Accuracy of TempZHF during SCTC or RCTC was compared with using the paired Student's t test corrected for multiple testing, with a P value equal to or lower than 0.05 considered as statistically significant.
The number and percentage (with 95% CI) of absolute difference of temperature pairs between reference and tested methods equal to or lower than 0.3, 0.5 and 0.8 °C were also determined.
Data were entered into Excel 2007 (Microsoft Office Excel 2007; Microsoft Corp., Redmond, Washington, USA) and statistical analysis using R software version 3.2.2 (R Development Core Team: R: A Language and Environment for Statistical Computing, Vienna, Austria, R Foundation for Statistical Computing, 2008. Available at: http://www.R-project.org).
Between June 2015 and March 2016, 50 patients were enrolled in the study; 22 were monitored with a thermodilution peripheral arterial catheter. One patient was excluded due to data recording problems. Consequently, the data from 49 patients were analysed, including fifteen treated with HIPEC. Their characteristics are summarised in Table 1.
Comparison between peripheral arterial catheter and oesophageal probe
A total of 1358 pairs of TempArt and TempEso (1281 during the SCTC period and 77 during the RCTC period) were collected, ranging from 33.4 to 38.8 °C. Compared with TempArt, bias and limits of agreement for TempEso were 0.1 ± 0.3 °C during the SCTC period and 0.0 ± 0.8 °C during the RCTC period (P = 0.725).
Comparison between zero-heat-flux and oesophageal probe
A total of 2446 pairs of TempZHF and TempEso (2345 during the SCTC period and 101 during the RCTC period) were collected, ranging from 34.0 to 40.2 °C. Compared with TempEso, bias and limits of agreement for TempZHF were 0.1 ± 0.5 °C during the SCTC period and 0.6 ± 1.8 °C during the RCTC period (P = 0.0002, Fig. 1). The number of temperature pairs with an absolute difference of 0.3, 0.5 and 0.8 °C or less is presented in Table 2.
Comparison between zero-heat-flux and peripheral arterial catheter
A total of 1780 pairs of TempZHF and TempArt (1708 during the SCTC period and 72 during the RCTC period) were collected, ranging from 33.0 to 39.8 °C. Compared with TempArt, bias and limits of agreement for TempZHF were −0.1 ± 0.4 °C during the SCTC period and 0.5 ± 1.7 °C during the RCTC period (P = 0.0039, Fig. 2). The number of temperature pairs with absolute difference of 0.3, 0.5 and 0.8 °C or less is presented in Table 2.
The ZHF sensors were well tolerated in all patients and no manifestation of burn or anaphylaxis was observed during the study period.
The current study performed in patients under anaesthesia during abdominal surgery showed that the ZHF device has good accuracy compared with an oesophageal probe or arterial catheter only during periods of slow core temperature changes. The ZHF sensors were well tolerated with no safety issue reported.
Thus, during periods of slow core temperature changes – which involve the majority of patients undergoing surgery – biases and limits of agreement of the ZHF method were equal to 0.1 ± 0.5 and −0.1 ± 0.4 °C compared with the oesophageal probe and the arterial catheter, respectively. Furthermore, in 94% (oesophageal probe) and 99% (arterial catheter) of measurements, absolute differences of temperature pairs were equal to or lower than 0.5 °C. Studies performed in cardiac surgical patients,19,20 gynaecological or trauma surgical patients23 and intensive care adults24 comparing the ZHF method to invasive methods (pulmonary artery catheter, oesophageal probe or femoral artery catheter) have reported comparable results. Eshraghi et al. compared a prototype 3M SpotOn sensor and a pulmonary artery catheter intra-operatively and for the first 4 postoperative hours. During surgery, they observed a limited bias at −0.1 ± 0.9 °C with absolute differences of temperature pairs equal to or lower than 0.5 °C in 84% of cases. Significantly, due to limited or absent blood flow in the pulmonary artery during cross-clamp, they excluded the CPB period from their analysis.19 In elective gynaecological and trauma surgery, Iden et al.23 compared temperature monitoring noncontinuously using a SpotOn sensor and a nasopharyngeal probe and found a mean bias at 0.1 ± 0.4 °C.
Our study highlighted some limitations on the reliability of the ZHF method during periods of rapid change in core temperature such as those caused by HIPEC. During periods of core temperature changes equal to or more than 1 °C within 30 min, biases and limits of agreement increased substantially and reached 0.6 ± 1.8 and 0.5 ± 1.7 °C compared with oesophageal probe and arterial catheter, respectively, with absolute differences of temperature pairs equal to or lower than 0.5 °C in less than 40% of cases. Few studies focussed on the reliability of the ZHF method during rapid changes in core temperature and, to the best of our knowledge, SpotOn has not been tested yet during HIPEC. Even so, rapid change in core temperature is not an uncommon phenomenon. In fact, during the first hour of general anaesthesia without warming manoeuvres, core temperature rapidly decreases by 1 to 1.5 °C.25 In rare but feared malignant hyperthermia, core temperature rapidly increases by around 2 °C per hour.26 In cardiac surgery, Akata et al. compared another ZHF device with jugular bulb temperature and pulmonary arterial catheters during deep hypothermic CPB. With a mean bias of 4.7 °C in comparison with jugular bulb temperature during induction of deep hypothermia, they concluded that the ZHF method was not reliable for brain temperature measurement.17 In a recent study, Mäkinen et al.20 compared the 3M SpotOn with nasopharyngeal temperature in 15 patients undergoing cardiac surgery with hypothermic CPB. Before and after CPB, bias and limits of agreement were close to our results (−0.10 ± 0.59 °C). During CPB, which corresponds to a phase of rapid changes in core temperature, agreement was weaker (−0.14 ± 1.09 °C), but the difference was not as big as in our study. The researchers did not report the range of absolute differences in temperature pairs. Eshraghi et al.19 excluded CPB from their analyses and explained their only slightly elevated limits of agreement of 0.82 °C – compared with pulmonary arterial catheter – by the many rapid thermal perturbations occurring during cardiac surgery. In critically ill patients, Dahyot-Fizelier et al.24 reported stable bias and precision of the ZHF method over temperatures ranging from 33.0 to 39.7 °C. Consequently, increased agreement during the rapid phase does not generally appear to be linked with extreme temperatures. Our results were not consistent with those of a previous study on healthy volunteers conducted by Teunissen et al.18 using a prototype ZHF sensor. In an experimental study performed in ambient conditions of 35 °C, the authors compared the accuracy of the ZHF method with an oesophageal probe during stable (at rest) and rapid changes in core temperature (during exercise, with an increase in core temperature of 1.5 °C within 30 min). They reported a comparable bias and limits of agreement (0.17 ± 0.19 °C at rest and −0.05 ± 0.18 °C during exercise). This difference could be explained by different experimental conditions and by the use of a different ZHF sensor.
A ZHF sensor insulates the skin locally, thereby creating a region of zero heat flow from the body core to the skin.12 Limitations on the reliability of the ZHF method during RCTC could be due to equilibration duration between core temperature and the insulated skin under the sensor. During HIPEC, rapid changes in core temperature are due to warming of the abdominal cavity which could explain differences between the oesophageal probe and the forehead ZHF sensor. But due to similar differences between ZHF and axillary arterial catheter, sensor location did not appear as the only explanation.
As previously reported, the ZHF sensor is well tolerated,24 but we were not able to effectively evaluate safety. More studies are warranted to address this issue.
Our study has some limitations. First, we did not check the position of the oesophageal probe with a chest radiograph, even though its position in the oesophagus could affect its accuracy.27 To limit this bias, the oesophageal probe was inserted through a nostril into the oesophagus under direct vision during laryngoscopy and the length of insertion was determined according to Mekjavić and Rempel21 formula. Second, as regards rapid temperature changes, the number of temperature measurement pairs was rather low (around 4% of total pairs recorded), due to our having preferred to observe a 5-min interval between temperature measurements to ensure that potential changes of core temperatures had sufficient time to occur. Third, we used a thermodilution arterial catheter inserted in the axillary artery as our method of reference, but the accuracy of PICCO (Pulsion Medical Systems AG, Munich, Germany) in temperature monitoring has been evaluated only in femoro-iliac artery.7
The major strengths of our study were continuous temperature recording and the participation of a substantial number of patients, which ensured a reliable comparison between the different devices.
To conclude, our study confirmed that for monitoring core temperature during abdominal surgery during periods of slow core temperature changes the SpotOn sensor using the ZHF method has good reliability in comparison with an oesophageal probe or a specific arterial catheter. However, SpotOn sensor accuracy is questionable during RCTC.
Acknowledgements relating to this article
Assistance with the study: we would like to thank Seguin Sabrina for her assistance with the study and Jeffrey Arsham for his assistance with the article.
Financial support and sponsorship: the study was supported by the University Hospital of Poitiers, the SpotOn temperature sensors and monitors were provided by 3M (3M, St Paul, MN, USA). The sponsor and 3M were not involved in study design, collection, analysis and interpretation of data, writing of article or in the decision to submit the article for publication.
Conflicts of interest: MB received travel expenses from Pulsion. OM has received honoraria to present work at a symposium and to attend an international conference supported by 3M and has also received honoraria for attendance at advisory board meetings.
Presentation: preliminary data were presented at the European Society of Anaesthesiology congress (Euroanaesthesia), 28 to 30 May 2016 and at the French Society of Anaesthesiology and Intensive Care (SFAR), 22 to 24 September 2016.
1. Sessler DI. Temperature monitoring and perioperative thermoregulation. Anesthesiology
2. Frank SM, Fleisher LA, Breslow MJ, et al. Perioperative maintenance of normothermia reduces the incidence of morbid cardiac events. A randomized clinical trial. JAMA
3. Schmied H, Kurz A, Sessler DI, et al. Mild hypothermia increases blood loss and transfusion requirements during total hip arthroplasty. Lancet
4. Rajagopalan S, Mascha E, Na J, et al. The effects of mild perioperative hypothermia on blood loss and transfusion requirement. Anesthesiology
5. Kurz A, Sessler DI, Lenhardt R. Perioperative normothermia to reduce the incidence of surgical-wound infection and shorten hospitalization. Study of Wound Infection and Temperature Group. N Engl J Med
6. Hypothermia: prevention and management in adults having surgery [Internet]. London: NICE National Institute for Health and Clinical Excellence; 2008 Available from: https://www.nice.org.uk/guidance/CG65
. [Accessed 17 April 2018]
7. Krizanac D, Stratil P, Hoerburger D, et al. Femoro-iliacal artery versus pulmonary artery core temperature measurement during therapeutic hypothermia: an observational study. Resuscitation
8. Robinson JL, Seal RF, Spady DW, et al. Comparison of esophageal, rectal, axillary, bladder, tympanic, and pulmonary artery temperatures in children. J Pediatr
9. Lefrant J-Y, Muller L, de La Coussaye JE, et al. Temperature measurement in intensive care patients: comparison of urinary bladder, oesophageal, rectal, axillary, and inguinal methods versus pulmonary artery core method. Intensive Care Med
10. Giuliano KK, Scott SS, Elliot S, et al. Temperature measurement in critically ill orally intubated adults: a comparison of pulmonary artery core, tympanic, and oral methods. Crit Care Med
11. Höcker J, Bein B, Böhm R, et al. Correlation, accuracy, precision and practicability of perioperative measurement of sublingual temperature in comparison with tympanic membrane temperature in awake and anaesthetised patients. Eur J Anaesthesiol
12. Fox RH, Solman AJ, Isaacs R, et al. A new method for monitoring deep body temperature from the skin surface. Clin Sci
13. Singer B, Lipton B. Monitoring of core temperature through the skin: a comparison with esophageal and tympanic temperatures. Bull N Y Acad Med
14. Muravchick S. Deep body thermometry during general anesthesia. Anesthesiology
15. Sakuragi T, Mukai M, Dan K. Deep body temperature during the warming phase of cardiopulmonary bypass. Br J Anaesth
16. Harioka T, Matsukawa T, Ozaki M, et al. ‘Deep-forehead’ temperature correlates well with blood temperature. Can J Anaesth
17. Akata T, Setoguchi H, Shirozu K, et al. Reliability of temperatures measured at standard monitoring sites as an index of brain temperature during deep hypothermic cardiopulmonary bypass conducted for thoracic aortic reconstruction. J Thorac Cardiovasc Surg
18. Teunissen LPJ, Klewer J, de Haan A, et al. Noninvasive continuous core temperature measurement by zero heat flux. Physiol Meas
19. Eshraghi Y, Nasr V, Parra-Sanchez I, et al. An evaluation of a zero-heat-flux cutaneous thermometer in cardiac surgical patients. Anesth Analg
20. Mäkinen MT, Pesonen A, Jousela I, et al. Novel zero-heat-flux deep body temperature measurement in lower extremity vascular and cardiac surgery. J Cardiothorac Vasc Anesth
21. Mekjavić IB, Rempel ME. Determination of esophageal probe insertion length based on standing and sitting height. J Appl Physiol
22. Bland JM, Altman DG. Agreement between methods of measurement with multiple observations per individual. J Biopharm Stat
23. Iden T, Horn E-P, Bein B, et al. Intraoperative temperature monitoring with zero heat flux technology (3M SpotOn sensor) in comparison with sublingual and nasopharyngeal temperature: an observational study. Eur J Anaesthesiol
24. Dahyot-Fizelier C, Lamarche S, Kerforne T, et al. Accuracy of zero-heat-flux cutaneous temperature in intensive care adults. Crit Care Med
25. Sessler DI, Olofsson CI, Rubinstein EH. The thermoregulatory threshold in humans during nitrous oxide-fentanyl anesthesia. Anesthesiology
26. Ramirez JA, Cheetham ED, Laurence AS, et al. Suxamethonium, masseter spasm and later malignant hyperthermia. Anaesthesia
© 2018 European Society of Anaesthesiology
27. Kaufman RD. Relationship between esophageal temperature gradient and heart and lung sounds heard by esophageal stethoscope. Anesth Analg