Sedentary behaviors are defined as seated or reclining postures that require low levels of energy expenditure (e.g., <1.5 METs) (24), and they comprise 55% to 70% of waking hours (22). Strong epidemiological evidence suggests that sedentary behaviors, independent of physical activity, are associated with a host of poor health outcomes, including increased risk of obesity, type 2 diabetes, cardiovascular disease, and premature mortality. Although these relationships have been predominantly established using self-reported surrogate measures of sedentary time (e.g., TV viewing time), studies using objective measures also support these findings (2,3,8,14–17,20). Using accelerometers to obtain global measures of total sedentary time, results from cross-sectional studies indicate sedentary behaviors are associated with increased risk for abnormal glucose and lipid metabolism, obesity, and several chronic diseases (8,14,17,20).
Several animal and human studies indicate that in addition to total sedentary time, how one breaks up sedentary time may influence the muscular, cardiovascular, and metabolic responses to sedentary behavior (1,12,15,29). Advances in accelerometer-based activity monitors have allowed researchers to investigate how breaks from sedentary time affect health in free-living humans. Using the ActiGraph GT1M accelerometer (Actigraph, Pensacola, FL), Healy et al. (15) reported that frequent breaks from sedentary time (i.e., sit-to-stand transitions) are beneficially associated with adiposity, triglycerides, and 2-h plasma glucose, independent of total sedentary time and physical activity. Data from the National Health and Examination Survey indicate more breaks from sitting are also associated with a smaller waist circumference and lower C-reactive protein levels (16). The observation that taking frequent breaks from sedentary behaviors attenuates the negative physiologic response is further supported by work from Chastin et al. (2) and Chastin and Granat (3). Using the activPAL monitor (PAL Technologies Ltd., Glasgow, Scotland) to assess sedentary time, there was no difference in total sedentary time between healthy individuals and individuals with chronic health conditions. However, there were significant differences in how sedentary time occurred between groups (2,3).
These data provide evidence that “breaks from sitting” may be an appropriate target for behavioral interventions. Encouraging individuals to “break up” and accumulate shorter bouts of sedentary time may be a feasible strategy for intervention recommendations. However, before establishing recommendations and implementing interventions, validity of wearable devices to detect breaks during free-living behavior must be determined. In laboratory studies, the activPAL was validated to estimate time in different postures (11), step count (21,26), static and dynamic behaviors (10), and sit-to-stand transitions (11). In free-living conditions, the activPAL produced highly accurate and precise estimate of total sedentary time (18). Given these data, it seems reasonable to suggest that the activPAL will accurately identify breaks from sedentary time in free-living settings; however, this has yet to be determined. Conversely, evidence suggests that the ActiGraph will not be able to identify breaks from sedentary time given its biased and imprecise estimates of total sedentary time (18). The ActiGraph is traditionally used to quantify time spent in different intensities of activity by summing time above or below specified count thresholds. This method, known as the cut-point method (9), works reasonably well for identifying moderate-to-vigorous physical activity (MVPA) but is less accurate for distinguishing sedentary and light intensity activities (18,19). In the context of sedentary behavior measurement, this method is inherently limited in that it estimates intensities, not posture. However, the validity of the ActiGraph in identifying breaks from sedentary time has yet to be tested. This is of particular importance given the ActiGraph was the tool used in all studies providing evidence that breaks from sedentary time may be beneficial for health (15,16). Thus, an empirical evaluation and comparison of both tools will further inform research aiming to measure and interpret sedentary behavior and its associated health outcomes.
As the first step in this validation, a suitable metric to describe breaks from sedentary time in free-living environments must be established. In cohort or etiologic studies where the goal is to relate a characteristic of sedentary behavior (e.g., total amount of sedentary time, total number of breaks from sedentary time) to a specific health outcome, it is appropriate to measure and statistically evaluate the independent, additive, and interactive effects of different features of sedentary behavior. This is easily done in statistical models. Conversely, when measuring behavioral changes after an intervention or establishing recommendations regarding sedentary behavior, it may be more appropriate to use a composite measure of sedentary behavior that collectively assesses changes in both total sedentary time and breaks from sedentary time. For example, in intervention studies where sedentary time decreases, the opportunity to accumulate breaks also decreases. Thus, change in the absolute number of breaks may not be sensitive to important behavioral changes consequent to an intervention. In contrast, if breaks are expressed as a rate per sedentary hour, the overall effects of the intervention may be more readily detected. In addition, a composite measure of sedentary behavior, such as breaks per sedentary hour, is ecologically relevant and may be easily translated for public health recommendations.
This study had three specific aims. Aim 1 was to express breaks from sedentary time as a metric specifically relevant to free-living sedentary behavior and to compare this metric with the absolute number of breaks. The metric, called “break rate,” is a measure of the rate of breaks per sedentary hour (brk·sed-h−1). Aim 2 was to validate the performance of the ActiGraph GT3X and activPAL to estimate absolute number of breaks and break rate. Aim 3 was to determine whether the ActiGraph GT3X and activPAL are sensitive enough to detect the effects of an intervention to reduce and break-up sedentary time.
Eligibility and recruitment.
Thirteen participants were recruited from the University of Massachusetts (Amherst, MA) and local communities. Eligible participants were between the ages of 20–60 yr and satisfied the physical activity guideline of accumulating at least 150 min·wk−1 of moderate intensity activity (4). Participants were healthy and free from any musculoskeletal problems that would compromise their ability to change how they accumulate sedentary time.
Participants reported to the University of Massachusetts and signed an informed consent document approved by the University of Massachusetts Institutional Review Board. Following the consenting process, height (to the nearest 0.1 cm) and weight (to the nearest 0.1 kg) were measured using a floor scale/stadiometer (Detecto, Webb City, MO). To determine initial eligibility, participants completed a health history questionnaire and a physical activity status questionnaire. To verify self-reported activity level, participants wore the ActiGraph GT3X for seven consecutive days. These data were analyzed using the Freedson cut points to determine time spent in MVPA per week (9). If MVPA was less than 150 min, participants were excluded from the study.
Free-living behavior was monitored for approximately 10 consecutive hours on two separate days. For the first monitoring day, participants were asked to perform their normal daily activity, including their normal exercise routine and normal activities of daily living (baseline condition). For the second monitoring day, they were asked to decrease their total sedentary time and to break up sedentary behaviors as much as possible (treatment condition). Participants were not provided a specific prescription to reduce and break up their sedentary time but were provided suggestions and examples of how to decrease sedentary time within their natural free-living environment. For example, if they normally spent the majority of their working day seated at a desk, they were encouraged to make standing workstations so they could easily alternate between sitting and standing. Other suggestions included standing during television commercials, standing to answer the phone, and standing while eating. The baseline and treatment conditions were performed at least 24 h apart, and no more than 2-wk separated the two conditions.
Criterion measure: direct observation.
In each condition, direct observation served as the criterion measure to assess total sedentary time and breaks from sedentary time. In the morning, a trained observer met the participants in their natural environment (e.g., place of work, school or home). Participants were then observed for approximately 10 consecutive hours. A hand-held personal digital assistant (PDA) programmed for focal sampling and duration coding was used to record participant behavior. Every time body position changed (e.g., went from sitting to standing), the observer recorded this in the PDA. Each entry was time stamped, and the length of each behavior bout was automatically recorded in the PDA. The start and stop of each behavior recorded by the direct observer was exported from the PDA using custom software (Observer 9.0; Noldus Information Technology, The Netherlands). During the 10-h observation, participants were allowed to have “private time” when needed. Reasons for “private time” included behaviors such as using the restroom and changing clothes. During these activities, the observer coded “private” on the PDA.
The DO PDA hardware (Pocket Observer, Noldus Information Technology) and software (Observer 9.0, Noldus Information Technology) used to conduct the direct observation are commercially available tools. The software was programmed to capture specific behaviors of interest (e.g., sitting and standing). Observers worked in 2–4 h shifts, and a total of three different observers completed all of the observation sessions for each participant. Observers completed extensive verbal, written, and video training and testing before observing participants in a free-living environment. The training material focused on a specific protocol to avoid disrupting free-living behavior and to accurately record sedentary time and breaks from sedentary time. Upon completion of training, the ability of each observer to identify activity type (e.g., sit, stand, and walk) and intensity (e.g., 3 METs) was tested using an approximately 15-min video of free-living behavior. The video was first coded by a group of experienced observers, and study observer responses (activity type and MET value) were compared with the experienced observers’ responses using a Cohen’s κ coefficient. To be considered “in agreement” for the study, observers needed to correctly identify both the type and intensity of the activity. There was a very high level of agreement between the study observers’ responses and the experienced observers’ responses (mean κ = 0.92).
Activity monitors: data synchronization.
A start/stop record was generated to record the start and stop times for each observation period using the custom Noldus software. The start/stop record was then used to process each of the activity monitor files using a custom R program (27). The monitor data corresponding directly with the start and stop times of direct observation were then extracted from the monitor file. The monitors were initialized on the same file as the computer; therefore, the activity monitor files were matched second by second with the direct observation time stamp. Behaviors coded as “private” by the observer along with the corresponding activity monitor data were eliminated from the analyses.
Each participant wore two GT3X monitors programmed to collect data from the vertical axis in 1-s epochs. One was initialized using the low-frequency extension (AG-LFE), and the other was initialized with the normal filter (AG-Norm). For both the AG-LFE and AG-Norm, two count cut points to estimate sedentary time were examined: 100 counts per minute (AG-LFE-100 and AG-Norm-100, respectively) and 150 counts per minute (AG-LFE-150 and AG-Norm-150, respectively). The 100 counts per minute cut point is the predominant cut point used by researchers (5,13,15,18,23). However, Kozey-Keadle et al. (18) recently reported that 150 counts per minute may be a more appropriate cut point to identify sedentary time from the ActiGraph monitor. The AG-LFE and AG-Norm were worn on a belt with the AG-LFE positioned directly on the anterior suprailiac spine and the AG-Norm positioned immediately distal to the AG-LFE. A custom R program was used to generate total sedentary time, breaks, and break rate from the AG monitor.
Participants wore the activPALs on their right thighs. It was attached using a nonallergenic adhesive pad and positioned on the midline of the thigh, one-third of the way between the hip and knee. The time-stamped “event” data file from the activPAL software (version 5.8.5) was used to identify the duration of each sitting/lying, standing, and stepping bout. The event file was converted to second-by-second data using an R program to synchronize data with the DO file.
For aims 1 and 3, direct observation data and valid monitor data from all three monitors were required for a participant to be included in the analyses. For aim 2, valid data from all monitors were not required. For example, if a participant had valid activPAL and AG-LFE data, but not AG-Norm, only the AG-Norm analyses were eliminated.
Operational definitions for total sedentary time and the absolute number of breaks for the criterion (direct observation) and each activity monitor were as follows:
- Total sedentary time (h)—the sum of the duration of behavior that was coded as sitting or lying.
- Break from sedentary time—any instance where a sitting or lying behavior was followed by a nonsitting or nonlying behavior.
activPAL—Output from the activPAL was downloaded using the “event” data in the activPAL software. These data provide information about the start time and duration of each sitting/lying, standing, and stepping bout.
- Total sedentary time (h)—determined by summing the duration of all sitting/lying bouts.
- Break from sedentary time—any instance where a sitting/lying bout was followed by a standing or stepping bout.
AG-LFE and AG-Norm
- Total sedentary time (h)—determined by summing the minutes where counts per minute were less than the cut point of interest (e.g., counts per minute <100).
- Break from sedentary time—any instance where a minute identified as sedentary (e.g., counts per minute <100) was followed by a minute identified as not sedentary (e.g., counts per minute ≥100).
Data from each measurement method were used to calculate break rate. Break rate was calculated by dividing the absolute number of breaks by total hours spent sedentary (brk·sed-h−1).
All statistical analyses were performed using the R statistical computing environment (27). To address aim 1, we computed and compared the break rate for the criterion and each prediction method. We used a repeated-measures linear mixed model and likelihood ratio tests to evaluate the sensitivity of these metrics to detect the possible effects of the intervention on measures of sedentary behavior. The repeated-measures model was used to account for possible intraindividual correlation. The likelihood ratio test was examined if the addition of condition (baseline/treatment) as an independent variable resulted in a significantly better fit.
To address aim 2, we used two statistical tools: percentage bias and precision. The percentage bias or mean difference between predicted and criterion estimates (Σ((predicted break rate − criterion break rate)/criterion break rate)/N) is a measure of accuracy and gives information about the systematic error of the prediction. We expressed bias as a percentage of the criterion to compare estimates between conditions and allow the size of the error to be proportional to the quantity that is measured. In this study, a negative bias indicated underestimation by the prediction method; a positive bias indicated overestimation by the prediction method. Precision is the inverse of variance and provides information about the size of the random error of the prediction. To evaluate precision, we determined the 95% confidence interval (CI) for the differences between the predictions and the criterion (bias). Use of the 95% CI allowed us to do two things. The width of the CI is proportional to the SE of the differences; a small CI width indicated high precision and a large CI width indicated low precision. In addition, if the upper and lower CIs of the bias spanned zero, then the predicted estimate was not significantly different from the criterion at α = 0.05. We used the 95% CI and SE instead of the SD because we were interested in how these metrics performed in a population, not for a specific individual. We note that a CI bias that does not include zero does not necessarily indicate that a measure is unbiased; instead, one may not have the power to detect a bias of a certain size.
To address aim 3, we used a repeated-measures linear mixed model with likelihood ratio testing. We fit separate models for the criterion, the activPAL, and each AG-LFE and AG-Norm cut point.
Thirteen participants (four males and nine females) completed the study. Participants’ mean ± SD age was 24.8 ± 5.2 yr and mean ± SD body mass index was 23.8 ± 1.9 kg·m−2. Of the 13 participants enrolled in the study, one participant’s AG-LFE and AG-Norm monitors did not record data. A second participant’s AG-Norm did not record data. These observations were eliminated from the analyses for aims 1 and 3 (n = 11). For aim 2, this resulted in a total of 234.8 h of direct observation with corresponding AG-LFE data (mean hours per observation = 9.4 (n = 12)), 234.2 h of direct observation with corresponding AG-Norm data (mean hours per observation = 9.8 (n = 11)), and 251.5 h of direct observation with corresponding activPAL data (mean hours per observation = 9.7 (n = 13)).
For aim 1, we expressed breaks from sedentary time as a rate of breaks per sedentary hour and compared this metric to the absolute number of breaks. In Table 1, we report the mean (95% CI) total sedentary time, absolute number of breaks, and break rate identified by direct observation. In the baseline condition, participants spent 6.3 h (5.7–7.0 h) in sedentary behavior and took 30.9 (22.2–39.6) breaks from sedentary time. This resulted in a break rate of 5.1 brk·sed-h−1 (3.3–6.8 brk·sed-h−1). Participants significantly decreased total sedentary time during the treatment condition to 4.5 h (3.6–5.4 h). However, there was no change in the absolute number of breaks (27.8 (20.5–35.1)) (P > 0.05), resulting in a significant increase in break rate to 7.3 brk·sed-h−1 (4.7–9.8 brk·sed-h−1) (P < 0.05).
For aim 2, we examined the accuracy and precision of the ActiGraph GT3X and activPAL for determining the absolute number of breaks and break rate. In Table 2, the percentage bias (95% CI) for each method’s estimate of the absolute number of breaks and break rate is reported. The activPAL accurately and precisely estimated the absolute number of breaks in the baseline (bias, 0.3% (−7.0 to 7.7)) and treatment conditions (bias, 10.9% (−6.8 to 28.6)), whereas all AG-LFE and AG-Norm cut points significantly overestimated breaks in both conditions. In general, the accuracy and precision of the AG-LFE and AG-Norm improved as the sedentary cut point increased. In addition, for a given cut point, AG-Norm was always more accurate and precise than AG-LFE. Both AG-LFE and AG-Norm overestimated the absolute number of breaks to a larger degree in the treatment condition compared with the baseline condition.
The activPAL accurately estimated the break rate in the baseline and treatment conditions, whereas no AG-LFE or AG-Norm cut point accurately estimated the break rate in either condition. The activPAL bias and CI widths were the smallest in both conditions (baseline condition bias, −1.0% (−9.1 to 7.0); treatment condition bias, 10.9% (−7.2 to 29.1)).
In both conditions, the AG-LFE and AG-Norm tended to overestimate break rate, and accuracy (bias) and precision (CI width) improved as the sedentary cut point increased. For each cut point, the AG-Norm had a smaller bias and narrower CI width compared with the AG-LFE. For each cut point, both the AG-LFE and AG-Norm were more accurate and precise in the treatment condition compared with the baseline.
Figure 1 illustrates the percentage bias for each estimate of the absolute number of breaks (top panel) and break rate (bottom panel). The horizontal axis represents the truth relative to direct observation; thus, the closer the bias falls to the horizontal axis, the more accurate the estimate. The error bars are 95% CI of the bias and illustrate the precision of the estimate. If the error bars cross the horizontal axis, the estimate from the monitor is not significantly different than the criterion DO measure. It is important to note that CIs are dependent on sample size. Although for aim 2, our sample sizes were different for each monitor (activPAL = 13, AG-LFE = 12 and AG-Norm = 11), these differences did not influence the relative precision of each tool, nor any conclusions about the significance of the estimates (data not shown). This was determined by performing the analyses on identical sample sizes (N = 11) (e.g., if AG-Norm data were missing for an observation, AG-LFE and activPAL data were also eliminated). Because our conclusions did not change when removed, we chose to keep these data in our analyses.
In Table 2, we also report the correlation between estimated metrics and direct observation. For all monitors, there were significant correlations between direct observation and estimated sedentary time (except AG-LFE-100), the absolute number of breaks, and break rate (range, 0.56–0.99). For each metric, correlations were the highest for estimates from the activPAL, and correlations for AG-Norm estimates tended to be higher than those for AG-LFE estimates.
For aim 3, we evaluated the ActiGraph GT3X and activPAL’s sensitivity to detect the effects of an intervention to reduce and break up sedentary time. Because the absolute number of breaks did not change between conditions, we performed the sensitivity analysis only using break rate as the response. Figure 2 illustrates the sensitivity of each method to detect the effects of the intervention on break rate. Agreeing with the direct observation results, the activPAL detected the effects of the intervention on break rate (P = 0.001). Results from AG-LFE and AG-Norm were indeterminate (range, P = 0.091 (AG-Norm-150) to P = 0.136 (AG-Norm-100)).
The epidemiological evidence linking sedentary time to poor health continues to expand (28). The use of wearable monitors to objectively measure behavior for days to weeks at a time has allowed researchers to investigate the specific role of breaks from sedentary time. The current study contributed to this literature in three ways. First, we demonstrated the use of a measure of sedentary behavior, the “break rate.” Even with a modest sample size, this study demonstrated that break rate is sensitive to an intervention to reduce and break up sedentary time. An existing measure of sedentary behavior, the absolute number of breaks was not sensitive to the intervention in this study. Second, we found that the activPAL estimates of the absolute number of breaks and break rate agreed with direct observation to a large extent. The activPAL estimates were also relatively precise. This suggests the differences between the activPAL estimates, and direct observation may not be practically different for many applications. Third, we found a statistically significant difference between the ActiGraph GT3X estimates of the absolute number of breaks and break rate compared with direct observation. This bias was statistically significant despite our modest sample size and the relatively low precision of the ActiGraph measures of those statistics. Whether the bias is practically significant (or whether the correlation between the ActiGraph estimates and direct observation is large enough) will depend on the application. In addition, in our study sample, the ActiGraph GT3X was not sensitive enough to detect the intervention.
Table 1 provides a summary of how participants changed behavior between the baseline and treatment conditions according to the criterion DO. Interestingly, when individuals were encouraged to reduce and break up their sedentary time, total sedentary time decreased, but the absolute number of breaks did not change significantly (baseline, 30.9 (22.2–39.6); treatment, 27.8 (20.5–35.1) (mean (95% CI)). It is important to note that when less time is spent being sedentary, there are fewer “opportunities” to take breaks and vice versa. This has implications for how we interpret change within individuals and differences between groups in intervention and observational studies, respectively. It is yet to be determined how this relationship affects health, but these results illustrate the advantages of expressing breaks as a rate.
The descriptive data reported in Table 1 are important because on the basis of the criterion DO measure, components of sedentary behavior clearly changed, but this change was not detected when the absolute number of breaks was used to evaluate sedentary behavior. To truly understand the relationship between behavior and disease, it may be desirable to accurately quantify as many of these behaviors as possible. However, we are unaware of an intervention or observational study that has obtained this level of measurement detail.
Metrics not sensitive to total sedentary time are difficult to interpret and translate into meaningful information. Healy et al. (15) were the first to introduce the idea that how sedentary time is accumulated may influence health, reporting a beneficial association between the absolute number of breaks and metabolic risk. Although appropriate to establish the independent effect of breaks from sedentary time, it is difficult to relate these data to real-world applications. Notably, Healy et al. (15) recognize the challenges in expressing breaks from sedentary time as an absolute number and suggest that future research examine the use of expressing breaks as a rate of sedentary time. However, using a composite measure such as break rate also has limitations. Change in break rate will indicate sedentary behavior has changed, but this metric will provide no indication if the change was in total amount of sedentary time, how sedentary time is broken up, or both. When we measure total sedentary time and the number of breaks independently, we can use statistical adjustment to evaluate the independent effects. Break rate cannot be statistically adjusted for because this would result in variables being entered in the model twice.
The results from this study further support the use of the activPAL to measure and characterize different components of sedentary behavior. The activPAL was accurate and precise for estimating the absolute number of breaks and break rate in both the baseline and treatment conditions. It was also sensitive to the effects of an intervention on the break rate. Previous studies indicate the activPAL is a valid tool to differentiate posture and estimate steps in both laboratory and free-living settings (10,11,18,21,26) and to estimate sit-to-stand and stand-to-sit transitions in a laboratory (11). Our results expand on this body of literature by providing free-living evidence that the activPAL is a valid tool to identify the number of transitions between sitting and standing, and it can accurately express this number as a rate per sedentary hour.
Unlike the activPAL, no AG-LFE or AG-Norm cut point accurately identified breaks (bias = 77.8% (AG-Norm-150) to 133.3% (AG-LFE-100)), nor break rate (bias = 30.4% (AG-Norm-150) to 102.3% (AG-LFE-100)) (Fig. 1 and Table 2). Moreover, AG-Norm-100, the cut point previously used by Healy et al. (13,15) to identify breaks from sedentary time, overestimated the absolute number of breaks 98.6% at baseline and 112.1% during the treatment condition.
In addition to the absolute number of breaks and break rate, Table 2 also reports the percentage bias of each monitor estimate of total sedentary time. Because break rate is a ratio of breaks per sedentary hour, we present these data to interpret the results for break rate. Three observations should be noted regarding the inaccuracies of the AG-LFE and AG-Norm cut points that will help to interpret break rate results: 1) as the sedentary cut point increased, overestimation of total sedentary time increased; 2) as the sedentary cut point increased, overestimation of the absolute number of breaks decreased; and 3) overestimates of both total sedentary time and the absolute number of breaks are larger in the treatment condition. These observations are consistent with previous results (18). Specifically, as the cut point increases, there is more “opportunity” for a given minute to be classified as sedentary; simultaneously, it is more “difficult” to achieve a break when the threshold increases. A detailed interpretation of the validity of the activPAL and ActiGraph in estimating total sedentary time is presented by Kozey-Keadle et al. (18).
The activPAL accurately and precisely estimated all metrics of sedentary behavior in both conditions, whereas both the AG-LFE and AG-Norm were considerably less accurate at estimating total sedentary time and the absolute number of breaks in the treatment condition. The mean percentage bias for all AG-LFE and AG-Norm estimates of total sedentary time at baseline was 11.9%, compared with 44.0% during the treatment condition. The mean percentage bias for all AG-LFE and AG-Norm estimates of the absolute number of breaks at baseline was 94.6%, compared with 113.5% during the treatment condition. These results indicate that the accuracy of AG-LFE and AG-Norm may be influenced by wearer behavior. ActiGraph output is similar for standing still and sedentary activities, and thus, it is possible the ActiGraph also measures “breaks from standing.” As a result, the observed differences are likely explained by increases in standing time during the treatment condition (Table 1), which subsequently increased the likelihood the ActiGraph would misclassify standing time as sedentary time. These data are important because if the ActiGraph consistently overestimates breaks in individuals who sit less, then the improved cardiometabolic profile associated with increased breaks (13,15) may actually be due to individuals sitting less and not taking more breaks from sitting.
In addition, it is reasonable to assume that the more active a group is, the more likely it is that the range of activity types performed will increase. It is likely that the precision of most accelerometer-based measurement tools will decrease as participants increase the range of activity types performed. This is supported by many laboratory-based calibrations where measurement errors are influenced by activity type (6,7,19,25). Although the current study was not specifically designed to address the question of varying activity type, our results suggest this will minimally affect the accuracy and precision of the activPAL on a group level, but it may influence ActiGraph estimates. More research is needed to fully understand how specific activity types influence the accuracy and precision of the activPAL and ActiGraph.
Sensitivity of the activPAL and ActiGraph GT3X to an intervention to reduce and break up sedentary time.
The precision of wearable monitors has implications for how well it can detect change between conditions. In both conditions, the activPAL produced the most precise estimates of break rate (Fig. 1). Not surprisingly, it also was the only predictor that detected the effects of the intervention in this study (Fig. 2). These results agreed with direct observation. The bias of a prediction method does not impact its ability to detection of change. In a repeated-measures analysis, the biases cancel each other when the difference is within the individual. For example, even if a tool is inaccurate (biased), it can still be precise if the bias for each individual is similar (e.g., the tool overestimates by 30% for each participant). Practically, this means that when used in an intervention, the activPAL will be capable of detecting a true increase or decrease in break rate. Conversely, the AG-LFE and AG-Norm did not detect the effects of the intervention on break rate.
This is the first study to validate the performance of activity monitors to measure and detect changes in breaks from sedentary time. An important strength of this study is that it was performed under free-living conditions, and sedentary time and breaks from sedentary time were measured using direct observation as the criterion measure. Participants were observed for approximately 10 consecutive hours per observation in their natural environment. By using a custom-designed direct observation system (Noldus Information Technology), we acquired 15,089 min of DO data synchronized with recordings from multiple activity monitors. Because these data were collected in the participant’s natural environment over an extended period, we were also able to capture sedentary behavior in a wide range of settings and contexts.
We also assessed performance under two free-living conditions representative of behavior before and after a sedentary behavior intervention. This is important for two reasons. First, Kozey-Keadle et al. (2010) (18) reported that activity monitor accuracy may be influenced by wearer behavior, and second, we assessed if the activity monitors were sensitive to change. Thus, we provide practical information that can be easily translated to natural settings.
This study also has several limitations. First, our sample was relatively young and lean and met the physical activity guidelines. We do not know how the activPAL and ActiGraph GT3X will perform for sedentary, overweight populations, particularly because the accuracy of the ActiGraph appears to be influenced by total sedentary time. Second, participants were only monitored for 1 d. This may have affected how participants changed their behavior from baseline to the treatment condition. It is not known how the monitors will perform when worn for longer periods or if the monitors will be sensitive to changes in habitual activity. Third, we tested the performance of the ActiGraph using output in 1-s epochs from the vertical axis only. We chose to assess performance in this way because previous studies (15,16) identified sedentary time and breaks in a similar manner. However, future studies should assess ActiGraph performance using additional epochs and axes to detect sedentary time. Future studies should also test the use of the ActiGraph inclinometer to detect transitions between seated and standing postures.
This study provides further evidence that the activPAL is a valid device for measuring total sedentary time (18), and our results provide the first evidence that the activPAL can identify specific features of free-living sedentary behavior. Relative to the criterion measure of direct observation, the activPAL accurately and precisely estimated sedentary time, the absolute number of breaks and break rate. It also agreed with direct observation and was sensitive enough to detect the effects of an intervention to reduce and break up sedentary time in free-living conditions.
This study also illustrates the difficulty in measuring sedentary behavior using a single hip-mounted accelerometer. In general, the ActiGraph GT3X did not accurately estimate total sedentary time, the absolute number of breaks or break rate. Its accuracy in estimating these metrics was dependent on 1) the cut point used to distinguish sedentary time, 2) the filter setting, and 3) the behavior of the sample population. In addition, the ActiGraph GT3X vertical axis did not detect the effects of an intervention to reduce and break up sedentary time regardless of the sedentary cut point and filter setting used.
By describing and validating measures sensitive to features of sedentary behavior, this study informs future investigation into how sedentary behavior influences health. When participants were given a simple recommendation relevant to and realistic for an increasingly sedentary public, break rate was sensitive to the behavioral changes observed. This is a relatively new field of research, and more evidence is needed to determine the specific features of sedentary behavior (e.g., total sedentary time, breaks per day, or break rate) that are important for health outcomes.
This study is funded by NHLBI 1RC1 HL099557-01. Patty Freedson is on the Scientific Advisory Board for ActiGraph, which manufactures one of the monitors used in this study. The authors would like to acknowledge the participants who volunteered for this study and all graduate and undergraduate students who helped with data collection.
Kate Lyden, Sarah Kozey-Keadle, and John Staudenmayer declare no conflicts of interest.
The results of this study do not constitute endorsement by the American College of Sports Medicine.
1. Bey L, Hamilton MT. Suppression of skeletal muscle lipoprotein lipase activity during physical inactivity: a molecular reason to maintain daily low-intensity activity. J Physiol
. 2003; 551 (Pt 2): 673–82.
2. Chastin SF, Baker K, Jones D, et al.. The pattern of habitual sedentary behavior
is different in advanced Parkinson’s disease. Mov Disord
. 2010; 25 (13): 2114–20.
3. Chastin SF, Granat MH. Methods for objective measure, quantification and analysis of sedentary behaviour and inactivity. Gait Posture
. 2010; 31 (1): 82–6.
4. Committee PAGA. Physical Activity Guidelines Advisory Committee Report
. In: Services USDoHaH, editor. Washington (DC): 2008. p. D-8.
5. Craig CL MA, Sjostrom M, Bauman AE, et al.. International Physical Activity Questionnaire: 12-country reliability and validity. Med Sci Sports Exerc
. 2003; 35 (8): 1381–95.
6. Crouter SE, Churilla JR, Bassett DR Jr. Estimating energy expenditure using accelerometers. Eur J Appl Physiol
. 2006; 98 (6): 601–12.
7. Crouter SE, Clowers K, Bassett DR Jr. A novel method for using accelerometer data to predict energy expenditure. J Appl Physiol
. 2006; 100: 1324–31.
8. Ekelund U, Brage S, Besson H, et al.. Time spent being sedentary and weight gain in healthy adults: reverse or bidirectional causality. Am J Clin Nutr
. 2008; 88: 612–7.
9. Freedson PS, Melanson E, Sirard J. Calibration of the Computer Science and Application Inc. Accelerometer. Med Sci Sports Exerc
. 1998; 30 (5): 777–81.
10. Godfrey A CK, Lyons GM. Comparison of the performance of the activpal
professional physical activity logger to a discrete accelerometer-based activity monitor. Med Eng Phys
. 2007; 29: 930–4.
11. Grant PM, Ryan CG, Tigbe WW, et al.. The validation of a novel activity monitor in the measurement of posture and motion during everyday activities. Br J Sports Med
. 2006; 40 (12): 992–7.
12. Hamilton MT, Hamilton DG, Zderic TW. Role of low energy expenditure and sitting in obesity, metabolic syndrome, type 2 diabetes, and cardiovascular disease. Diabetes
. 2007; 56 (11): 2655–67.
13. Healy GN, Clark BK, Winkler EA, et al.. Measurement of adults’ sedentary time in population-based studies. Am J Prev Med
. 2011; 41 (2): 216–27.
14. Healy GN, Dunstan DW, Salmon J, et al.. Objevtively measured light-intensity activity physical activity is independently associated with 2-h plasma glucose. Diabetes Care
. 2007; 30 (6): 1384–9.
15. Healy GN, Dunstan DW, Salmon J, et al.. Break in sedentary time. Diabetes Care
. 2008; 31 (4): 661–6.
16. Healy GN, Matthews CE, Dunstan DW, et al.. Sedentary time and cardio-metabolic biomarkers in US adults: NHANES 2003–06. Eur Heart J
. 2011; 32 (5): 590–7.
17. Helmerhorst HJ, Wijndaele K, Brage S, et al.. Objectively measured sedentary time may predict insulin resistance independent of moderate- and vigorous-intensity physical activity. Diabetes
. 2009; 58 (8): 1776–9.
18. Kozey-Keadle S, Libertine A, Lyden K, et al.. Validation of wearable monitors for assessing sedentary behavior
. Med Sci Sports Exerc
. 2011; 43 (8): 1561–7.
19. Lyden K, Kozey SL, Staudenmeyer JW, et al.. A comprehensive evaluation of commonly used accelerometer energy expenditure and MET prediction equations. Eur J Appl Physiol
. 2011; 111 (2): 187–201.
20. Lynch BM, Dunstan DW, Winkler E, et al.. Objectively assessed physical activity, sedentary time and waist circumference among prostate cancer survivors: findings from the National Health and Nutrition Examination Survey (2003–2006). Eur J Cancer Care (Engl)
. 2011; 20 (4): 514–9.
21. Maddocks M, Petrou A, Skipper L, Wilcock A. Validity of three accelerometers during treadmill walking and motor vehicle travel. Br J Sports Med
. 2010; 44: 606–8.
22. Matthew CE. Calibration of accelerometer output for adults. Med Sci Sports Exerc
. 2005; 37 (Suppl 11): S512–22.
23. Matthews CE, Chen KY, Freedson PS, et al.. Amount of time spent in sedentary behaviors in the United States, 2003–2004. Am J Epidemiol
. 2008; 167 (7): 875–81.
24. Owen N, Healy GN, Matthew CE, et al.. Too much sitting: the population health science of sedenatry behavior. Exerc Sport Sci Rev
. 2010; 38 (3): 105–13.
25. Rothney MP, Schaefer EV, Neumann MM, et al.. Validity of physical activity intensity predictions by ActiGraph
, Actical, and RT3 accelerometers. Obesity (Silver Spring)
. 2008; 16 (8): 1946–52.
26. Ryan CG, Grant PM, Tigbe WW, Granat MH. The validity and reliability of a novel activity monitor as a measure of walking. Br J Sports Med
. 2006; 40: 779–84.
27. Team RDC. R: A language and environment for statistical computing. Vienna, Austria2009 . Available from: http://www.R-project.org
28. Thorp AA, Owen N, Neuhaus M, et al.. Sedentary behaviors and subsequent health outcomes in adults a systematic review of longitudinal studies, 1996–2011. Am J Prev Med
. 2011; 41 (2): 207–15.
29. Zderic TW, Hamilton MT. Physical inacivity amplifies the sensitivity of skeletal muscle to the lipid-induced downregulation of lipoprotein lipase activity. J Appl Physiol
. 2006; 100: 249–57.