailey RC, Moses S, Parker CB, et al. Male circumcision
for HIV prevention
in young men in Kisumu, Kenya: a randomised controlled trial. Lancet. 2007;369:643–656
3. Auvert B, Taljaard D, Lagarde E, et al. Randomized, controlled intervention trial of male circumcision
for reduction of HIV infection risk: the ANRS 1265 Trial. PLoS Med. 2005;2:e298
4. Byakika-Tusiime J. Circumcision and HIV infection: assessment of causality. AIDS Behav. 2008;12:835–841
5. Mills E, Cooper C, Anema A, et al. Male circumcision
for the prevention of heterosexually acquired HIV infection: a meta-analysis of randomized trials involving 11,050 men. HIV Med. 2008;9:332–335
6. Siegfried N, Muller M, Deeks JJ, et al. Male circumcision
for prevention of heterosexual acquisition of HIV in men. Cochrane Database Syst Rev. 2009 CD003362
7. Uthman OA, Popoola TA, Yahaya I, et al. The cost-utility analysis of adult male circumcision
for prevention of heterosexual acquisition of HIV in men in sub-Saharan Africa
: a probabilistic decision model. Value Health. 2011;14:70–79
8. WHO/UNAIDS. . New data on male circumcision
and HIV prevention
: policy and programme implications [WHO/UNAIDS Technical Consultation] Presented at: male circumcision
and HIV prevention
: research implications for policy and programming; March 6–8, 2007; Montreux, Switzerland; Available at: http://data.unaids.org/pub/Report/2007/mc_recommendations_en.pdf
. Accessed August 15, 2010
9. Dickson KE, Reed J, Rech DWorld Health Organization. Considerations for Implementing Models for Optimizing the Volume and Efficiency of Male Circumcision
Services. February 2010 Geneva, Switzerland World Health Organization
10. World Health Organization. Manual for Male Circumcision
Under Local Anesthesia, Version 2.5C. 2008 Geneva, Switzerland World Health Organization
11. Rwanda Interim Demographic and Health Survey (2007–08). Ministry of Health of Rwanda, Kigali, Rwanda; National Institute of Statistics of Rwanda, Kigali, Rwanda; ICF Macro, Calverton, MD; April 2009
12. Peng YF, Cheng Y, Wang GY, et al. Clinical application of a new device
for minimally invasive circumcision. Asian J Androl. 2008;10:447–454
13. Ge CG, Liu DL. Clinical observation of adult circumcision with removal ring device
. Chin J Mod Med. 2005;15:3028–3030
14. Masson P, Li PS, Barone MA, et al. The ShangRing device
for simplified adult circumcision. Nat Rev Urol. 2010;7:638–642
15. Cheng Y, Peng YF, Liu YD, et al. A recommendable standard protocol of adult male circumcision
with the Chinese Shang Ring: outcomes of 328 cases in China. [in Chinese]. Zhonghua Nan Ke Xue. 2009;15:584–592
16. Li HN, Xu J, Qu LM. Shang Ring circumcision versus conventional surgical procedures: comparison of clinical effectiveness. [in Chinese]. Zhonghua Nan Ke Xue. 2010;16:325–327
17. Barone MA, Ndede F, Li PS, et al. The Shang Ring device
for adult male circumcision
: a proof of concept study in Kenya. J Acquir Immune Defic Syndr. 2011;57:E7–E12
18. Peng YF, Yang BH, Jia C, et al. Standardized male circumcision
with Shang Ring reduces postoperative complications: a report of 351 cases. [in Chinese]. Zhonghua Nan Ke Xue. 2010;16:963–966
19. . British Association of Paediatric Surgeons. Management of Foreskin Conditions. Statement from the British Association of Paediatric Urologists on behalf of the British Association of Paediatric Surgeons and The Association of Paediatric Anaesthetists Available at: http://www.baps.org.uk/documents/Circumcision_2007
. Accessed July 27, 2011
20. Perera CL, Bridgewater FH, Thavaneswaran P, et al. Safety and efficacy of nontherapeutic male circumcision
: a systematic review. Ann Fam Med. 2010;8:64–72
21. Krieger JN, Bailey RC, Opeya JC, et al. Adult male circumcision
outcomes: experience in a developing country setting. Urologia Int. 2007;78:235–240
22. Decastro B, Gurski J, Peterson A. Adult template circumcision: a prospective, randomized, patient-blinded, comparative study evaluating the safety and efficacy of a novel circumcision device
. Urology. 2010;76:810–814
23. Binagwaho A, Pegurri E, Muita J, et al. Male circumcision
at different ages in Rwanda: a cost-effectiveness study. PLoS Med. 2010;7:e1000211
Keywords:© 2011 Lippincott Williams & Wilkins, Inc.
Africa; bloodless; device; HIV prevention; male circumcision; nonsurgical