For statistical analysis, IBM-SPSS version 22.0 for Windows (SPSS, Chicago, Illinois, United States) was used. Pre-experimental sample size calculation was not performed because reference data on ‘measured leakage volume’ were not available. Comparisons between tracheal tubes were performed with the Kruskal–Wallis test followed by pairwise Mann–Whitney U tests. Data are presented as medians (range). Statistical significance was accepted at a P value < 0.05.
In the static model, leakage started earlier with PVC cuffs compared with polyurethane cuffs: 2 (1 to 3) vs. 60 (33 to 136) s, PVCcyl (Portex Soft-Seal) vs. PUcyl (Microcuff) P = 0.002 and 5 (3 to 37) vs. 76 (25 to 720) s, PVCcon (Taperguard) vs. PUcon (Sealguard) P = 0.009. The leakage times were also statistically significantly different between the PVC cylindrical and conical cuffs (P < 0.004) but not between the polyurethane cylindrical and conical cuffs (P < 0.09). During the 1-h study period, the PVC cuffs with conical form allowed less dye to pass than the cylindrical ones: 9.8 (6.2 to 20) vs. 1.3 (0.2 to 3.8) ml min−1, PVCcyl (Portex Soft-Seal) vs. PVCcon (Taperguard) P = 0.004). No difference in leakage flow was noted between cylindrical and conical PU cuffs: 0.03 (0.007 to 0.1) vs. 0.04 (0.003 to 0.2) ml min−1, PUcyl (Microcuff) vs. PUcon (Sealguard) P = 0.94. Regardless of the cuff form, the polyurethane material consistently protected better against leakage than PVC: P < 0.004 for the difference between PVC and polyurethane cylindrical tracheal tubes and P < 0.005 for the difference between PVC and polyurethane conical cuffs. No leakage flow [0.0 (0.0 to 0.0) ml min−1] was observed when the PVCdc tracheal tube was used (P < 0.001, PVCdc vs. all other cuffs).
In the dynamic setting, no leakage was detected for up to 60 min with any of the cuffs studied. Only after loss of PEEP or tracheal tube disconnection did dye inflow occur alongside all cuffs except for the PVCdc (P < 0.001, PVCdc vs. all other cuffs). Leakage was rapidly detected with the PVC-cuffed tubes, but occurred significantly later with polyurethane cuffs: 2 (1 to 4) vs. 55 (15 to 120) s, PVCcyl (Portex Soft-Seal) vs. PUcyl (Microcuff) and 3 (2 to 5) vs. 114 (36 to 264) s, PVCcon (Taperguard) vs. PUcon (Sealguard); both P = 0.002 but was not influenced by cuff shape: P = 0.16 and P = 0.26 for comparisons between PVC cuffs and between polyurethane cuffs, respectively.
As regards, the prevention of leakage in a static trachea model, the prototype tracheal tube outperformed all the commercially available PVC and polyurethane-cuffed tracheal tubes studied. Under dynamic conditions, although PEEP guaranteed adequate sealing in all tracheal tubes, only the prototype tube protected against leakage after release of PEEP or tracheal tube disconnection.
The two most important mechanisms implicated in the development of tracheal tube-related VAP are biofilm formation inside the lumen of the tracheal tube and micro-aspiration around the cuff.8 Interventions to prevent biofilm formation include either antiseptic coating of the tracheal tube or mechanical removal of the biofilm. Within this context, the largest study to date demonstrated a decrease in microbiologically proven VAP in patients intubated with a silver-coated tracheal tube compared with a standard tracheal tube. However, no difference was observed in relevant outcome parameters such as time on the ventilator, length of ICU stay or mortality. Also, a worrisome, and unexplained, trend toward a higher mortality in patients randomised to the silver-coated tracheal tube was noticed.9
Experimental and clinical research has focused extensively on the prevention of airway intrusion by bacteriologically contaminated secretions that accumulate above the inflated cuff of the tracheal tube. Suggested solutions to decrease or avoid leakage and subsequent aspiration include preservation of a constant pressure (20 to 30 cmH2O) inside the cuff, the use of specific cuff materials (e.g., polyurethane) or shapes (e.g., conical) and tracheal tubes equipped with a suction lumen for aspiration of subglottic secretions.10
Madjdpour et al.11 demonstrated better air-sealing properties of conical-shaped vs. cylindrical-shaped cuffs as well as superior sealing characteristics of PU compared with PVC material. With regard to preventing fluid ingress past tracheal tube cuffs, our study results in particular under static conditions, point in the same direction and may add theoretical support for the preferential use of tracheal tubes with conical-shaped polyurethane cuffs. However, the clinical advantage of such tracheal tubes is less convincing. Poelaert et al.12 reported a lower incidence of early pneumonia in after cardiac surgery patients intubated with a conical polyurethane-cuffed tube. However, diagnosis of pneumonia was based on clinical criteria and lower airway cultures were positive in only a fraction of the study population. In a small cohort of critically ill patients, Bulpa et al.13 used scintigraphy to detect micro-aspiration alongside cuffs and found that a conical polyurethane-cuffed tracheal tube did not perform better than a PVC-cuffed one. In addition, we recently demonstrated that polyurethane-cuffed tracheal tubes were prone to internal condensation causing false manometric cuff pressure readings.14 Finally, a recent study showed that polyurethane and/or conically shaped cuffs were not superior to PVC and/or cylindrical cuffs in preventing tracheal colonisation and VAP.15
Subglottic secretion drainage (SSD) is recommended as part of many VAP prevention protocols. This recommendation is based on a recent meta-analysis showing that SSD resulted in a 50% reduction in the risk of acquiring VAP. Although SSD may shorten the duration of mechanical ventilation and ICU length of stay, it does not improve survival.16 However, other VAP prevention measures were not specified in 11 of the 13 studies analysed. Also, in one17 of the four studies marked as of ‘high methodological quality’, a polyurethane cuff was used in addition to SSD, and this prevents evaluation of their respective roles as a VAP preventive measure. In addition, SSD is not always effective and is not without risk. Rello et al.18 reported SSD failure in up to one-third of patients, whereas Girou et al.19 noted laryngeal oedema necessitating re-intubation in 25% of patients who had received continuous SSD. Data from animal models show convincingly that SSD produces tracheal injury in areas immediately adjacent to the subglottic suction port, and this injury did not recover within up to 96 h following extubation.20 We,21 and others22,23 have shown in vivo that tracheal injury occurs when mucosa is aspirated into a suction port. Recently, a tracheal tube equipped with SSD, that had provided excellent results with regard to VAP prevention,24 had to be withdrawn from the market because of repeated tube kinking which resulted in inadequate ventilation causing severe, including hypoxic injury.25
In conclusion, in terms of preventing leakage, the novel PVCdc tracheal tube outperformed all the PVC and polyurethane-cuffed comparator tracheal tubes. The sealing effect persisted after acute release of PEEP or tracheal tube disconnection. Clinically, the above may translate into a 100% prevention of micro-aspiration of the bacteria-enriched secretions which accumulate above the cuff, hence, substantially reducing the risk of patients developing ventilator-associated infection. This new tracheal tube also counters the intrusion of secretions into the lower airways in any situation or procedure characterised by abrupt loss of PEEP: for example, during an extubation procedure partial deflation of the cuffs combined with a momentary increase in the pressure between the cuffs will move secretions upward toward the oropharynx where they can be easily aspirated.
Some limitations of our study must be acknowledged. First, we used a vertically mounted benchtop rigid trachea model at room temperature which does not match the clinical situation where a tracheal tube is positioned in a more compliant trachea at body temperature in a semi-recumbent position. Second, subglottic secretions and saliva have higher viscosities compared with the liquid dye used in the study, and hence their potential for leaking past the cuff is less. Third, the study results were obtained over only a 60-min observation period, whereas VAP occurs after at least 48 h of mechanical ventilation. Nevertheless, our findings remain relevant because even a small bacterial load passing the cuff will colonise the trachea and bronchial tract. Subsequent bacterial growth and increasing inflammation then may cause tracheobronchitis, an important risk factor for VAP.26 Fourth, tracheal suctioning was not performed in our model but is unlikely to influence leakage as the inter-cuff pressure is kept constant. Finally, in contrast with the comparator tracheal tubes, the long-term in-vivo effects of a double-cuffed tracheal tube have not been investigated. Whether the advantages of the prototype tracheal tube seen in our benchtop study will translate in to clinical advantages will need to be investigated by an appropriate clinical trial.
Acknowledgements relating to this article
Assistance with the study: none.
Financial support and sponsorship: the study received a research grant from the ‘Wetenschappelijk Fonds Willy Gepts’ of the University Hospital, Vrije Universiteit, Brussels.
Conflicts of interest: none.
Presentation: the study was presented in part as an oral communication at the 33rd Congress of the Scandinavian Society of Anesthesiology and Intensive Care Medicine, 10 to 12 June 2015, Reykjavik, Iceland.
1. Kalanuria AA, Ziai W, Mirski M. Ventilator-associated pneumonia in the ICU. Crit Care
2. Kollef MH, Hamilton CW, Ernst FR. Economic impact of ventilator-associated pneumonia in a large matched cohort. Infect Control Hosp Epidemiol
3. Deem S, Treggiari MM. New endotracheal tubes designed to prevent ventilator-associated pneumonia: do they make a difference? Respir Care
4. Pitts R, Fisher D, Sulemanji D, et al. Variables affecting leakage past endotracheal cuffs: a bench study. Intensive Care Med
5. Zanella A, Scaravilli V, Isgrò S, et al. Fluid leakage across tracheal tube cuff, effect of different cuff material, shape, and positive expiratory pressure: a bench-top study. Intensive Care Med
6. Ouanes I, Lyazidi A, Danin PE, et al. Mechanical influences on fluid leakage past the tracheal tube cuff in a benchtop model. Intensive Care Med
7. Dave MH, Koepfer N, Madjdpour C, et al. Tracheal fluid leakage in benchtop trials: comparison of static versus dynamic ventilation model with and without lubrication. J Anesth
8. Pneumatikos IA, Dragoumanis CK, Bouros DE. Ventilator-associated pneumonia or endotracheal tube-associated pneumonia? An approach to the pathogenesis and preventive strategies emphasizing the importance of endotracheal tube. Anesthesiology
9. Kollef MH, Afessa B, Anzueto A, et al. Silver-coated endotracheal tubes and incidence of ventilator-associated pneumonia: the NASCENT randomized trial. JAMA
10. Fernandez JF, Levine SM, Restrepo MI. Technologic advances in endotracheal tubes for prevention of ventilator-associated pneumonia. Chest
11. Madjdpour C, Mauch J, Dave MH, et al. Comparison of air-sealing characteristics of tapered- vs. cylindrical-shaped high-volume, low-pressure tube cuffs. Acta Anaesthesiol Scand
12. Poelaert J, Depuydt P, De Wolf A, et al. Polyurethane cuffed endotracheal tubes to prevent early postoperative pneumonia after cardiac surgery: a pilot study. J Thorac Cardiovasc Surg
13. Bulpa P, Evrard P, Bouhon S, et al. Polyurethane does not protect better than polyvinyl cuffed tracheal tubes from microaspirations. Minerva Anestesiol
14. Spapen H, Moeyersons W, Stiers W, et al. Condensation of humidified air in the inflation line of a polyurethane cuff precludes correct continuous pressure monitoring during mechanical ventilation. J Anesth
15. Philippart F, Gaudry S, Quinquis L, et al. Randomized intubation with polyurethane or conical cuffs to prevent pneumonia in ventilated patients. Am J Respir Crit Care Med
16. Muscedere J, Rewa O, McKechnie K, et al. Subglottic secretion drainage for the prevention of ventilator-associated pneumonia: a systemic review and meta-analysis. Crit Care Med
17. Lorente L, Lecuona M, Jiménez A, et al. Influence of an endotracheal tube with polyurethane cuff and subglottic secretion drainage on pneumonia. Am J Respir Crit Care Med
18. Rello J, Soñora R, Jubert P, et al. Pneumonia in intubated patients: role of respiratory airway care. Am J Respir Crit Care Med
19. Girou E, Buu-Hoi A, Stephan F, et al. Airway colonization in long-term mechanically ventilated patients. Effect of semi-recumbent position and continuous subglottic suctioning. Intensive Care Med
20. Li Bassi G, Luque N, Martí JD, et al. Endotracheal tubes for critically ill patients: an in vivo analysis of associated tracheal injury, mucociliary clearance, and sealing efficacy. Chest
21. Suys E, Nieboer K, Stiers W, et al. Intermittent subglottic secretion drainage may cause tracheal damage in patients with few oropharyngeal secretions. Intensive Crit Care Nurs
22. Chandler Harvey R, Miller P, Lee JA, et al. Potential mucosal injury related to continuous aspiration of subglottic secretion device. Anesthesiology
23. Dragoumanis CK, Vretzakis GI, Papaioannou VE, et al. Investigating the failure to aspirate subglottic secretions with the Evac endotracheal tube. Anesth Analg
24. Damas P, Frippiat F, Ancion A, et al. Prevention of ventilator-associated pneumonia and ventilator-associated conditions: a randomized controlled trial with subglottic secretion suctioning. Crit Care Med
© 2017 European Society of Anaesthesiology
26. Craven DE, Chroneou A, Zias N, et al. Ventilator-associated tracheobronchitis. The impact of targeted antibiotic therapy on patient outcomes. Chest