Study Design. A retrospective analysis of 1078 spinal surgical procedures with lumbar pedicle screw placement at a single institution.
Objective. Based on previously established normative values, triggered electromyographic stimulation (TrgEMG) was re-examined to evaluate its efficacy in determining screw malposition.
Summary of Background Data. Threshold values for confirmation of intraosseous placement of pedicle screws with EMG stimulation is controversial.
Methods. TrgEMG threshold values for 4857 pedicle screws placed from L2 to S1 from 1996 to 2005 were analyzed. An ascending method of constant current stimulation was applied to each pedicle screw to obtain a compound muscle action potential (CMAP) from lower extremity myotomes. Previously determined threshold value normative data from a published clinical series of 233 screws were as follows: 0 to 4 mA, high likelihood of pedicle wall breach; 4 to 8 mA, possible pedicle wall breach; >8 mA, no pedicle wall defect.
Results. A total of 7.74% (376 of 4857) of all screws tested had threshold values <8.0 mA. A total of 19.1% (72 of 376) of these were <4.0 mA: 54% (39 of 72) were repositioned (26) or removed (13) while the remaining 33 screws were left in place following repalpation. A total of 80.9% (304 of 376) had thresholds between 4 and 8 mA: 17.4% (53) were repositioned (38) or removed (15). Nine screws had thresholds of ≤2.8 mA and were either repositioned or removed following confirmation of a medial wall breach. A total of 74.5% (280 of 376) of all screws with thresholds <8.0 mA were verified as correctly placed by repalpation/radiography and therefore left in place.
Conclusion. The probability of a medial breach pedicle screw detected by triggered EMG stimulation increases with decreasing triggered EMG thresholds: 0.31% for >8.0 mA, 17.4% for 4.0 to 8.0 mA, 54.2% for <4.0 mA, and 100% for <2.8 mA. At 2.8 mA, triggered EMG has a specificity of 100%, with sensitivity of 8.4%; at 4.0 mA, specificity of 99% and sensitivity of 36%; and at 8.0 mA, 94% specificity and 86% sensitivity. TrgEMG is an adjuncttechnique and should always be used in conjunction with palpation and radiography to optimize safe pedicle screw placement.
A total of 4857 pedicle screws were placed in 1078 patients from L2–S1 using lower extremity myotome triggered EMGs for directional confirmation. Misdirection rarely occurred with threshold values of >8.0 mA, between 4 and 8 mA misdirection occurred in 17.4%, <4 mA in 54.2%, and <2.8 mA were misdirected. All screws placed were verified with intraoperative anteroposterior and lateral CT or radiograph.
From the Washington University Medical Center, Department of Orthopaedic Surgery, Barnes-Jewish Hospital Health Systems, St. Louis, MO.
Acknowledgment date: November 16, 2005. First revision date: April 28, 2006. Second revision date: January 25, 2007. Acceptance date: January 26, 2007.
The device(s)/drug(s) is/are FDA-approved or approved by corresponding national agency for this indication.
No funds were received in support of this work. No benefits in any form have been or will be received from a commercial party related directly or indirectly to the subject of this manuscript.
Address correspondence and reprint requests to Lawrence G. Lenke, MD, Washington University School of Medicine, Department of Orthopaedic Surgery, One Barnes-Jewish Hospital Plaza, Suite 11300, West Pavilion, Campus Box 8233, St. Louis, MO 63110; E-mail: email@example.com