Other factors, which could theoretically influence M(P), UFR, single-pool Kt/V (spKt/V), and cumulative blood flow per procedure (VB), did not differ significantly.
There are two principal messages clearly emerging from this study:
The first one is dissociation of dialysis efficiency in urea removal as assessed by the Kt/V and phosphorus removal assessed by its removed amount. It is merely a confirmation of a know phenomenon caused by more complex phosphorus kinetics11–13 compared with simple first-order kinetics of urea. Recently, it was nicely demonstrated by a study,14 which compared two dialysis schedules with 4 and 8 hours sessions but the same Kt/V (achieved by much lower blood flow in the latter schedule): Although the amount of removed urea was identical in both schedules, the amount of removed phosphorus was by about 40% higher in the schedule with longer session time.
The second and probably clinically more important and novel finding of our study is the very minor role of convective transport in phosphorus dialytic removal. Comparison of the data from HFHD and HDF clearly indicate that convective component adds very little, if anything, to the diffusive removal. Unsatisfactory phosphorus balance in a patient treated by LFHD can be significantly improved by his/her transfer to HFHD. No amelioration can, however, be expected in a patient treated by HFHD with his/her transfer to HDF. This may appear in contradiction with some papers which claimed that phosphorus clearance or its removal may be improved by adding convection as an additional element to standard diffusional transport. However, those papers15–18 compared HDF with LFHD and were thus not able to separate the effect of increased diffusibility of high-flux membranes (seen already in HFHD) from the impact of significant convection (taking place in HDF only). Significant improvement in both phosphorus clearance and its removed amount with switchover from HFHD to HDF has been reported by Zehnder et al. 19 But this study used lower dialysate flow for HFHD (500 ml/min) than in HDF (800 ml/min). Unfortunately, inappropriate and in view of the results of our study false statement of benefits of HDF in phosphorus removal can be found also in the European Best Practice Guidelines (EBPG) document on dialysis strategy.20 Although we can agree with its guideline G1.4 (beneficial impact of longer session time and increased HD frequency in patients with impaired phosphorus control) and G2.1 (recommendation to use high-flux membranes in hyperphosphataemia), evidence of the G2.2 guideline on superior position of HDF over HFHD in phosphorus removal seems rather weak—out of three references one21 compares HDF with acetate-free biofiltration (i.e., also principally an HDF process), one22 compares LFHD (F8 dialyzer, Fresenius Medical Care, Bad Homburg, Germany) with HDF (F B3-2 Filtryzer dialyzer, Toray Medical Company Ltd., Tokyo, Japan) and is primarily focused on evaluation of postdialysis rebound, and one23 compares HD and HDF but its abstract available on the Internet does not specify the HD mode (LFHD or HFHD). On the other side, our finding of decisive role of diffusional transport for phosphorus removal is in good correspondence with theoretical calculation of phosphorus clearances by means of the simulation software CCT (Clearance Calculation Toot, based on Michaels transport equation,24 updated to include also convective transport, Fresenius Medical Care, Figure. 2). Calculation was done for FX 10 dialyzer (for LFHD) and FX 80 (for HFHD and HDF), QB was 300 ml/min for all procedures and QS was 100 ml/min for HDF, and KoA (mass transfer area coefficient) mass transfer area coefficient for phosphorus was used to calculate phosphorus clearance.
There is a steep increase (+29 ml/min, i.e., nearly 14%) in phosphorus clearance with switchover from LFHD with low-flux polysulfone dialyzer FX10 (210 ml/min) to the corresponding high-flux type FX80 with the same surface area (239 ml/min). Further increase in clearance when using FX 80 dialyzer in HDF mode (251 ml/min) is much less pronounced (+12 ml/min, i.e., just 5%). The fact that it is still higher that the increase in removed phosphorus amount in vivo seen in our study (Figure 1) can be explained by specific behavior of Fresenius dialysis machines—the produced substitution flow is added on the top of the dialysate flow (i.e., overall production of the solution is increased in HDF when compared with HD). With Gambro machines that were mostly used in our study, substitution fluid is taken from the fixed amount of produced dialysis solution (i.e., dialysate flow rate is thus reduced in HDF when compared with HD mode).
It is fair to say that the detailed statistical analysis of our data resulted in some surprising and difficult-to-explain findings. It is for instance much lower statistical significance of the difference in M(P) obtained in LFHD versus HFHD when compared with the LFHD versus HDF, despite insignificance of this difference in HDF versus HFHD. The second one is weak correlation between phosphorus removal in individual patients obtained by different dialytic modes. The key may be in dependence of this correlation on the predialysis plasma phosphorus value. There are several factors that may have affected the predialysis value. The first one is variation caused by putting one procedure with a different phosphorus removal efficiency in a sequence of standard sessions (LFHD or HDF performed in a patient who was normally dialyzed on LFHD, or vice versa one LFHD used in a patient otherwise treated by HFHD or HDF). Predialysis phosphatemia can also be significantly influenced by a short-term event such as single excess in dietary phosphorus intake and noncompliance with phosphorus binding agents. Such sudden peak in phosphatemia would logically overestimate phosphorus elimination by dialysis during the next session. Phosphorus removal is much more variable than removal of urea or creatinine, as has been recently shown25 and stressed.26 Also different saturation of inner phosphorus compartments in the different patients may have lead to different releases of phosphorus into plasma during dialysis depending on different dialytic clearances provided by the tested modes. Although existence of such compartments is generally accepted, their nature is far from being clear. Some consider bones27,28 as a possible store compartment able to release phosphorus during its massive dialytic removal from plasma, others suggest glycophosphates in body fluids13 or leave this question entirely open.11 The recently introduced “mobilization clearance” of phosphorus from inner compartments29–31 may once prove useful to asses those differences. All the previously mentioned possibilities (regardless of which came into effect in our study), however, do not change the principal finding that predialysis plasma phosphorus level seems the only confounding factor worth to be considered in any statistical evaluation of phosphorus removal.
In fact, consideration of patient-related confounding factors (namely predialysis phosphatemia and ultrafiltration) and practical inability of eliminate them in a longer time period led us to use this single procedure—in vivo—single cohort study design instead of a cross-over one done for a longer time.
It would be difficult both to fix strictly procedure parameters (as we did for those three assessed sessions) and to eliminate the previously mentioned patient-related confounding factors for several weeks required for a cross-over study.
Rather a by-product of our study is evidence that the Gutzwiller method can be easily applied to estimate phosphorus removal by dialysis, be it HD or HDF. Data on phosphorus dietary intake as the second factor determining overall phosphorus balance can be obtained either by analysis of phosphorus content in the ingested diet or roughly estimated from the protein catabolic rate.7 With the knowledge of those two factors (intake and dialytic removal), prescription of necessary phosphorus binders (accounting for their binding capacity—see for instance work of Daugirdas et al.32) could possibly be more objectively controlled. With regard to complexity of such a procedure, its use can, however, be reasonably expected only in patients with highly difficult or puzzling problems of overall phosphorus balance.
Similar results come from published papers in the area of phosphate metabolism; however, the number of published papers with this topic is markedly lower. None of these papers investigated whether the enhanced phosphorus elimination in HDF treatment is caused by increased phosphorus clearance of high permeable membrane (e.g., increased diffusion) or by high-volume ultrafiltration (partly replaced by substitution fluid, e.g., increased convection).
Based on comparison of phosphorus removal between LFHD versus HFHD and HFHD versus HDF in our study, it can be concluded that
- - It is the increased diffusibility of high-flux dialyzers which significantly augments phosphorus dialytic removal.
- - Addition of convection by using the high-flux dialyzers in the HDF mode, however, does not lead to any significant further increase compared with HFHD.
Practical applicability and accuracy of Gutzwiller’s and Gotch’s formulas for estimation of phosphorus removal M(P) both in HD and in HDF were investigated in our center in a small single center study in 2007, and the results were presented at the 34th ESAO (European Society for Artificial Organs) Congress.9 Those results were, however, subsequently not published in extenso in any article. We have, therefore, included brief information on that study to this article as an Appendix. Aim of the study9 was not comparison of elimination efficiency of different dialytic modes as in this article but verification of applicability of Gutzwiller’s and Gotch’s methods against PDC as a reference both for HD and for HDF. The detailed results for Gutzwiller’s method are shown in Figure 3.
It displays correlation between the amount of phosphorus removed during a single session in a group of 16 patients on HD (six of them on LFHD and 10 on HFHD) and 10 patients on HDF.
Next step was a comparison of combined data from LFHD, HFHD, and HDF, obtained by PDC, with results calculated by Gotch and Gutzwiller formula. Comparison of the correlations revealed slightly weaker value for the Gotch’s method when compared with that of Gutzwiller:
This together with easier and less laborious measurement procedure made us to choose the Gutzwiller’s approach for this study. The fact that it appears to underestimate M(P) against PDC did not matter in this study comparing efficacy of different blood-cleansing modes because it was applied uniformly in all of them.
1. Levin A, Bakris GL, Molitch M, et al. Prevalence of abnormal serum vitamin D, PTH, calcium, and phosphorus in patients with chronic kidney disease: results of the study to evaluate early kidney disease. Kidney Int. 2007;71:31–38
2. Tentori F, Blayney MJ, Albert JM, et al. Mortality risk for dialysis patients with different levels of serum calcium, phosphorus, and PTH: the Dialysis Outcomes and Practice Patterns Study (DOPPS). Am J Kidney Dis. 2008;52:519–530
3. Noordzij M, Korevaar JC, Boeschoten EW, Dekker FW, Bos WJ, Krediet RTNetherlands Cooperative Study on the Adequacy of Dialysis (NECOSAD) Study Group. . The Kidney Disease Outcomes Quality Initiative (K/DOQI) Guideline for bone metabolism and disease in CKD: association with mortality in dialysis patients. Am J Kidney Dis. 2005;46:925–932
4. Block GA, Hulbert-Shearon TE, Levin NW, Port FK. Association of serum phosphorus and calcium x phosphate product with mortality risk in chronic hemodialysis patients: a national study. Am J Kidney Dis. 1998;31:607–617
5. Kidney Disease: Improving Global Outcomes (KDIGO) CKD-MBD Work Group. . KDIGO clinical practice guideline for the diagnosis, evaluation, prevention, and treatment of chronic kidney disease-mineral and bone disorder (CKD-MBD). Kidney Int Suppl. 2009:S1–S130
6. National Kidney Foundation. . K/DOQI Clinical Practice Guidelines for Bone Metabolism and Disease in Chronic Kidney Disease. Am J Kidney Dis. 2003;42:S1–S202
7. Gotch FA, Panlilio F, Sergeyeva O, et al. A kinetic model of inorganic phosphorus mass balance in hemodialysis therapy. Blood Purif. 2003;21:51–57
8. Gutzwiller JP, Schneditz D, Huber AR, Schindler C, Gutzwiller F, Zehnder CE. Estimating phosphate removal in haemodialysis: an additional tool to quantify dialysis dose. Nephrol Dial Transplant. 2002;17:1037–1044
9. Lopot F, Turzová M, Švára F, Polakovič V, Nejedlý B. Quantification of phosphate removal during low- and high-flux haemodialysis (LFHD, HFHD) and haemodiafiltration (HDF). Int J Artif Organs. 2007;30:699
10. Daugirdas JT. Second generation logarithmic estimates of single-pool variable volume Kt/V: an analysis of error. J Am Soc Nephrol. 1993;4:1205–1213
11. Spalding EM, Chamney PW, Farrington K. Phosphate kinetics during hemodialysis: evidence for biphasic regulation. Kidney Int. 2002;61:655–667
12. Sugisaki H, Onohara M, Kunitomo T. Dynamic behavior of plasma phosphate in chronic dialysis patients. Trans Am Soc Artif Intern Organs. 1982;28:302–307
13. Maasrani M, Jaffrin MY, Fischbach M, Boudailliez B. Urea, creatinine and phosphate kinetic modeling during dialysis: application to pediatric hemodialysis. Int J Artif Organs. 1995;18:122–129
14. Sampaio MS, Ruzany F, Dorigo DM, Suassuna JH. Phosphate mass removal during hemodialysis: a comparison between eKT/V-matched conventional and extended dialysis. Am J Nephrol. 2012;36:121–126
15. Penne EL, van der Weerd NC, van den Dorpel MA, et al.CONTRAST Investigators. Short-term effects of online hemodiafiltration on phosphate control: a result from the randomized controlled Convective Transport Study (CONTRAST). Am J Kidney Dis. 2010;55:77–87
16. Movilli E, Camerini C, Gaggia P, et al. Effect of post-dilutional on-line haemodiafiltration on serum calcium, phosphate and parathyroid hormone concentrations in uraemic patients. Nephrol Dial Transplant. 2011;26:4032–4037
17. Davenport A, Gardner C, Delaney MPan Thames Renal Audit Group. . The effect of dialysis modality on phosphate control: haemodialysis compared to haemodiafiltration. The Pan Thames Renal Audit. Nephrol Dial Transplant. 2010;25:897–901
18. Vaslaki L, Major L, Berta K, et al. Decrease of serum phosphate with on-line hemodiafiltration. Nephrol Dial Transplant. 2003;18(Suppl. 4):194
19. Zehnder C, Gutzwiller JP, Renggli K. Hemodiafiltration–a new treatment option for hyperphosphatemia in hemodialysis patients. Clin Nephrol. 1999;52:152–159
20. . European Best Practice Guidelines on Haemodialysis: EBPG guideline on dialysis strategies. Guideline 2. Flux and convection. Nephrol Dial Transplant. 2007;22(Suppl 2):5–21
21. Ding F, Ahrenholz P, Winkler RE, et al. Online hemodiafiltration versus acetate-free biofiltration: a prospective crossover study. Artif Organs. 2002;26:169–180
22. Minutolo R, Bellizzi V, Cioffi M, et al. Postdialytic rebound of serum phosphorus: pathogenetic and clinical insights. J Am Soc Nephrol. 2002;13:1046–1054
23. Tuccillo S, Bellizzi V, Catapano F, et al. [Acute and chronic effects of standard hemodialysis and soft hemodiafiltration on interdialytic serum phosphate levels]. G Ital Nefrol. 2002;19:439–445
24. Michaels AS. Operating parameters and performance criteria for hemodialyzers and other membrane-separation devices. Trans Am Soc Artif Intern Organs. 1966;12:387–392
25. Debowska M, Wojcik-Zaluska A, Ksiazek A, Zaluska W, Waniewski J. Phosphate, urea and creatinine clearances: haemodialysis adequacy assessed by weekly monitoring. Nephrol Dial Transplant. 2015;30:129–136
26. Basile C, Schneditz D. Haemodialysis adequacy monitoring for phosphate: an old problem with new solutions? Nephrol Dial Transplant. 2015;30:9–11
27. Sugisaki H, Onohara M, Kunitomo T. Phosphate in dialysis patients. Trans Am Soc Artif Intern Organs. 1983;29:38–43
28. Aebisher P, Schorderet D, Noetzli P, Fellay G. Phosphate kinetics during hemodialysis. Progress in Artif Organs – 1985. 1986 Cleveland ISAO Press:128–131
29. Agar BU, Troidle L, Finkelstein FO, Kohn OF, Akonur A, Leypoldt JK. Patient-specific phosphorus mobilization clearance during nocturnal and short daily hemodialysis. Hemodial Int. 2012;16:491–496
30. Agar BU, Akonur A, Cheung AK, Leypoldt JK. A simple method to estimate phosphorus mobilization in hemodialysis using only predialytic and postdialytic blood samples. HemodialInt. 2011;15:S8–S13
31. Leypoldt JK, Agar BU, Akonur A, Culleton BF. Phosphorus kinetics during hemodiafiltration: analysis using a pseudo-one-compartment model. Blood Purif. 2013;35(Suppl 1):59–63
32. Daugirdas JT, Finn WF, Emmett M, Chertow GMFrequent Hemodialysis Network Trial Group. . The phosphate binder equivalent dose. Semin Dial. 2011;24:41–49
Keywords:Copyright © 2016 by the American Society for Artificial Internal Organs
low-flux hemodialysis; high-flux hemodialysis; hemodiafiltration; Gutzwiller formula