Optometry & Vision Science:
Feature Article on Line
Bacterial Adhesion to Unworn and Worn Silicone Hydrogel Lenses
Vijay, Ajay Kumar*,†; Zhu, Hua†; Ozkan, Jerome*; Wu, Duojia§; Masoudi, Simin§; Bandara, Rani§; Borazjani, Roya N.†; Willcox, Mark D. P.†
Brien Holden Vision Institute (AKV, HZ, JO, DW, SM, RB, MDPW), and School of Optometry and Vision Science (HZ, MDPW), University of New South Wales, Sydney, New South Wales, Australia, and Alcon Labs, Fort Worth, Texas (RNB) Roya N. Borazjani is currently at CooperVision, Pleasanton, California.
Received February 6, 2012; accepted June 4, 2012.
Mark D. P. Willcox School of Optometry and Vision Science, University of New South Wales Sydney, New South Wales 2052 Australia e-mail: firstname.lastname@example.org
Purpose. The objective of this study was to determine the bacterial adhesion to various silicone hydrogel lens materials and to determine whether lens wear modulated adhesion.
Methods. Bacterial adhesion (total and viable cells) of Staphylococcus aureus (31, 38, and ATCC 6538) and Pseudomonas aeruginosa (6294, 6206, and GSU-3) to 10 commercially available different unworn and worn silicone hydrogel lenses was measured. Results of adhesion were correlated to polymer and surface properties of contact lenses.
Results. S. aureus adhesion to unworn lenses ranged from 2.8 × 104 to 4.4 × 105 colony forming units per lens. The highest adhesion was to lotrafilcon A lenses, and the lowest adhesion was to asmofilcon A lenses. P. aeruginosa adhesion to unworn lenses ranged from 8.9 × 105 to 3.2 × 106 colony forming units per lens. The highest adhesion was to comfilcon A lenses, and the lowest adhesion was to asmofilcon A and balafilcon A lenses. Lens wear altered bacterial adhesion, but the effect was specific to lens and strain type. Adhesion of bacteria, regardless of genera/species or lens wear, was generally correlated with the hydrophobicity of the lens; the less hydrophobic the lens surface, the greater the adhesion.
Conclusions. P. aeruginosa adhered in higher numbers to lenses in comparison with S. aureus strains, regardless of the lens type or lens wear. The effect of lens wear was specific to strain and lens. Hydrophobicity of the silicone hydrogel lens surface influenced the adhesion of bacterial cells.
Silicone hydrogels are rapidly becoming the contact lens of choice worldwide, as in many countries silicone hydrogel lenses are the most common lens prescribed for new and refits for daily wear.1 Initially developed for the extended-wear market, these materials have now been embraced for various modes of wear, including overnight, daily wear, and daily disposable wear modalities. Although these lenses have achieved excellent oxygen delivery to the cornea,2–4 they are still associated with adverse events during wear.5–7 Contact lens–related microbial keratitis (MK) is the most serious contact lens adverse response.8 MK is most frequently attributed to bacteria, although Acanthamoeba and fungal cases have also been reported.9–13 Other adverse responses, such as contact lens acute red eye, contact lens-induced peripheral ulcer (CLPU), and infiltrative keratitis, have also been attributed to bacterial contamination of lenses.14–18
The gram-negative bacterium Pseudomonas aeruginosa is the most common bacteria isolated in contact lens–related MK,9,11,19–21 whereas staphylococci are often most commonly isolated from non–contact lens–related MK.22 Furthermore, P. aeruginosa has been isolated from contact lenses at the time of a contact lens acute red eye response,14 and Staphylococcus aureus has been isolated from lenses at the time of a CLPU response.23 Silicone hydrogel lenses appear to exhibit higher bacterial adhesion than conventional hydrogel contact lenses in vitro.24–26 However, such lens materials possess diverse physical, chemical, and structure properties, which may affect attachment of microorganisms. The relative adhesion profiles of bacteria to recently available silicone hydrogel lenses are unknown.
The effect of lens wear on bacterial adhesion remains controversial. Boles et al.27 found that viable or total cells of a single strain of P. aeruginosa adhered better to unworn etafilcon A (HEMA-based hydrogel) lenses. Bruinsma et al.,28 using contact lenses, described only as “commercially available hydrogel contact lenses” demonstrated that in vitro adsorption of the tear film resulted in a decrease in the adhesion of a strain of P. aeruginosa (strain 3) but an increase in adhesion of a strain of S. aureus (strain 799). Willcox et al.29 found that a strain of P. aeruginosa adhered in lower numbers to worn etafilcon A lenses but in higher numbers to worn polymacon (HEMA-based hydrogel) lenses or balafilcon A silicone hydrogel lenses. However, a strain of Aeromonas hydrophilia adhered the same or better to those same types of worn lenses, whereas a strain of Stenotrophomonas maltophilia adhered in lower numbers to worn etafilcon A or polymacon lenses but in higher numbers to worn balafilcon A lenses.29 Borazjani et al.30 found that adhesion of a strain of P. aeruginosa to balafilcon A lenses was not affected by lens wear.
The objective of this study was to determine the adhesion of P. aeruginosa and S. aureus strains to various silicone hydrogel lens materials and to determine whether lens wear modulates adhesion. The specific aims of the study were to establish the adhesion of strains to unworn silicone hydrogel lenses and to determine whether daily disposable lens wear changed the levels of adhesion of these bacteria to individual lens types. In addition, correlations were sought between lens surface and polymer properties and bacterial adhesion.
MATERIALS AND METHODS
Ten commercially available silicone hydrogel contact lenses were used in the study (Table 1). Unworn contact lenses were used from blister packs after washing three times in phosphate buffered saline (PBS; NaCl, 8 g/l; KCl, 0.2 g/l; Na2HPO4, 1.15 g/l; KH2PO4, 0.2 g/l; pH 7.4), and worn lenses were collected from the clinical study described later in the text and washed three times in PBS.
All procedures were conducted in accordance with the 2000 Declaration of Helsinki and were approved by a local ethics committee. Participants were advised of potential adverse reactions and completed written informed consent before enrollment. Twenty-nine experienced contact lens wearers (mean age: 32 ± 13 years) of both sexes (15 female and 14 male) were enrolled for the study. Subjects attended up to six scheduled clinic visits to have their ocular health and contact lens fitting assessed, as well as lenses dispensed. Subjects were consecutively dispensed a 2-week supply of each lens type. Lens wear was bilateral, and the order of wear for each lens type was randomized for each subject. Contact lenses were worn for a minimum of 2 weeks on a daily disposable basis (i.e., removed before sleep each night and not worn again). Lenses were worn for a minimum of 6 h per day, after which time they were removed and stored dry in lens collection vials in the subject's refrigerator. Subjects were advised to wash their hands with soap and dry them before handling lenses. A minimum of six pairs of lenses (three pairs/week) were collected from each subject per lens material before performing the adhesion assays. Once a sufficient number of lenses were collected, subjects were then dispensed with a 2-week supply of the next lens type, as per the randomized allocation. Subjects were strictly instructed not to use any contact lens storage or disinfection solutions.
Micro-Organisms and Culture Preparation
P. aeruginosa strains 6294 (MK isolate), 6206 (MK isolate), GSU-3 (MK isolate) and S. aureus strains 031 (CLPU isolate), 038 (MK isolate), and ATCC 6538 (human isolate) were used in the adhesion experiments, and all were obtained from stock cultures kept at −80°C. After growth on chocolate agar plates, strains were grown overnight at 37°C in 10 ml of minimal medium (1.0 g/l D-glucose, 7.0 g/l K2HPO4, 2.0 g/l, KH2PO4, 0.5 g/l sodium citrate, 1.0 g/l (NH4)2SO4, 0.1 g/l MgSO4, pH 7.2)30 for viable adhesion, or minimal medium containing 0.2% v/v 3H-uridine (Perkin Elmer, Waltham, MA) for total adhesion. The bacterial cells were collected and washed three times by centrifugation and resuspended in sterile PBS. The concentration of the bacterial suspension was adjusted to 0.1 at 660 nm using a spectrophotometer (Heliosβ, Unicam Instruments, Cambridge, UK; approximately 1.0 × 108 colony forming units per ml; CFU/ml). The suspension was diluted 1/10 in PBS to obtain the final concentration of 1.0 × 107 CFU/ml.
Bacterial Adhesion Assay
Three unworn lenses of each lens material were used in the adhesion experiments, and each experiment was repeated twice for all six bacterial strains. A minimum of six pairs of worn lenses per lens material were collected from each subject. As stated previously, unworn and worn lenses were washed three times each time in 2 ml of PBS. After the washes in PBS, lenses were aseptically transferred to individual wells of a 24-well cell culture plate containing 1 ml of the bacterial suspension prepared as mentioned earlier and incubated at 37°C for 18 h. For recovery of viable cells, each lens was removed aseptically and washed three times in PBS, then placed in sterile plastic tubes containing 2 ml of PBS and a sterile magnetic stirrer, and vortexed for 2 min at maximum speed to detach adhered bacterial cells. The dislodged bacterial suspension was serially diluted 10-fold in PBS. Aliquots of each dilution were inoculated on nutrient agar plates in triplicate and incubated at 37°C for 18 h. After incubation, numbers of CFU were counted, and the numbers of viable bacteria per lens were calculated. Pilot studies confirmed that there were no remaining viable cells on contact lens fragments after vortexing (data not shown). The total number of bacteria adhered on lens surfaces was estimated after measuring the radioactivity of lenses using a β scintillation counter (Wallac, Perkin Elmer) after washing three times in PBS and converting the radioactivity into the numbers of cells using a standard curve of cell numbers.
Log transformation of the adhesion data was performed before data analysis. For unworn lenses, the general linear model was used to determine differences between lens types. For worn lenses, differences in bacterial adhesion between lens types were analyzed using linear mixed models to account for the repeated participant effects. A separate analysis was performed for each bacterial strain and viable or total counts. If the overall effect was significant, post hoc multiple comparisons were performed using the Bonferroni correction. Differences between worn and unworn lens types were analyzed for each lens type using t-test, but the p value was adjusted for multiple comparisons using the Bonferroni correction.
In addition, correlations were sought between bacterial adhesion to lenses and previously published data31–38 on surface characteristics of contact lenses, and contact lens oxygen permeability and water content, as given by the lens manufacturers (Table 2), using Pearson's correlation coefficient. Level of significance was set at 5% for each analysis. Polymer and surface characteristic data were only used for comparisons if more than three different lens materials had been measured for an individual characteristic and if lenses had been washed in PBS after removal from their packaging, with the exception of values for surface roughness and adhesion energy where no measurements had been reported for washed lenses. For lotrafilcon A and B lenses, data from previous publications were only used if it was clear that the lotrafilcon B lenses used contained AQUA, and that the lotrafilcon A lenses did not contain AQUA. If lenses had been analyzed by more than one research group, an average value was taken for the analysis. Thus, essentially the same captive bubble technique had been used by the same group on two separate occasions to measure the receding contact angle on balafilcon A (27.8°, 18.3°)31,23 or senofilcon A (27.8°, 22.1°).31,32 Essentially the same sessile drop technique had been used by several groups to measure the advancing contact angle of galyfilcon A (112°, 112.7°, 108°),33,34 senofilcon A (80°, 97°, 100°),31,33,34 comfilcon A (33°, 37.6°, 38°),33,34 lotrafilcon A (Focus Night & Day) (50.2°, 48°, 48°),31,33,34 asmofilcon A (105°, 82.1°),33 or balafilcon A (86.5°, 88°, 89.3°, 118°).31,33,34 Essentially the same measurement for the lens surface roughness (root mean square roughness) at 25 μm2 was made by the same research group and reported in two separate publications35,36 for galyfilcon A (6.75 and 3.68 nm, respectively), lotrafilcon A (Focus Night & Day) (4.67 and 4.98 nm, respectively), and balafilcon A (12.26 and 15.19 nm, respectively), and different groups had reported the surface roughness at 25 μm2 for comfilcon A (3.62 and 2.34 nm).36,37 Essentially the same measurement for the lens surface roughness (average roughness) at 25 μm2 was made by the same research group and reported in two separate publications35,36 for galyfilcon A (5.39 and 2.81 nm, respectively), lotrafilcon A (Focus Night & Day) (3.6 and 3.67 nm, respectively), and balafilcon A (9.55 and 11.62 nm, respectively), and different groups had reported the average surface roughness at 25 μm2 for comfilcon A (2.87 and 1.56 nm).36,37
Adhesion to Unworn Lenses
Table 3 reports the mean adhesion of the combined strains with the 95% confidence intervals; statistical significance (p < 0.05) is shown when the confidence intervals for adhesion to the lenses do not overlap. Fig. 1 represents the data graphically.
Combining the adhesion data for all three Staphylococcus strains (Table 3; Fig. 1A), significant differences in levels of bacterial adhesion were observed among test lens types. The major findings were as follows: (1) total adhesion to unworn asmofilcon A lenses was significantly less than to all other lens types (p ≤ 0.005); (2) total adhesion to unworn lotrafilcon A was greater than to unworn narafilcon A (p = 0.001), galyfilcon A (p = 0.012), enfilcon A (p = 0.002), and asmofilcon A (p = 0.000); (3) viable adhesion to unworn asmofilcon A was significantly less than to most lens types (p ≤ 0.001) except unworn enfilcon A (p = 1.000) or comfilcon A (p = 0.169); (4) viable adhesion to unworn enfilcon A was significantly less than to most lens types (p ≤ 0.001) except unworn comfilcon A (p = 1.000) or asmofilcon A (p = 1.000). The only other significant differences were greater viable adhesion to unworn lotrafilcon B than unworn comfilcon A (p = 0.019), unworn lotrafilcon A than unworn comfilcon A (p = 0.000), and unworn lotrafilcon A than unworn balafilcon A (p = 0.014).
Comparing the three Staphylococcus strains, S. aureus 38 adhered in greater total numbers (p = 0.000, Table 4) compared with S. aureus 31 and S. aureus ATCC 6538, regardless of the lens material. However, significantly lower numbers of viable S. aureus 38 (p = 0.000; Table 4) were recovered from test lenses compared with other test strains, indicating that there were non-cultivable cells of S. aureus 38 adherent to the lens surfaces.
TABLE 4-a. Compariso...Image Tools
Combining the adhesion data for all three Pseudomonas strains (Table 3; Fig. 1B), significant differences in levels of bacterial adhesion were observed among test lens types. The differences were as follows: (1) total adhesion to unworn lotrafilcon B, enfilcon A, comfilcon A, filcon II 3, or lotrafilcon A was greater than to unworn narafilcon A (p ≤ 0.002), galyfilcon A (p ≤ 0.021), senofilcon A (p ≤ 0.008), or asmofilcon A (p ≤ 0.004); (2) total adhesion to unworn balafilcon A was significantly less than to unworn lotrafilcon B (p = 0.037), comfilcon A (p = 0.000), or lotrafilcon A (p = 0.007); (3) viable adhesion to unworn asmofilcon A was significantly less than to all other lens types (p ≤ 0.001) except unworn narafilcon A (p = 0.468) or balafilcon A (p = 1.000); (4) viable adhesion to unworn balafilcon A was significantly less than to all other lens types (p = 0.000) except unworn narafilcon A (p = 0.267) or asmofilcon A (p = 1.000); (5) viable adhesion to unworn narafilcon A was significantly less than to unworn comfilcon A (p = 0.000) or filcon II 3 (p = 0.001) only.
Comparing P. aeruginosa strains, total counts of P. aeruginosa 6206 were significantly higher than P. aeruginosa 6294 (Table 4). Overall, there was no significant difference between the strains for viable counts on unworn lenses (Table 4). Regardless of the lens polymer type, P. aeruginosa strains adhered in greater numbers to unworn lenses compared with S. aureus strains (p < 0.001).
Adhesion to Worn Lenses
Table 3 and Fig. 1C, D show the mean adhesion of the combined strains with 95% confidence intervals.
Combining the adhesion data for all three Staphylococcus strains (Table 3; Fig. 1C), significant differences in levels of bacterial adhesion were observed among test lens types. The differences were as follows: (1) total adhesion to worn asmofilcon A was significantly less (p ≤ 0.01) than to all other worn lenses except to worn galyfilcon A (p = 1.000) and worn balafilcon A (p = 0.665); (2) total adhesion to worn filcon II 3 was significantly greater (p ≤ 0.028) than to all other worn lenses except to narafilcon A (p = 0.893) and lotrafilcon A (p = 0.475); (3) total adhesion to worn narafilcon A was significantly greater than to worn galyfilcon A (p = 0.000), asmofilcon A (p = 0.000), or balafilcon A (p = 0.021); (4) total adhesion to worn lotrafilcon A was significantly greater than to worn galyfilcon A (p = 0.004) or asmofilcon A (p = 0.000) or balafilcon A (p = 0.029); (5) viable adhesion to worn asmofilcon A was significantly less (p = 0.000) than to all other worn lenses except to worn filcon II 3 (p = 1.000) and worn balafilcon A (p = 0.751); (6) viable adhesion to worn balafilcon A was significantly less than to worn narafilcon A (p = 0.002), worn senofilcon A (p = 0.000), and worn lotrafilcon A (p = 0.029); (7) viable adhesion to worn lotrafilcon A was significantly greater than to worn filcon II 3 (p = 0.021), worn asmofilcon A (p = 0.000), and worn balafilcon A (p = 0.029); (8) viable adhesion to worn filcon II 3 was significantly less than to worn narafilcon A (p = 0.001), worn senofilcon A (p = 0.000), and worn lotrafilcon A (p = 0.021); (9) viable adhesion to worn enfilcon A was significantly less than to worn senofilcon A (p = 0.024).
For the individual S. aureus strains, the order of viable adhesion on worn lenses was S. aureus 31 > S. aureus ATCC 6538 (p = 0.000) > S. aureus 38 (p = 0.000; Table 3). S. aureus 38 adhered least to worn lenses (regardless of lens type; Table 4).
Combining the adhesion data for all three Pseudomonas strains (Table 3; Fig. 1D), significant differences in levels of bacterial adhesion were observed among test lens types. The differences were as follows: (1) total adhesion to worn asmofilcon A was significantly lower than to worn narafilcon A (p = 0.000), worn enfilcon A (p = 0.000), worn comfilcon A (p = 0.023), and worn filcon II 3 (p = 0.004); (2) total adhesion to worn enfilcon A was significantly higher than to worn galyfilcon A (p = 0.000), worn senofilcon A (p = 0.001), worn lotrafilcon B (p = 0.000), and worn asmofilcon A (p = 0.000); (3) total adhesion to worn galyfilcon A was significantly lower than to worn narafilcon A (p = 0.006) or enfilcon A (p = 0.000); (4) viable adhesion to worn narafilcon A was significantly (p ≤ 0.001) higher than to all other lenses except worn enfilcon A (1.000), worn comfilcon A (0.483), or worn filcon II 3 (0.170); (5) viable adhesion to worn enfilcon A was significantly (p ≤ 0.006) higher than to all other lenses except for worn narafilcon A (p = 1.000), worn comfilcon A (p = 1.000) or worn filcon II 3 (p = 0.596); (6) viable adhesion to worn asmofilcon A was significantly less than to worn narafilcon A (p = 0.000), worn enfilcon A (p = 0.000), worn comfilcon A (p = 0.001), and worn filcon II 3 (p = 0.001); (7) viable adhesion to worn galyfilcon A was significantly less than to worn narafilcon A (p = 0.000), worn enfilcon A (p = 0.000), or worn comfilcon A (p = 0.031).
Combining adhesion data from all the worn lens materials (Table 4), P. aeruginosa 6206 showed the least viable counts compared with the other strains, whereas strain GSU-3 showed the least total counts compared with strain 6206 only (p < 0.05). Similar to the unworn lenses, P. aeruginosa strains adhered in greater numbers to all silicone hydrogel lens materials compared with S. aureus strains.
Effect of Lens Wear on Adhesion
There were more total than viable S. aureus adherent to all the lens materials after wear (Table 3; Fig. 1C). For S. aureus overall, significantly higher numbers of total bacterial cells adhered to worn narafilcon A, enfilcon A, and filcon II 3 lenses compared with the unworn lenses (p < 0.05). No significant differences were observed for the overall number of viable cells on worn compared with unworn lenses.
S. aureus 31 showed significantly higher total adhesion to worn compared with unworn narafilcon A, senofilcon A, lotrafilcon B, enfilcon A, comfilcon A, and asmofilcon A lenses (p ≤ 0.009, Fig. 2). The viable adhesion of S. aureus 31 on most of the worn lenses was similar to unworn lenses except for increases in adhesion to senofilcon A (p = 0.003) and enfilcon A (p = 0.000) lenses. S. aureus 38 showed significantly less total adhesion to worn compared with unworn senofilcon A and balafilcon A lenses (p ≤ 0.03, Fig. 2); no significant differences were seen for the other lens materials. S. aureus 38 had higher levels of viable cells adhered to worn narafilcon A, enfilcon A, and comfilcon A lenses compared with unworn lenses (p ≤ 0.041, Fig. 2). Total adhesion of S. aureus ATCC 6538 was not affected by wear to any lens type. Lens wear reduced the numbers of viable S. aureus ATCC 6538 on galyfilcon A, lotrafilcon B, and lotrafilcon A, but increased the numbers of viable cells on asmofilcon A lenses compared with unworn lenses (p ≤ 0.015, Fig. 2).
When the effect of lens wear on overall adhesion of P. aeruginosa was analyzed using combined data from three test strains, there was a significant increase in Pseudomonas total adhesion to narafilcon A lenses (p = 0.0001) compared with unworn lenses, whereas significantly lower total numbers adhered to worn comfilcon A lenses (p < 0.05) compared with unworn lenses. Lower viable numbers of P. aeruginosa adhered to worn galyfilcon A, lotrafilcon B, and lotrafilcon A lenses (p < 0.05) compared with unworn lenses, but higher viable numbers adhered to worn balafilcon A lenses (p < 0.05) compared with unworn lenses.
Significantly lower numbers of viable P. aeruginosa 6206 adhered to worn galyfilcon A (p = 0.042) compared with unworn lenses. Significantly higher total adhesion of P. aeruginosa 6294 was seen to worn senofilcon A lenses (p = 0.024, Fig. 2) compared with unworn lenses. Significantly higher numbers of viable cells of P. aeruginosa 6294 adhered to worn narafilcon A (p = 0.003) or enfilcon A (p = 0.042) lenses compared with unworn lenses. P. aeruginosa GSU-3 showed significantly higher total adhesion to worn narafilcon A lenses (p = 0.003, Fig. 2) compared with unworn lenses, but there was significantly lower total adhesion of cells to worn lotrafilcon B (p = 0.001) or comfilcon A (p = 0.005) lenses compared with unworn lenses. There was significantly increased adhesion of viable GSU-3 to worn narafilcon A lenses (p = 0.000), but lower numbers of viable cells adhered to worn lotrafilcon B (p = 0.000) or comfilcon A (p = 0.008) lenses compared with unworn lenses.
Correlations between Bacterial Adhesion and Contact Lens Polymer and Surface Properties
Correlations were sought for data on the contact lens properties (Table 2) and bacterial adhesion (Table 3). There was a relatively strong (r > ±0.5) correlation between lens oxygen permeability and adhesion for S. aureus total cells; the higher the oxygen permeability, the higher the adhesion (Table 5). However, overall, there was no strong relationship between bacterial adhesion and oxygen permeability of a lens, as this was the only correlation seen for this material property. There was a relatively strong correlation (r > −0.5) between adhesion of S. aureus (total or viable cells) and the water content of a lens such that the higher the water content of the lens, the lower the adhesion of S. aureus.
In general, there were moderate (±0.3 ≤ r ≤ ±0.5) to strong (r > ±0.5) correlations between Pseudomonas or Staphylococcus adhesion and contact lens surface hydrophobicity (however measured) such that the more hydrophobic the surface, the less bacterial adhesion there was. In particular, there was a strong correlation between total adhesion of P. aeruginosa and contact angle measured by receding captive bubble, advancing sessile drop, or Wilhelmy balance techniques (Table 5); a strong correlation between viable adhesion of P. aeruginosa and contact angle measured by advancing captive bubble, advancing sessile drop, or Wilhelmy balance techniques (Table 5); and a strong correlation between total adhesion of S. aureus and advancing captive bubble, advancing sessile drop, or Wilhelmy balance techniques (Table 5).
The number of total or viable P. aeruginosa that adhered to lenses appeared to be dependent on the surface roughness and surface energy of the contact lenses such that surfaces with higher roughness or lower surface energy tended to adhere fewer bacterial cells (Table 5). There was no such relationship to total or viable S. aureus adhesion.
This study found that strains of P. aeruginosa adhere in higher numbers than strains of S. aureus to silicone hydrogel lenses, regardless of lens polymer type or surface properties of the lens. For both bacterial genera, there was a relationship between the surface hydrophobicity of the lens and numbers of bacterial cells adhering to the lens. Wearing the contact lenses for 1 day (on a daily disposable basis) had different effects on adhesion of the bacteria, depending on the bacterial strain and lens type. Bacterial adhesion to contact lenses may be significant, as it is likely to be the initial event in the production of adverse responses, such as MK, contact lens-induced acute red eye, and contact lens-induced peripheral ulcers.
The finding that P. aeruginosa strains tend to adhere to contact lenses in higher numbers than other bacterial types has been previously reported by others. Kodjikian et al.25 found that the number of viable P. aeruginosa cells on lenses was generally higher than the number of viable Staphylococcus epidermidis. Henriques et al.24 also found that a strain of S. epidermidis (strain 12,228) adhered in lower numbers to lotrafilcon A and balafilcon A (as well as the HEMA-based etafilcon A) lenses than a strain of P. aeruginosa (10145). However, another strain of S. epidermidis (9142) adhered to these silicone hydrogel lens types in similar numbers as the strain of P. aeruginosa. P. aeruginosa strains 6206 and 6294 previously have been shown to adhere in higher numbers to unworn etafilcon A, lotrafilcon A,39 senofilcon A, or balafilcon A39,40 lenses compared with S. aureus strain 31, and this was confirmed in the present study. Using one of the same strains as in the current study, namely, P. aeruginosa GSU-3 (strain 328), Bruinsma et al.28 found that P. aeruginosa GSU-3, with a relatively hydrophobic cell surface, adhered in higher numbers to contact lenses (type of contact lens not stated) than the relatively hydrophilic S. aureus 799, and this was regardless of whether the lenses had been soaked in tears. The ability of P. aeruginosa to adhere in high numbers to many contact lens types may be one of the reasons that this bacterium is the predominant causative microbe for contact-lens–related MK,9,11,19–21 and it can also cause other adverse events.14,29
Using a method (ATP bioluminescence) that measures numbers of viable bacteria adhered to a surface, Kodjikian et al.25 found that two strains of S. epidermidis and one strain of P. aeruginosa adhered maximally to unworn lotrafilcon B lenses, and one strain of S. epidermidis and the strain of P. aeruginosa adhered better to unworn balafilcon A lenses than unworn galyfilcon A lenses. The lotrafilcon B lenses used in the Kodjikian et al.25 study were slightly different than those used in the current assay, as they were an older version that did not contain the “AQUA” internal wetting agent. Even so, in the current study, the number of viable cells of P. aeruginosa or S. aureus was generally greater on unworn lotrafilcon B lenses than on unworn galyfilcon A or balafilcon A lenses.
The effect of the hydrophobicity of substrata on bacterial adhesion is somewhat controversial. Fletcher and Loeb41 showed that adhesion of a marine pseudomonad was driven by surface hydrophobicity such that advancing contact angles and bacterial adhesion had a correlation coefficient of 0.73 (R2 = 0.54), i.e., the more hydrophobic the substratum, the higher the adhesion. Similarly, Fletcher and Marshall42 found that more hydrophobic surfaces tend to adhere more of a particular strain of Pseudomonas. Giraldez et al.37 found that, when comparing adhesion of S. epidermidis with three HEMA-based and two silicone hydrogel contact lenses, the hydrophobicity of the lens surface was partly correlated to adhesion; the more hydrophobic silicone hydrogel lenses adhere more viable bacterial cells than the less hydrophobic HEMA-based lenses. However, the formation of dental plaque on surfaces in the human oral cavity is far less on hydrophobic than on hydrophilic surfaces.43 Hydrophobic silicone rubber voice prostheses in patients who had had a laryngectomy were contaminated by less biofilm than a prosthesis with a relatively hydrophilic surface.44 Tang et al.,45 using silicone samples but not contact lenses, showed that adhesion of S. epidermidis (measured by counting viable cells) decreased with increasing substrate hydrophobicity. Data from the present study suggest that adhesion of P. aeruginosa or S. aureus is generally less to more hydrophobic silicone hydrogel lenses. Henriques et al.24 reported the advancing contact angle of balafilcon A, lotrafilcon A, and galyfilcon A lenses after they had been soaked in artificial tear fluid to be approximately 75°, 51°, and 35°. If these values are substituted for the values of non–artificial tear-soaked lenses and then compared with adhesion of P. aeruginosa or S. aureus to these lens types after wear from the present study, the correlation coefficients (r) for P. aeruginosa are 0.75 and 0.85 for total and viable cells, respectively, and for S. aureus, they are 0.07 and −0.99 for total and viable cells, respectively. In general, the contact angles are reduced after soaking in artificial tear fluid (especially for galyfilcon A), and this leads to (at least for P. aeruginosa) a change in adhesion characteristics such that adhesion (total or viable) is now increased with increasing hydrophobicity. This result from the current investigation is in general agreement with Henriques et al.24 and their reported adhesion data. Bruinsma et al.28 also found that adsorbing tears to contact lenses (types not stated) generally decreased their surface hydrophobicity. The current study did not examine the adhesion to any conventional HEMA-based hydrogel materials. However, Kodjikian et al.25 found that a strain of P. aeruginosa adhered to a HEMA-based (etalfilcon A) or silicone hydrogel lenses in higher numbers than two strains of S. aureus. If the data from Kodjikian et al.25 are compared with the data from Read et al.,31 who examined the surface contact angle using the advancing sessile drop technique for the same HEMA-based and silicone hydrogel lenses, it appears that adhesion to the HEMA-based lenses is less for both genera of bacteria, and that the relationship between adhesion and surface hydrophobicity is parabolic [equation for P. aeruginosa is adhesion = −0.02(contact angle)2 + 2.6775 × –22.535; R2 = 0.92; for S. aureus is adhesion = y = −0.0006(contact angle)2 + 0.0759 × –0.3005; R2 = 0.6294] with a maximum adhesion lying between 40° and 60° for both genera, and a rapid decline toward minima below 20° or at or above 100°.
When adhesion to surfaces with similar hydrophobicities, but differing surface roughness, was examined, a higher surface roughness resulted in an increase in bacterial adhesion.45 The present study found that surface roughness was moderately to strongly negatively correlated with adhesion of total or viable P. aeruginosa, but not with adhesion of S. aureus. Giraldez et al.37 found that lenses with a higher average surface roughness (nm at 25 μm2) adhered more viable S. epidermidis cells. Santos et al.46 found that during wear of galyfilcon A, balafilcon A, lotrafilcon A, or lotrafilcon B (non-AQUA) lenses, microbial colonization of the lens during wear was different, with colonization of balafilcon A lenses being greater than the other lens types; they speculated that the surface roughness of balafilcon A lenses contributed to their ability to be colonized by microbes during wear. González-Méijome et al.36 have published on the root-mean-square and average roughness on lenses (galyfilcon A, comfilcon A, lotrafilcon A, and balafilcon A) worn on a daily wear schedule for 2 weeks (galyfilcon A) or 1 month (other lens types). If those values were used to correlate with adhesion to worn lenses, there is still a moderate correlation between adhesion of total cells of P. aeruginosa and root-mean-square (r = −0.41; r2 = 0.17) and average (r = −0.45; r2 = 0.20) roughness of the worn lenses; strong correlations between viable cell adhesion of P. aeruginosa and root-mean-square (r = −0.83; r2 = 0.67) and average (r = −0.87; r2 = 0.75) roughness of the worn lenses. There were also moderate correlations between root-mean-square roughness and total (r = −0.41; r2 = 0.17) or viable (r = −0.35; r2 = 0.13) adhesion of S. aureus; average surface roughness and total adhesion (r = −0.32; r2 = 0.10) of S. aureus. Differences in the effects of substrate surface roughness may simply be because of the different bacterial types used in the studies. The hydrophobicity of cells of P. aeruginosa can contribute to adhesion (total cells) on a HEMA-based hydrogel contact lens; the more hydrophobic the surface of the bacteria, the more they adhered to the contact lens.47 Similar to the results from the current study, Miller and Ahearn48 found no correlation between the water content of HEMA-based hydrogel lenses and adhesion (total cells) of strains of P. aeruginosa.
The observation that adhesion of P. aeruginosa was associated with surface energy confirms data of others.49–51 Adhesion of several different bacterial types has been shown to be minimal between 20 to 30 mN/m.49–51 The results of the current analyses demonstrate that for viable adhesion of P. aeruginosa, the lowest adhesion occurred for the contact lens, balafilcon A, with the lowest surface free energy (15 nM/m). This was also generally the case for total adhesion of P. aeruginosa and total and viable adhesion of S. aureus. However, as the surface free energy results had not been measured after rinsing in PBS, the results may not be fully compatible, and further research is warranted.
Differences between the ability to adhere to contact lenses of different strains within a species/genera of bacteria have been previously reported. Fleiszig et al.52 demonstrated that adhesion of viable S. epidermidis to HEMA-based soft lenses was strain dependent. Williams et al.53 showed that adhesion to worn HEMA-based etafilcon A lenses by P. aeruginosa was strain dependant, with strain 6294 adhering in higher numbers to this lens type than strain 6206 that correlated to the findings of the current study for worn silicone hydrogel lenses.
The effect of contact lens wear on bacterial adhesion has been controversial. There have been reports that P. aeruginosa can adhere (total cells) more to worn extended wear than unworn HEMA-based soft contact lenses.54 Yet Boles et al.,27 measuring both total (using scanning electron microscopy) and viable cells of P. aeruginosa, found greater adhesion to unworn HEMA-based (etafilcon A) lenses than to lenses worn on an extended wear basis. Babaei Omali et al.40 found that after daily lens wear, when subjects used multipurpose disinfecting solution overnight, there was a significant reduction in viable cell adhesion of P. aeruginosa 6294 and S. aureus 31 to balafilcon A, but not to senofilcon A, lenses. In the present study that examined adhesion to lenses after daily disposable wear (i.e., no use of multipurpose disinfecting solution), there was an increase in viable S. aureus on senofilcon A lenses after wear, but no significant change in viable adhesion to worn balafilcon A lenses, and no significant change in viable adhesion of P. aeruginosa 6294 to either of the lens types. These differences possibly reflect the effect of adsorbed components of the multipurpose disinfecting solution to the lens during daily wear. Williams et al.53 found that adhesion of total cells was greater than adhesion of viable cells, as was found in the present study, and that lens wear generally increased total adhesion, but decreased viable adhesion, for a strain (Paer1) of P. aeruginosa to etafilcon A lenses.53 In the present study, for the silicone hydrogels, lens wear decreased viable cells adhesion to galyfilcon A, lotrafilcon B, and lotrafilcon A lenses (p < 0.05), but enhanced viable cells adhesion to balafilcon A lenses (p < 0.05). However, there were strain differences in the effect of lens wear on adhesion. Differences in effect of wear during extended vs. daily disposable wear could be because of different lens polymers that have been investigated and the effect of adsorbed tear film components. Butrus et al.54 found that addition of tear film components, such as albumin, lysozyme, and lactoferrin to a HEMA-based contact lens, increased adhesion. Although lactoferrin deposition on either HEMA-based or silicone hydrogel contact lenses can increase total adhesion, this protein can also decrease the viability of adhered P. aeruginosa.39,53 As subjects handled lenses during the removal process and did not wear gloves, it is possible that adhesion to lenses was also affected by deposition of biological material from hands onto the lens surface.
Bacterial adhesion to contact lens surfaces may be dependent on the surface properties of the bacterial cell (differences between strains) and the contact lens. Hydrophobicity of a silicone hydrogel lens surface appears to affect the ability of bacteria to adhere. Although different strains of bacteria can behave differently, overall, daily disposable contact lens wear results in increased numbers of total cells of S. aureus, but had less overall effect on total numbers of P. aeruginosa. Lens wear did not affect the viable numbers of S. aureus adhered to lenses, but did affect viable numbers of P. aeruginosa, and this was contact lens type dependent. The finding that P. aeruginosa can adhere in higher numbers (approximately 1 to 1.5 log increase) to all silicone hydrogel lenses compared with S. aureus may be clinically relevant, as P. aeruginosa is usually more frequently associated with MK than S. aureus,21,55–57 and a first step in the production of MK is likely to be adhesion to contact lens surfaces. On the other hand, there have been no reports of differences in rates of MK or other corneal infiltrative events with respect to different silicone hydrogel materials.7,58,59 Thus, it may be that the differences in adhesion of P. aeruginosa or S. aureus to various silicone hydrogel lenses of between 0.51 to 0.33 logs are perhaps not sufficient to impact the adverse event rate as a consequence of the adhesion of these bacteria to lenses.
Mark D. P. Willcox
School of Optometry and Vision Science,
University of New South Wales
Sydney, New South Wales 2052
1. Morgan PB, Efron N, Helland M, Itoi M, Jones D, Nichols JJ, van der Worp E, Woods CA. Twenty first century trends in silicone hydrogel contact lens fitting: an international perspective. Cont Lens Anterior Eye 2010;33:196–8.
2. Stern J, Wong R, Naduvilath TJ, Stretton S, Holden BA, Sweeney DF. Comparison of the performance of 6- or 30-night extended wear schedules with silicone hydrogel lenses over 3 years. Optom Vis Sci 2004;81:398–406.
3. Dumbleton K, Keir N, Moezzi A, Feng Y, Jones L, Fonn D. Objective and subjective responses in patients refitted to daily-wear silicone hydrogel contact lenses. Optom Vis Sci 2006;83:758–68.
4. Moezzi AM, Fonn D, Simpson TL. Overnight corneal swelling with silicone hydrogel contact lenses with high oxygen transmissibility. Eye Contact Lens 2006;32:277–80.
5. Carnt NA, Evans VE, Naduvilath TJ, Willcox MD, Papas EB, Frick KD, Holden BA. Contact lens-related adverse events and the silicone hydrogel lenses and daily wear care system used. Arch Ophthalmol 2009;127:1616–23.
6. Schein OD, McNally JJ, Katz J, Chalmers RL, Tielsch JM, Alfonso E, Bullimore M, O'Day D, Shovlin J. The incidence of microbial keratitis among wearers of a 30-day silicone hydrogel extended-wear contact lens. Ophthalmology 2005;112:2172–9.
7. Stapleton F, Keay L, Edwards K, Naduvilath T, Dart JK, Brian G, Holden BA. The incidence of contact lens-related microbial keratitis in Australia. Ophthalmology 2008;115:1655–62.
8. Keay L, Stapleton F, Schein O. Epidemiology of contact lens-related inflammation and microbial keratitis: a 20-year perspective. Eye Contact Lens 2007;33:346–53.
9. Sharma S, Gopalakrishnan S, Aasuri MK, Garg P, Rao GN. Trends in contact lens-associated microbial keratitis in Southern India. Ophthalmology 2003;110:138–43.
10. Toshida H, Kogure N, Inoue N, Murakami A. Trends in microbial keratitis in Japan. Eye Contact Lens 2007;33:70–3.
11. Green M, Apel A, Stapleton F. Risk factors and causative organisms in microbial keratitis. Cornea 2008;27:22–7.
12. Moriyama AS, Hofling-Lima AL. Contact lens-associated microbial keratitis. Arq Bras Oftalmol 2008;71:32–6.
13. Saeed A, D'Arcy F, Stack J, Collum LM, Power W, Beatty S. Risk factors, microbiological findings, and clinical outcomes in cases of microbial keratitis admitted to a tertiary referral center in ireland. Cornea 2009;28:285–92.
14. Holden BA, La Hood D, Grant T, Newton-Howes J, Baleriola-Lucas C, Willcox MD, Sweeney DF. Gram-negative bacteria can induce contact lens related acute red eye (CLARE) responses. CLAO J 1996;22:47–52.
15. Sankaridurg PR, Vuppala N, Sreedharan A, Vadlamudi J, Rao GN. Gram negative bacteria and contact lens induced acute red eye. Indian J Ophthalmol 1996;44:29–32.
16. Keay L, Harmis N, Corrigan K, Sweeney D, Willcox M. Infiltrative keratitis associated with extended wear of hydrogel lenses and Abiotrophia defectiva
. Cornea 2000;19:864–9.
17. Sankaridurg PR, Sharma S, Willcox M, Naduvilath TJ, Sweeney DF, Holden BA, Rao GN. Bacterial colonization of disposable soft contact lenses is greater during corneal infiltrative events than during asymptomatic extended lens wear. J Clin Microbiol 2000;38:4420–4.
18. Wu P, Stapleton F, Willcox MD. The causes of and cures for contact lens-induced peripheral ulcer. Eye Contact Lens 2003;29:S63–6.
19. Lam DS, Houang E, Fan DS, Lyon D, Seal D, Wong E. Incidence and risk factors for microbial keratitis in Hong Kong: comparison with Europe and North America. Eye 2002;16:608–18.
20. Bourcier T, Thomas F, Borderie V, Chaumeil C, Laroche L. Bacterial keratitis: predisposing factors, clinical and microbiological review of 300 cases. Br J Ophthalmol 2003;87:834–8.
21. Pachigolla G, Blomquist P, Cavanagh HD. Microbial keratitis pathogens and antibiotic susceptibilities: a 5-year review of cases at an urban county hospital in north Texas. Eye Contact Lens 2007;33:45–9.
22. Shah A, Sachdev A, Coggon D, Hossain P. Geographic variations in microbial keratitis: an analysis of the peer-reviewed literature. Br J Ophthalmol 2011;95:762–7.
23. Jalbert I, Willcox MD, Sweeney DF. Isolation of Staphylococcus aureus
from a contact lens at the time of a contact lens-induced peripheral ulcer: case report. Cornea 2000;19:116–20.
24. Henriques M, Sousa C, Lira M, Elisabete M, Oliveira R, Azeredo J. Adhesion of Pseudomonas aeruginosa
and Staphylococcus epidermidis
to silicone-hydrogel contact lenses. Optom Vis Sci 2005;82:446–50.
25. Kodjikian L, Casoli-Bergeron E, Malet F, Janin-Manificat H, Freney J, Burillon C, Colin J, Steghens JP. Bacterial adhesion to conventional hydrogel and new silicone-hydrogel contact lens materials. Graefes Arch Clin Exp Ophthalmol 2008;246:267–73.
26. Santos L, Rodrigues D, Lira M, Real Oliveira ME, Oliveira R, Vilar EY, Azeredo J. Bacterial adhesion to worn silicone hydrogel contact lenses. Optom Vis Sci 2008;85:520–5.
27. Boles SF, Refojo MF, Leong FL. Attachment of Pseudomonas
to human-worn, disposable etafilcon A contact lenses. Cornea 1992;11:47–52.
28. Bruinsma GM, van der Mei HC, Busscher HJ. Bacterial adhesion to surface hydrophilic and hydrophobic contact lenses. Biomaterials 2001;22:3217–24.
29. Willcox MD, Harmis N, Cowell, Williams T, Holden BA. Bacterial interactions with contact lenses; effects of lens material, lens wear and microbial physiology. Biomaterials 2001;22:3235–47.
30. Borazjani RN, Levy B, Ahearn DG. Relative primary adhesion of Pseudomonas aeruginosa
, Serratia marcescens
and Staphylococcus aureus
to HEMA-type contact lenses and an extended wear silicone hydrogel contact lens of high oxygen permeability. Cont Lens Anterior Eye 2004;27:3–8.
31. Read ML, Morgan PB, Maldonado-Codina C. Measurement errors related to contact angle analysis of hydrogel and silicone hydrogel contact lenses. J Biomed Mater Res B Appl Biomater 2009;91:662–8.
32. Read ML, Morgan PB, Kelly JM, Maldonado-Codina C. Dynamic contact angle analysis of silicone hydrogel contact lenses. J Biomater Appl 2011;26:85–99.
33. Menzies KL, Jones L. The impact of contact angle on the biocompatibility of biomaterials. Optom Vis Sci 2010;87:387–99.
35. González-Méijome JM, Lopez-Alemany A, Almeida JB, Parafita MA, Refojo MF. Microscopic observation of unworn siloxane-hydrogel soft contact lenses by atomic force microscopy. J Biomed Mater Res B Appl Biomater 2006;76:412–8.
36. González-Méijome JM, Lopez-Alemany A, Almeida JB, Parafita MA. Surface AFM microscopy of unworn and worn samples of silicone hydrogel contact lenses. J Biomed Mater Res B Appl Biomater 2009;88:75–82.
37. Giraldez MJ, Serra C, Lira M, Real Oliveira ME, Yebra-Pimentel E. Soft contact lens surface profile by atomic force microscopy. Optom Vis Sci 2010;87:475–81.
38. Lin MC, Svitova TF. Contact lenses wettability in vitro: effect of surface-active ingredients. Optom Vis Sci 2010;87:440–7.
39. Subbaraman LN, Borazjani R, Zhu H, Zhao Z, Jones L, Willcox MD. Influence of protein deposition on bacterial adhesion to contact lenses. Optom Vis Sci 2011;88:959–66.
40. Babaei Omali N, Zhu H, Zhao Z, Ozkan J, Xu B, Borazjani R, Willcox MD. Effect of cholesterol deposition on bacterial adhesion to contact lenses. Optom Vis Sci 2011;88:950–8.
41. Fletcher M, Loeb GI. Influence of substratum characteristics on the attachment of a marine pseudomonad to solid surfaces. Appl Environ Microbiol 1979;37:67–72.
42. Fletcher M, Marshall KC. Bubble contact angle method for evaluating substratum interfacial characteristics and its relevance to bacterial attachment. Appl Environ Microbiol 1982;44:184–92.
43. Quirynen M, Marechal M, Busscher HJ, Weerkamp AH, Arends J, Darius PL, van Steenberghe D. The influence of surface free-energy on planimetric plaque growth in man. J Dent Res 1989;68:796–9.
44. Everaert EP, Mahieu HF, Wong Chung RP, Verkerke GJ, van der Mei HC, Busscher HJ. A new method for in vivo evaluation of biofilms on surface-modified silicone rubber voice prostheses. Eur Arch Otorhinolaryngol 1997;254:261–3.
45. Tang H, Cao T, Liang X, Wang A, Salley SO, McAllister J II, Ng KY. Influence of silicone surface roughness and hydrophobicity on adhesion and colonization of Staphylococcus epidermidis
. J Biomed Mater Res A 2009;88:454–63.
46. Santos L, Rodrigues D, Lira M, Oliveira ME, Oliveira R, Vilar EY, Azeredo J. The influence of surface treatment on hydrophobicity, protein adsorption and microbial colonisation of silicone hydrogel contact lenses. Cont Lens Anterior Eye 2007;30:183–8.
47. Klotz SA, Butrus SI, Misra RP, Osato MS. The contribution of bacterial surface hydrophobicity to the process of adherence of Pseudomonas aeruginosa
to hydrophilic contact lenses. Curr Eye Res 1989;8:195–202.
48. Miller MJ, Ahearn DG. Adherence of Pseudomonas aeruginosa
to hydrophilic contact lenses and other substrata. J Clin Microbiol 1987;25:1392–7.
49. Baier RE. Substrata influences on the adhesion of micro-organisms and their resultant new surface properties. In: Bitton G, Marshall KC, eds. Adsorption of Microorganisms to Surfaces. New York, NY: Wiley; 1980:59–104.
50. Zhao Q, Wang S, Müller-Steinhagen H. Tailored surface free energy of membrane diffusers to minimize microbial adhesion. J Appl Surface Sci 2004;230:371–8.
51. Pereni CI, Zhao Q, Liu Y, Abel E. Surface free energy effect on bacterial retention. Colloids Surf (B) 2006;48:143–7.
52. Fleiszig SM, Evans DJ, Mowrey-McKee MF, Payor R, Zaidi TS, Vallas V, Muller E, Pier GB. Factors affecting Staphylococcus epidermidis
adhesion to contact lenses. Optom Vis Sci 1996;73:590–4.
53. Williams TJ, Schneider RP, Willcox MD. The effect of protein-coated contact lenses on the adhesion and viability of gram negative bacteria. Curr Eye Res 2003;27:227–35.
54. Butrus SI, Klotz SA, Misra RP. The adherence of Pseudomonas aeruginosa
to soft contact lenses. Ophthalmology 1987;94:1310–4.
55. van der Meulen IJ, van Rooij J, Nieuwendaal CP, Van Cleijnenbreugel H, Geerards AJ, Remeijer L. Age-related risk factors, culture outcomes, and prognosis in patients admitted with infectious keratitis to two Dutch tertiary referral centers. Cornea 2008;27:539–44.
56. Bharathi MJ, Ramakrishnan R, Shivakumar C, Meenakshi R, Lionalraj D. Etiology and antibacterial susceptibility pattern of community-acquired bacterial ocular infections in a tertiary eye care hospital in south India. Indian J Ophthalmol 2010;58:497–507.
57. Stapleton F, Keay LJ, Sanfilippo PG, Katiyar S, Edwards KP, Naduvilath T. Relationship between climate, disease severity, and causative organism for contact lens-associated microbial keratitis in Australia. Am J Ophthalmol 2007;144:690–8.
58. Chalmers RL, Keay L, McNally J, Kern J. Multicenter case-control study of the role of lens materials and care products on the development of corneal infiltrates. Optom Vis Sci 2012;89:316–25.
59. Radford CF, Minassian D, Dart JK, Stapleton F, Verma S. Risk factors for nonulcerative contact lens complications in an ophthalmic accident and emergency department: a case-control study. Ophthalmology 2009;116:385–92.
contact lens; bacterial adhesion; Pseudomonas; Staphylococcus
This article has been cited 1 time(s).
Contact Lens & Anterior Eye
Ocular surface health with contact lens wear
Contact Lens & Anterior Eye, 36():
© 2012 American Academy of Optometry
What does "Remember me" mean?
By checking this box, you'll stay logged in until you logout. You'll get easier access to your articles, collections,
media, and all your other content, even if you close your browser or shut down your
To protect your most sensitive data and activities (like changing your password),
we'll ask you to re-enter your password when you access these services.
What if I'm on a computer that I share with others?
If you're using a public computer or you share this computer with others, we recommend
that you uncheck the "Remember me" box.
Highlight selected keywords in the article text.
Data is temporarily unavailable. Please try again soon.
Readers Of this Article Also Read