Reliability study comparing computed tomography (CT) to biomechanics.
To measure the accuracy and precision of such measurements in comparison with a standard method.
Rotations of lumbar spinal motion segments can be measured with dynamic CT imaging. This may be a useful tool to measure intersegmental motion. Validation of its use is lacking.
Human cadaveric lumbar spines were fixed in a rigid rotation device and rotated, whereas rotation at each level was measured with extensiometers. Rotation at each level was calculated as a percent of total rotation. The spines were placed in a CT scanner and imaged after rotation of the spine in each direction. The percent of total rotation that occurred at each level was calculated with a software program. Accuracy of the CT method was calculated as the average difference between methods. Precision was measured as the standard deviation of the CT measurement. Biomechanical testing and CT were repeated after the posterior anulus fibrosus at L3–L4 was incised with a scalpel. The power of the CT method to detect a change in rotation was tested by calculating the difference between the pre- and postsurgery rotation at L3–L4 and testing it for significance with a Student t test of paired samples.
Differences between CT and biomechanical measurements averaged 0.2%. Precision was 6.0% (Table 1). Postanular injury, percent rotation at the L3–L4 level increased, whereas it decreased at the other 4 levels. The change at L3–L4 was statistically significant (P = 0.047).
Dynamic CT measures vertebral rotations sufficiently accurately to study the effect of a radial tear on axial rotation can.
Measurement of axial rotation by computed tomography was compared with biomechanical testing in human lumbar cadaver spines. Computed tomography was shown to be accurate and precise and maybe a useful clinical tool.
From the Departments of *Orthopedic Surgery and Rehabilitation and †Radiology, University of Wisconsin, Madison, WI.
Acknowledgment date: March 18, 2008. Revision date: June 27, 2008. Acceptance date: August 18, 2008.
The manuscript submitted does not contain information about medical device(s)/drug(s).
No funds were received in support of this work. No benefits in any form have been or will be received from a commercial party related directly or indirectly to the subject of this manuscript.
Address correspondence and reprint requests to Paul A. Anderson, MD, 600 Highland Ave K4/736, Madison, WI 53792; E-mail: firstname.lastname@example.org