The Quick Spectral Modulation Detection (QSMD) test provides a quick and clinically implementable spectral resolution estimate for cochlear implant (CI) users. However, the original QSMD software (QSMD(MySound)) has technical and usability limitations that prevent widespread distribution and implementation. In this article, we introduce a new software package EasyQSMD, which is freely available software with the goal of both simplifying and standardizing spectral resolution measurements.
QSMD was measured for 20 CI users using both software packages.
No differences between the two software packages were detected, and based on the 95% confidence interval of the difference between tests, the difference between the tests is expected to be <2% points. The average test duration was under 4 minutes.
EasyQSMD is considered functionally equivalent to QSMD(MySound) providing a clinically feasible and quick estimate of spectral resolution for CI users.
1Department of Otolaryngology, New York University School of Medicine, New York, New York, USA
2Department of Hearing and Speech Sciences, Vanderbilt University Medical Center, Nashville, Tennessee, USA.
Received July 28, 2018; accepted January 12, 2019.
The authors have no conflicts of interest to disclose.
Address for correspondence: David M. Landsberger, Department of Otolaryngology, New York University School of Medicine, 550 1st Avenue, STE NBV 5E5, New York, NY 10016, USA. E-mail: David.Landsberger@nyumc.org
Online date: April 04, 2019