We did not consider the Stiles-Crawford effect, but this can be incorporated into the pupil function by using a Gaussian model for the pupil aperture, p. In any case, we have previously verified that the Stiles-Crawford effect is practically negligible in the computed PSF.
The Entropy of the PSF.
The entropy is mathematically calculated as follows:MATH where (x,y) indicates the position (pixels) and log is the decimal logarithm. This metric is a measure of the spatial variance of the PSF, i.e., a measure of how the energy is distributed in the image. The aberration-free PSF has the minimum entropy. Aberrations increase the entropy. 23
The Intensity Variance of the PSF.
The intensity variance of the PSF is calculated as the average value of squared PSF minus the average PSF squared:MATH
In the Fourier Domain
The Volume of the MTF between 0 and 60 cpd.
The MTF is the modulus of the Fourier transform of the PSF (alternatively the autocorrelation of the generalized pupil function):MATH
The Volume of the Contrast Sensitivity Function (CSF).
The CSF is calculated as the product of the optical MTF and the neural contrast sensitivity function, 24 NCSF:MATH
The independent variables for the three-dimensional search were the coefficients b20, b22, and b2−2. The starting point was determined by the criterion of minimum RMS of the WA (i.e., b20 = b22 = b2−2 = 0), not by the subjective refraction. To avoid local maxima we used an exhaustive method of search of the tridimensional space. The search ran in steps of the parameters b20, b22, and b2−2 that corresponded to 0.05 D and 5° for the axis. The extremes of the search intervals were chosen far enough apart to ensure that the maxima identified were global maxima.
The derivation of the sphere and cylinder from the Zernike coefficients that made optimum the image-quality metric will now be described. The WAlens in the Seidel form is MATH where ρ is the normalized radial distance from the axis, θ is the azimuthal angle in the pupil, and θa indicates the axis of the astigmatism. The sphere (S) and cylinder (C), in D, of the correcting lens is MATH where r0 is the radius of the pupil that the WA describes. We have the following relationship between Zernike coefficients and Seidel coefficients:MATH with the axis given by θa=1/2arctan a2−2/a22. The Zernike coefficients for the lens are obtained from the coefficients of the eye’s WA and the coefficients yielded by the search a20, ±2 = b20, ±2 − c20, ±2.
To find out whether a metric performs better than the others, we carried out an experiment consisting of measuring the subjective refractions in a sample of eyes and comparing them with the objective refractions calculated from the wave aberration data. See Fig. 3 for a schematic of the experiment.
Measurements were obtained in six normal young subjects. Subjects were selected after an ophthalmological examination that confirmed no pathology and normal visual acuity. Subjects ranged in age from 23 to 35 years (mean ± SD, 29 ± 4 years). Refractive errors ranged from −1.75 to −4 D for sphere (−3.3 ± 0.8 D), and from 0 to −1 for astigmatism (−0.6 ± 0.4 D). The eye’s pupil was dilated and the accommodation paralyzed by instilling two drops of cyclopentolate 1%, 5 min between drops. The study followed the tenets of the Declaration of Helsinki, and signed informed consent was obtained from the every subject after explanation of the nature and possible consequences of the study.
We measured the wave aberrations with a Shack-Hartmann wavefront sensor 25 (221 lenslets, lenslet spacing = 0.4 mm, and focal length = 24 mm). A 790-nm infrared superluminescent diode serves as a beacon, forming a point source on the retina. Light reflected from the retina emerges through the eye’s pupil as an aberrated wavefront. The eye’s pupil is conjugate with the lenslet array of the sensor, which forms the Shack-Hartmann image. From that image, the wave aberration is determined. An artificial pupil of 6 mm in diameter was used. We measured the aberrations for a 6-mm-diameter pupil to the sixth radial order (28 Zernike coefficients). The error corresponding to axial chromatic aberration from 790 to 570 nm was added to the term of defocus in the wave aberration.
To obtain the subjective refraction, the best vision sphere was first determined with a Snellen chart at 6 ft. Then the astigmatism was measured by means of a fan chart and the fogging method. This first estimation of the refraction was then refined by means of a Snellen chart displayed on a cathode ray tube that the subject viewed through the optics of the experimental system with the 6-mm pupil. The end-point criterion for determining the final subjective refraction consisted of achieving the maximum visual acuity. The spherical and cylindrical power adjustments were made in step sizes of 0.25 D. The subjective refraction was always monocular.
After the subjective refraction, the determined correction was kept in front of the eye by means of spherocylindrical trial lenses. Thus, the measured wave aberration is that of the eye plus that of the correcting lens. If the correct metric is used, the wave aberrations measured in those conditions should directly yield a computed image quality that is optimum or close to the optimum because the eye’s aberrations are measured after the correction of defocus and astigmatism. Any possible aberrations in the trial lenses are taken into account with this experimental procedure.
Fig. 4 shows the average error from the six eyes between the subjective and the objective refraction (in spherical equivalent), for the five image-quality metrics and for the two pupil plane metrics. On average, the error when one uses the methods based on the pupil plane (either, the minimum RMS criterion or the paraxial approximation) was about 0.5 D (range, 0.1 to 0.8 D). By optimizing the image-quality metrics, the error between subjective and objective refraction was 0.1 ± 0.08 D (range, 0 to 0.25 D). This error was lower than the smallest step size (0.25 D) typically used in prescribing spectacles and contact lenses. There was no significance difference between the five image-quality metrics.
For the pupil plane methods, the error estimating the subjective refraction is dependent on the eye’s higher-order aberrations. Fig. 5 shows the error for each eye between the subjective and the objective refraction, estimated by means of the two pupil plane methods, as a function of the RMS of the wave aberration of each eye (without including defocus and astigmatism). The error increases when the amount of higher-order aberrations increases. This is expected for the method based on the paraxial approximation because it directly neglects the effect of higher-order aberrations. But we also see that the minimum RMS is not a good indicator of the best image quality. This is because the method based on the minimum RMS criterion takes into account the higher-order aberrations in the pupil plane, not in the image plane, and the mathematical transformation (based on Fourier transforms) from one plane to the other often produces results that are not intuitive.
Test of the Method in a Large Population of Eyes.
To test further the reliability of the method, we compared the objective refraction calculated by image-quality optimization and the subjective refraction in a large population of subjects. We used the wave aberration data previously measured 12 in a population of 146 eyes from 73 normal subjects. Subjects ranged in age from 24 to 66 years (mean, 45 ± 10). The aberrations were measured with a Shack-Hartmann sensor for a pupil 5.7 mm in diameter to the fifth radial order with natural accommodation. The Shack-Hartmann sensor had a total of 57 lenslets (spacing = 0.6 mm and focal length = 40 mm). Aberrations were measured at 780 nm. The clinical subjective refraction was available for each eye. This refraction was binocular. Refractive errors ranged from −10.75 to 4.25 D for sphere (mean, −3 ± 3 D), and from 0 to −4.25 for astigmatism (mean, −0.5 ± 0.75 D).
Fig. 6 a shows the correlation between subjective spherical equivalent (sphere plus half the cylinder) and the objective spherical equivalent calculated by metric optimization. (The results shown are for the optimization of the Strehl ratio, but there was no significant difference for the other metrics.) The correlation coefficient was close to one (r2 = 0.98). The average error across the 146 eyes between subjective and objective refraction in spherical equivalent was 0.4 ± 0.5 D. Fig. 6 b shows the correlation between the subjective and the objective cylinder (r2 = 0.87). The average error in the estimated cylinder was 0.2 ± 0.3 D. Fig. 6 c shows the same correlation for the axis of the cylinder (r2 = 0.89). The errors for the whole population were <1 D for sphere and <0.5 D for cylinder.
Fig. 7 shows the error between subjective and objective refraction as a function of the eye’s aberrations. Fig. 7 a and c correspond to the estimation of objective refraction by using pupil plane methods, and Fig. 7 b and d correspond to the optimization method using image-quality metrics. When the pupil plane methods are used, both the spherical equivalent and the cylinder show systematic errors that correlate with the higher-order aberrations. With image-quality optimization, there is no correlation between the error in the refraction and the aberrations.
We showed how the amount of sphere and cylinder required to correct vision is different from both the amount that minimizes the RMS of the wave aberration and the total amount in the wave aberration. Thus, methods that use the coefficients of the wave aberration (i.e., based on the pupil plane) fail in predicting the eye’s refraction. The error in the prediction increases when the amount of higher-order aberrations increases, indicating the need to consider the role of higher-order aberrations. Although the minimum wave aberration RMS criterion involves balanced aberrations, we have shown that it is not the best metric for the typical amounts of aberrations in the human eye.
On the other hand, we have proposed an objective method based on the optimization of retinal image quality and demonstrated that it performs well in predicting the subjective refraction of the eye. The experiment performed on six subjects showed that the error between predicted and subjective refraction can be as low as 0.1 D when the subjective and the objective refraction are obtained under identical experimental conditions. In the large population of 146 eyes, we found excellent correlation between subjective and objective refraction, either for sphere, cylinder, or axis. There is, however, some variability with the average error between subjective and objective refraction being 0.4 D for sphere and 0.2 D for cylinder. We attribute these larger errors to several factors. First, it should be noted that the wave aberrations were measured in this large population with natural accommodation and for a 5.7-mm pupil. Although a fixation target was used, small fluctuations of the accommodation may increase the experimental error. It is also likely that subjective refractions were performed for pupil sizes other than 5.7 mm. Another possible factor could be that whereas the WA measurements took into account the possible aberrations of the correcting lenses in the sample of six subjects, this was not the case in the measurements of the aberrations in the large population. Finally, whereas the subjective refraction was monocular and the end-point criterion consisted of reaching the maximum visual acuity (comparable with a maximum in the metric) in the prospective study in six eyes, in the large population, the refraction was binocular and the end-point criterion likely was not the maximum visual acuity but perhaps consisted of reaching the 20/20 line.
Although vision performance ultimately depends on the combination of optical and retinal and neural factors, it seems from our results that the optimization of the retinal image taking into account only the optics of the eye is an accurate method to predict the refractive errors of the eye. In other words, the best objective image quality closely corresponds to the best subjective image quality in terms of refraction. It is important to note that the vision task that we have evaluated is the subjective refraction based on the subjective quality and recognition of letters. However, the present results do not guarantee that retinal image optimization will correlate with the best performance for other visual tasks.
A goal of our study was to find an adequate metric that describes image quality and allows us, by means of its optimization, to predict the subjective refraction. We tested a number of these metrics both in the spatial and in the Fourier domain and found no significant difference between them. It has sometimes been debated whether the Strehl ratio is a good indicator of image quality for the eye. When the Strehl ratio falls to low levels, the extent and the shape of the PSF could become more important than its peak value. 26 In addition, any increases in the Strehl ratio associated with optimization of spatial frequencies in the MTF above 60 cpd would not be relevant for vision because of the limitations of receptoral and neural factors. 27 Finally, the PSF may have several peaks that would make a metric based on the Strehl ratio ambiguous. It happens, however, that for the distribution of aberrations of the human eye, the MTF is very small above 60 cpd, so both the Strehl ratio and the volume of the MTF are robust metrics. Additionally, the similarity of using the volume of the CSF or the volume of the MTF indicates no significant effect of the neural transfer function on the best subjective refraction. With respect to the intensity variance of the PSF, the larger the variance, the sharper the PSF. The entropy of the PSF is a measure of the spatial variance or spread of the PSF. The lower the entropy, the more compact and sharper the PSF. Thus, the correlation between Strehl ratio, variance, and entropy is expected. Although not presented here, we also tested as an image-quality metric the encircled energy that falls within an area corresponding to the Airy disk. The results were similar to those with the other metrics. In sum, we proposed several image-quality metrics, which must be optimized to find the refraction of the subject, and found that all of them correlated well and are equally reliable. It seems likely that additional experiments with greater statistical power will be able to distinguish between these metrics. Additional experiments that involve a wider variety of visual stimuli and an assessment of subjective image quality rather than merely optimizing it may reveal interesting differences between image-quality metrics. Nonetheless, it is of some interest that if differences do exist, they are not large enough to affect this particular task.
As a practical application, wavefront sensors that capture the complete wave aberration of the eye can improve objective refraction by using image plane metrics to incorporate the effect of higher-order aberrations. One of the advantages of refracting the eye from wavefront sensing instead of an autorefractor is that the refraction can be computed for any desired pupil size. Wavefront sensors provide the complete information on the optics of the eye and can tailor the prescription for the patient’s pupil size. The cost in time of performing metric optimization is not large. Commercial computers of about 1 GHz can make the PSF and image metric calculations in 1 to 10 s (depending on the amount of aberrations).
The optimization of an image-quality metric allows not only an estimation of the subjective refraction, but also a simulation of scenes that the subject would see after a best correction of defocus and astigmatism. Of course, there are retinal and neural factors that will ultimately affect the perceptual experience of the observer. However, these computational simulations provide at least an estimation of what will optimize vision as well as a view of how much vision improves when the refractive errors of the eye are corrected. In Fig. 8, there is a series of simulated images in white light after correcting a typical eye with the refraction calculated with the pupil plane methods (Fig. 8 a and b) and with the prescription obtained with the image-quality optimization (Fig. 8 c). The comparison of the three images shows that the optimization of image-plane metrics produces the least blurred image.
This work was supported in part by National Eye Institute, National Institutes of Health grants EY04367 and EY01319. A. Guirao was supported, in part, by a postdoctoral fellowship from Ministerio de Educación y Cultura (Spain).
Part of the results of this manuscript were presented as a paper at the Optical Society of America Annual Meeting, October 2000, Providence, RI, and at a meeting of the Association for Research in Vision and Ophthalmology, May 2001, Fort Lauderdale, FL.
1. Elliott M, Simpson T, Richter D, Fonn D. Repeatability and accuracy of automated refraction: a comparison of the Nikon NRK-8000, the Nidek AR-1000, and subjective refraction. Optom Vis Sci 1997; 74: 434–8.
2. Walline JJ, Kinney KA, Zadnik K, Mutti DO. Repeatability and validity of astigmatism measurements. J Refract Surg 1999; 15: 23–31.
3. Thompson AM, Li T, Peck LB, Howland HC, Counts R, Bobier WR. Accuracy and precision of the Tomey ViVA infrared photorefractor. Optom Vis Sci 1996; 73: 644–52.
4. Bullimore MA, Fusaro RE, Adams CW. The repeatability of automated and clinician refraction. Optom Vis Sci 1998; 75: 617–22.
5. Harvey EM, Miller JM, Dobson V, Tyszko R, Davis AL. Measurement of refractive error in Native American preschoolers: validity and reproducibility of autorefraction. Optom Vis Sci 2000; 77: 140–9.
6. Smirnov MS. Measurement of the wave aberration of the human eye. Biophysics 1962; 7: 766–95.
7. Howland HC, Howland B. A subjective method for the measurement of monochromatic aberrations of the eye. J Opt Soc Am 1977; 67: 1508–18.
8. Walsh G, Charman WN, Howland HC. Objective technique for the determination of monochromatic aberrations of the human eye. J Opt Soc Am (A) 1984; 1: 987–92.
9. Artal P, Marcos S, Navarro R, Williams DR. Odd aberrations and double-pass measurements of retinal image quality. J Opt Soc Am (A) 1995; 12: 195–201.
10. Liang J, Williams DR. Aberrations and retinal image quality of the normal human eye. J Opt Soc Am (A) 1997; 14: 2873–83.
11. He JC, Marcos S, Webb RH, Burns SA. Measurement of the wave-front aberration of the eye by a fast psychophysical procedure. J Opt Soc Am (A) 1998; 15: 2449–56.
12. Porter J, Guirao A, Cox IG, Williams DR. Monochromatic aberrations of the human eye in a large population. J Opt Soc Am (A) 2001; 18: 1793–803.
13. Campbell MC, Bobier WR, Roorda A. Effect of monochromatic aberrations on photorefractive patterns. J Opt Soc Am (A) 1995; 12: 1637–46.
14. Roorda A, Bobier WR. Geometrical technique to determine the influence of monochromatic aberrations on retinoscopy. J Opt Soc Am (A) 1996; 13: 3–11.
15. Joubert L, Harris WF. Excess of autorefraction over subjective refraction: dependence on age. Optom Vis Sci 1997; 74: 439–44.
16. Guirao A, Gonzalez C, Redondo M, Geraghty E, Norrby S, Artal P. Average optical performance of the human eye as a function of age in a normal population. Invest Ophthalmol Vis Sci 1999; 40: 203–13.
17. Artal P, Berrio E, Guirao A, Piers P. Contribution of the cornea and internal surfaces to the change of ocular aberrations with age. J Opt Soc Am (A) 2002; 19: 137–43.
18. Liang J, Williams DR. Aberrations and retinal image quality of the normal human eye. J Opt Soc Am (A) 1997; 14: 2873–83.
19. Mahajan VN. Strehl ratio for primary aberrations: some analytical results for circular and annular pupils. J Opt Soc Am 1982; 72: 1258–66.
20. Thibos LN, Applegate RA, Schwiegerling JT, Webb R. Standards for reporting the optical aberrations of eyes. In: Lakshminarayanan V, ed. Trends in Optics and Photonics. Vision Science and Its Applications, Vol 35. OSA Technical Digest Series. Washington, DC: Optical Society of America, 2000: 232–44.
21. King WB. Dependence of the Strehl ratio on the magnitude of the variance of the wave aberration. J Opt Soc Am 1968; 58: 655–61.
22. Mahajan VN. Aberration Theory Made Simple. Bellingham, WA: SPIE Press, 1991.
23. Barakat R. Some entropic aspects of optical diffraction imagery. Optics Commun 1998; 156: 235–9.
24. Williams DR. Visibility of interference fringes near the resolution limit. J Opt Soc Am (A) 1985; 2: 1087–93.
25. Liang J, Williams DR. Aberrations and retinal image quality of the normal human eye. J Opt Soc Am (A) 1997; 14: 2873–83.
26. Mouroulis P, Zhang HP. Visual instrument image quality metrics and the effects of coma and astigmatism. J Opt Soc Am (A) 1992; 9: 34–42.
27. Williams DR. Aliasing in human foveal vision. Vision Res 1985; 25: 195–205.
Keywords:© 2003 American Academy of Optometry
aberrations; subjective refraction; objective refraction; refractive errors; image-quality metrics; retinal image quality