Neonatal herpes simplex virus (HSV) occurs in up to 1 per 3,200 live births,1 with an estimated incidence of 1,500 cases in the United States annually.2 Neonatal HSV causes disseminated or central nervous system disease in approximately 50% of cases; up to 30% of these neonates die and up to 40% of survivors have neurologic damage, despite antiviral therapy.2 Current guidelines for prevention of neonatal herpes recommend visual inspection for maternal herpetic genital lesions at the time of labor with cesarean delivery if lesions are present3; this strategy has not resulted in an appreciable reduction in incidence.4 The main mode of transmission to the neonate is exposure to infected maternal secretions at the time of passage through the birth canal. Isolation of HSV from the maternal genital tract at the time of labor is the primary risk factor for neonatal HSV with an estimated relative risk of greater than 300 for infection.1 As such, prevention of neonatal exposure to HSV during birth by identifying women with HSV in the genital tract during labor is a potential approach for reducing risk of infection. Interventions that could be tested among women shedding HSV during labor include cesarean delivery, antiviral prophylaxis for mother or the neonate or both, and enhanced observation for signs of neonatal HSV infection.
Currently, there is no clinically useful method to detect HSV in the genital tract at the time of labor. Lesions consistent with genital herpes indicate an increased risk of HSV; however, both sensitivity and specificity of genital lesions for viral shedding is poor.5–7 Most women with HSV in the genital tract at the time of labor are without signs or symptoms of genital herpes.1 In the past, viral cultures of genital swabs collected in the third trimester of pregnancy were used to identify women at risk of perinatal HSV transmission. However, results from these prenatal specimens rarely correlated with culture results of swabs collected at the time of delivery and thus were not helpful in predicting neonatal exposure.8 Therefore, this practice was abandoned in the late 1980s.3 Viral cultures to detect HSV from genital swabs collected at the time of labor are not practical because results are not available in a time frame that would allow for effective interventions. Moreover, false-negative viral cultures were reported in up to 30% cases of neonatal HSV. In most reports, polymerase chain reaction (PCR)-based assays detected HSV in these negative viral cultures.1,5
Real-time quantitative PCR has proved to be sensitive and reproducible in the diagnosis and management of viral diseases.9 However, due to the complexity of the procedures involved and the potential of contamination, clinical diagnostic PCR usually is performed in state-of-the-art clinical laboratories. Transfer of specimens from clinical sites to comprehensive clinical laboratories often takes hours or even days. Once received, the entire analytic procedure, including nuclear acid extraction and 45 cycles of Taqman PCR, takes about 4 hours. Although point-of-care molecular diagnostic systems are available for adaptation,10 to adopt one of these systems for specific pathogen detection will heavily involve the commercial proprietary issues, higher cost per test, and less flexibility for any in-house development. As such, we chose to develop an in-house, sensitive, specific, rapid, automated, self-contained PCR assay using the extraction and PCR equipment available commercially that could be utilized to detect HSV-1 or HSV-2 from samples taken in the delivery room and reported back rapidly (within 2 hours) to clinicians providing care to the laboring woman and newborn. The assay is designed to be utilized by any trained laboratory technician, so as to be feasibly implemented in at any hour of the day in a clinical laboratory serving obstetric patients.
MATERIALS AND METHODS
Swab samples for analysis were derived from three different sources. First, a buffer-only solution was used as a negative control group (n=299). For this group, vials were filled with 1 mL of buffer solution and a sterile swab was inserted. Second, swabs were collected from nonpregnant patients with herpetic genital lesions (n=211) who presented to the Virology Research Clinic and placed in a vial containing 1 mL of PCR digestion buffer. Third, mucocutaneous genital swab samples from pregnant women presenting in labor at the University of Washington Medical Center were included (n=206). These swabs were collected between October 2006 and March 2009 from women who were aged 18 or older, presented in labor, and understood English. Verbal consent was obtained for collection of genital secretions on sterile polyester-tipped applicators (Puritan Medical Products, Guilford, ME) at the time of clinically indicated routine speculum examination at admission. Each swab was placed in a vial containing 1 mL of PCR digestion buffer; no identifying information was retained with the sample so that the samples were anonymous. University of Washington Institutional Review Board approved all study procedures.
To make the rapid PCR procedure as simple as possible to be carried out in a less stringent laboratory setting and performed by less experienced technologists, the reagents for the samples and controls were prepared and aliquoted by experienced PCR technologists in the Molecular Diagnostic Laboratory. All the reagents were quality control tested before being sent to the Microbiology Laboratory where the rapid PCR procedure was performed.
Each aliquot contained enough reagents to perform a single run of PCR reactions for four clinical samples, one negative control sample, one positive control sample, no DNA control, and five standard curve points in duplicate. An appropriate amount of ABI Taqman Fast Universal PCR Master Mix (2×) (Applied Biosystems, Inc., Foster City, CA) also was pre-aliquoted accordingly.
DNA was extracted from 200 microliters of buffer using the EZ1 Virus Kit on Qiagen BioRobot EZ1 (Qiagen, Germantown, MD) and eluted into 100 milliliters of AVE buffer (Qiagen, Inc., Valencia, CA). This machine was chosen because of its speed, small size, ability to extract six samples simultaneously, ease of operation, and minimization of cross-contamination by use of a single reagent cartridge for each sample. Ten microliters of eluted DNA was then used for PCR. To control contamination and DNA recovery, one negative control and one positive control were included in each extraction run.
The ABI7500 Fast machine (Applied Biosystems) was used to run PCR reactions. The glycoprotein B gene primers and probe were used for this study as described previously.11,13 These primers and probes pick up both HSV-1 and HSV-2 and have been validated in a variety of clinical and investigative studies.11,14,15 The glycoprotein B probe was labeled with FAM (6-carboxyfluorescein) at 5′-end and TAMRA (6-carboxy-tetramethyl-rhodamine) at 3′-end. Each 30-microliter rapid HSV PCR reaction contained 15 microliters of 2× ABI Taqman Fast Univeral PCR Master Mix, 830 nM primer, 100 nM probe. The PCR conditions were 95°C for 20 seconds, followed by 45 cycles of 95°C for 3 seconds and 60°C for 30 seconds. All the PCR reactions were spiked with internal control, as described below, to monitor for false-negative results. The rapid test was run in duplicate wells, and we reported the average value of the two wells when both wells were positive. However, to determine the cutoff for positivity, we considered the sensitivity and specificity with requiring only one compared with both wells to be positive. The duration of the procedure was recorded for each PCR run.
After the rapid real-time HSV PCR was performed at the Microbiology Laboratory, samples were sent to Molecular Diagnostic Laboratory where they were reanalyzed using the standard real-time TaqMan HSV PCR assay.11 Briefly, DNA was extracted from 200-microliter buffer using Qiagen QIAmp 96 Blood kit and eluted into 100 microliters. Ten microliters of DNA was then used for real-time TaqMan PCR with glycoprotein primer/probe. Each 30-microliter reaction contained 15 microliters of QuantiTech Multiplex PCR Master Mix (2×) (Qiagen), and 830 nM primer, 100 nM probe, and internal control consisting of a jellyfish DNA fragment as template, specific primers, and a specific probe. ABI 7900 HT was used to run the assay. The PCR conditions were 52°C for 2 minutes, 95°C for 15 minutes, followed by 45 cycles of 94°C for 1 minute and 60°C for 1 minute. Each PCR run contained several negative controls, including two reaction mixtures without DNA and several specimens that are known to contain no HSV DNA, a positive control, and a standard dilution curve DNA.
We hypothesized that results of the rapid HSV PCR would not vary significantly from those of standard 4-hour HSV PCR. An a priori sample size calculation determined that analyzing 650 specimens could determine a specificity of at least 97% with a precision of 1.5%, assuming that 518 of these specimens did not contain HSV using standard HSV PCR as the gold standard. Positivity determination that optimized sensitivity and specificity was determined with receiver operating characteristic curves. Under any reasonable sensitivity assumptions (at least 80%), specificity of 97% provides a positive predictive value of at least 76% at a baseline HSV standard PCR positive rate of 10.8%, a rate of positivity observed in our previous cohort of pregnant women (unpublished data).1
We estimated the sensitivity and specificity of the rapid test to detect HSV DNA, using the standard TaqMan HSV PCR result as the gold standard and obtained confidence intervals using the logit transformation. A TaqMan HSV PCR result of greater than 150 HSV copies/mL was considered positive, consistent with the cutoff currently utilized in our standard PCR assay for clinical diagnoses.12 We ascertained whether the results of the rapid test and the standard TaqMan test varied significantly for each specimen and describe the distribution of virion copy numbers of HSV DNA for specimens with detectable HSV DNA. Receiver operating characteristic curves were used to evaluate the summary measure for the two wells and the range of potential cutoffs for positivity determination. Since both the TaqMan and the Rapid PCR quantitate copies of HSV DNA per milliliter, we compare both the quantity obtained as well as the binary indicator of response: positive or negative. Positive predictive values and negative predictive values of the assay were estimated in the target population of interest (laboring women) based on demonstrated accuracy over all samples and the previously observed prevalence of HSV DNA.
We tested 732 swabs on 732 unique persons; 16 (2%) swabs were excluded because the negative controls in the runs were positive, and the remaining 716 were valid and used for analyses. The median time to perform the Rapid Test from initiation of test procedures to result was 2 hours (interquartile range 1.75–2.33 hours, range 1.5–3.5 hours).
Using a cutoff of 150 copies/mL, our Rapid PCR assays detected HSV DNA in 207 of 716 (28.9%, 95% confidence interval [(CI] 25.7–32.3%) swabs, compared with 212 of 716 (29.6%, 95% CI 26.4–33.1%) in the TaqMan assay. Below, we provide additional detail on sensitivity and specificity that result in establishment of the cutoff for PCR positivity as both wells positive at 150 copies/mL or greater.
We evaluated the specificity of the Rapid PCR using samples containing buffer only. Seven (7 of 299, 2.3%, 95% CI 1.1–4.8%) samples yielded HSV in at least one well. None of the seven was positive in the TaqMan assay. However, the median amount of HSV DNA in the samples with HSV detected was low (106 copies/mL, range 47–147 copies/mL). In addition, none of the buffer-only samples had HSV detected in both wells. Thus, using positivity in both wells to define a positive test minimized the potential for false-positive test results related to low-level contamination.
Two hundred eleven swabs were obtained from genital lesions, and 202 (95.7%, 95% CI 92.0–97.8%) tested positive by the Rapid Assay. The concordance between Rapid PCR and our standard TaqMan results varied by the cutoff for the Rapid Test. For example, at a cutoff for Rapid PCR at 150 copies/mL, both tests were positive in 201 samples, neither in eight samples, and the results were discordant in two samples, one of which was positive only by Rapid PCR and the other only by TaqMan. Thus, overall, 209 samples (99.1%, 95% CI 96.3–99.8%) had concordant results.
We collected 206 samples from pregnant women presenting to Labor and Delivery; five (2.4%, 95% CI 1.0–5.7%) tested positive by the Rapid assay and 10 (4.9%, 95% CI 2.6–8.8%) tested positive by TaqMan. Using 150 copies/mL as the cutoff for positive results and requiring both wells to be positive for Rapid PCR, both tests were positive in four samples, both tests were negative in 195 samples, and the results were discordant in seven, with only Rapid PCR being positive in one and only TaqMan positive in six. The median amount of HSV DNA in the samples that were positive by TaqMan only was 431 copies/mL (range 160–1901 copies/mL). The amount of HSV DNA in the one sample that was positive solely in the rapid assay was 596 copies/mL.
Using the above data in aggregate, we determined the sensitivity and specificity of the test using three different interpretation strategies applied to cutoff values from 50 to 500 copies/mL (Table 1). Strategy A averaged the quantitative values of both wells of the rapid test into a single value that is compared with the cutoff value to determine test positivity. Strategy B used the quantitative data from both wells separately with both wells required to exceed the cutoff value of 150 copies/mL for the specimen to be considered positive. Strategy C used the average of both wells after excluding specimens with one positive and one negative well, with positive defined as greater than 0 copies/mL. As shown in Figure 1 and Table 1, comparisons of accuracy rates suggest that strategy B that requires both wells to detect 150 copies or greater of HSV per milliliter to consider the test positive maximized specificity (96.7%, 95% CI 93.2–98.4%), without appreciable loss of sensitivity (99.6%, 95% CI 98.4–99.9%). The receiver operating characteristic curves shown in Figure 1 indicate that strategy B (the gray dashed line) is closer to the top left corner at all cutoffs relative to strategy A and that the standard Taqman cutoff of 150 copies is near optimal.
Among the Rapid PCR results, the quantitative difference between the two wells run for each sample varied from 0 to 0.8 log copies/mL (Fig. 2A). Among positive samples (150 or greater copies/mL) in both wells (n=207), the correlation between the number of copies of HSV DNA detected by Rapid PCR and TaqMan was excellent with a Spearman correlation coefficient of 0.99, P<.001 (Fig. 2B).
Using the accuracy rates for each cutoff (Table 1), we also computed the expected positive predictive value and negative predictive value for a prevalence of 10.8%, which was the prevalence of HSV shedding noted in our studies among HSV-2 seropositive women in labor and without genital lesions (unpublished data).1 We calculated a positive predictive value of 96.7% and negative predictive value of 99.6% using strategy B (the cutoff of 150 copies/mL and both wells positive). Estimates of positive and negative predictive value using other cutoff values at both a 10.8% and 5% prevalence (for comparison) are shown in Table 2.
We developed and validated a sensitive and specific rapid HSV real-time PCR assay procedure that can be performed in 2 hours from receipt of sample to result reporting. By using the automatic extraction machine, prepacked nuclear acid extraction reagents, and premixed and pre-aliquoted PCR reagents, the procedure is self contained and can be performed in a regular laboratory by technologists with very little PCR training. Although the procedure is not performed by the clinical staff within the vicinity of the patients as other point-of-care devices, as defined by the College of American Pathologists, the procedure is a useful point-of-care alternative since the technique is easy and fast, and the assay is sensitive and specific.
Weighing the consequences of a false positive, which may potentially expose the patient to unnecessary interventions, and a false negative, which may result in transmission of neonatal HSV, we determined that positivity in both wells maximized specificity and low copy number cutoff maximized sensitivity. The threshold amount of virus that may result in transmission is unknown; however, we have reported transmission of culture negative, PCR-positive cases at between 500 and 1,000 copies of DNA/swab sample in our standard TaqMan assay.1
Ideally, a rapid test for HSV detection in labor should be combined with a serologic test for HSV-2 antibodies. Those women who lack HSV-2 antibodies but are shedding HSV at the time of labor are at very high risk (30–50%) of transmitting the infection to their neonate.1 In this situation, cesarean delivery is appropriate, and some experts would also administer acyclovir to the neonate, especially if the membranes were ruptured.13 The risk of HSV transmission to the neonate is much lower if the woman has antibodies to HSV-2 and the neonate is exposed to HSV at the time of delivery. A course of intrapartum acyclovir may reduce further the already low risk of neonatal infection. Our approach would parallel the successful strategies for group B streptococcus prevention14 and human immunodeficiency virus (HIV) prevention,15 in which providing the neonate with pre-exposure prophylaxis with an antibiotic or an antiviral prevents the pathogen from establishing productive infection. In animal models, preloading mice with acyclovir before HSV exposure prevents infection,16 although the relevance of these models to human HSV infection is unclear. Our rapid assay was developed as a tool to help organize scientific evaluations to define how best to manage neonates exposed to HSV at delivery.
Our approach uses a type-common probe. Although type-specific PCR assays have been developed, they may be somewhat less sensitive,17 and it is not clear that such are needed for this use. Women who have HSV-2 detected in the genital tract are at high risk of infecting their newborns if they are seronegative and experiencing a new HSV infection. In contrast, women who have HSV-1 detected in the genital tract are at high risk of infecting their newborns regardless of whether they have new HSV-1 infection or reactivation of HSV-1.1 As such, detection of HSV in the genital tract during labor, combined with a negative serologic test for HSV-2, represents a high risk scenario regardless of whether the virus detected is HSV-1 or HSV-2. In contrast, women who are HSV-2 seropositive and are shedding HSV in labor are at low, albeit not absent, risk.
Studies that have evaluated the interest of women in being tested for HSV during pregnancy suggest that women are willing and that some expect it as part of the routine care.21,22 Even when asked specifically about a hypothetical HSV test that would be administered during labor, 85% of women expressed interest.18 While theoretical agreement to testing may not always correspond to consent for actual testing, experience at our institution where HSV antibody testing has been incorporated into routine prenatal care suggests that women rarely object to this test, which is similar to the high acceptability of HIV testing when incorporated into routine medical care for pregnant women.19 Concerns about routine serologic testing for HSV have been raised, both from the point of view of shortcomings of the commercially available serologic assays, as well as cost-benefit calculations. The use of rapid PCR for HSV in labor can obviate some of these concerns because only women who are HSV PCR positive need to be tested for HSV-2 antibody, and rapid bedside antibody tests are commercially available.25,26
In summary, we have developed a rapid test for HSV in genital secretions that can give results within 2 hours, a time frame that allows the physician to institute appropriate interventions to reduce the risk of HSV transmission to the neonate. While the technical development of this test needs to be followed by clinical studies among pregnant women, and appropriate management strategies for women who are HSV positive in labor need to be evaluated, we hope that this tool will allow clinical studies to be conducted toward defining effective strategies for reducing the burden of neonatal HSV in the United States.
1. Brown ZA, Wald A, Morrow RA, Selke S, Zeh J, Corey L. Effect of serologic status and cesarean delivery on transmission rates of herpes simplex virus from mother to infant. JAMA 2003;289:203–9.
2. Kimberlin DW. Neonatal herpes simplex infection. Clin Microbiol Rev 2004;17:1–13.
3. Practice Bulletin. Management of genital herpes in pregnancy. ACOG Practice Bulletin No. 81. American College of Obstetricians and Gynecologists. Obstet Gynecol 2007;109:1233–48.
4. Morris SR, Bauer HM, Samuel MC, Gallagher D, Bolan G. Neonatal herpes morbidity and mortality in California, 1995–2003. Sex Transm Dis 2008;35:14–8.
5. Cone RW, Hobson AC, Brown Z, Ashley R, Berry S, Winter C, et al. Frequent detection of genital herpes simplex virus DNA by polymerase chain reaction among pregnant women. JAMA 1994;272:792–6.
6. Waldhuber MG, Denham I, Wadey C, Leong-Shaw W, Cross GF. Detection of herpes simplex virus in genital specimens by type-specific polymerase chain reaction. Int J STD AIDS 1999;10:89–92.
7. Gardella C, Brown ZA, Wald A, Morrow RA, Selke S, Krantz E, Corey L. Poor correlation between genital lesions and detection of herpes simplex virus in women in labor. Obstet Gynecol 2005;106:268–74.
8. Arvin AM, Hensleigh PA, Prober CG, Au DS, Yasukawa LL, Wittek AE, et al. Failure of antepartum maternal cultures to predict the infant's risk of exposure to herpes simplex virus at delivery. N Engl J Med 1986;315:796–800.
9. Ryncarz AJ, Goddard J, Wald A, Huang ML, Roizman B, Corey L. Development of a high-throughput quantitative assay for detecting HSV DNA in clinical samples. J Clin Microbiol 1999;37:1941–7.
10. Holland CA, Kiechle FL. Point-of-care molecular diagnostic systems–past, present and future. Curr Opin Microbiol 2005;8:504–9.
11. Jerome KR, Huang ML, Wald A, Selke S, Corey L. Quantitative stability of DNA after extended storage of clinical specimens as determined by real-time PCR. J Clin Microbiol 2002;40:2609–11.
12. Magaret AS, Wald A, Huang ML, Selke S, Corey L. Optimizing PCR positivity criterion for detection of herpes simplex virus DNA on skin and mucosa. J Clin Microbiol 2007;45:1618–20.
13. Pediatrics AA. Herpes Simplex. In: Pickering LK, ed. Red Book: 2009 Report of the Committee on Infectious Diseases. 28th ed. Elk Grove Village (IL): American Academy of Pediatrics; 2009. p. 363–73.
14. Schuchat A. Neonatal group B streptococcal disease–screening and prevention. N Engl J Med 2000;343:209–10.
15. Achievements in public health. Reduction in perinatal transmission of HIV infection–United States, 1985–2005. MMWR Morb Mortal Wkly Rep 2006;55:592–7.
16. Dobson AT, Little BB, Scott LL. Prevention of herpes simplex virus infection and latency by prophylactic treatment with acyclovir in a weanling mouse model. Am J Obstet Gynecol 1998;179:527–32.
17. Corey L, Huang ML, Selke S, Wald A. Differentiation of herpes simplex virus types 1 and 2 in clinical samples by a real-time taqman PCR assay. J Med Virol 2005;76:350–5.
18. Schalkwyk J, Amiri N, Lalji S, Gardella C, Wald A, Money D. Acceptance of a rapid herpes test in labour: survey of attitudes of patients and health care providers. J Obstet Gynaecol Can 2008;30:776–80.
© 2010 by The American College of Obstetricians and Gynecologists.
19. Chou R, Smits AK, Huffman LH, Fu R, Korthuis PT, US Preventive Services Task Force. Prenatal screening for HIV: A review of the evidence for the U.S. Preventive Services Task Force. Ann Intern Med 2005;143:38–54.