Enter your Email address:
Wolters Kluwer Health may email you for journal alerts and information, but is committed
to maintaining your privacy and will not share your personal information without
You currently have no recent searches
Melby, Spencer J.; Zierer, Andreas; Lall, Shelly C.; Voeller, Rochus K.; Bailey, Marci S.; Damiano, Ralph J. Jr
Department of Surgery, Division of Cardiothoracic Surgery, Washington University School of Medicine, St. Louis, MO, USA
Bipolar radiofrequency (BPRF) ablation has been used to replace many of the surgical incisions of the Cox-Maze procedure for the treatment of atrial fibrillation. Real-time impedance controlled BPRF reliably produces transmural lesions in both experimental and clinical studies. Other available technologies rely on predetermined time and/or temperature criteria to determine ablation duration. The purpose of this study was to determine the energy required in different atrial structures to create transmural lesions, and to evaluate the ability of a novel impedance algorithm to tailor energy delivery to the particular tissue geometry.
Initial impedance, total energy, temperature, and ablation time were measured in 38 patients undergoing the Cox-Maze procedure using an impedance controlled BPRF device (AtriCure Isolator®; Cincinnati, OH). Lesions were categorized into the following groups: right atrial free wall, left atrial free wall, mitral valve annulus, tricuspid valve annulus, and right or left pulmonary veins. Time, energy delivered, impedance, and maximum temperature were recorded for each ablation.
The table summarizes the data from 654 ablations done in 38 patients. There was a wide range of initial impedance (32.3–760.7 Ohms), and this correlated with total energy delivered (r=−0.31, p<0.001). There was also a significant variation in the total energy delivered (15.4–457.3 joules). Ablation times varied widely (2.0–29.9 sec) and were longer on left atrial structures than right (p<0.005) and shortest near the tricuspid annulus (p<0.005). Mean tissue temperature one mm from the electrode was only 45.7 ± 7.8 °C (range 23.7–69.3 oC).
Bipolar radiofrequency ablation of different atrial structures required widely different amounts of energy and ablation times, likely due to the inhomogeneity of atrial geometry and tissue impedance. Impedance controlled ablation optimizes the delivery of BPRF energy. This reduces ablation time, and Results in minimal lateral spread. This customization of ablation decreases the chance for collateral tissue injury.
© 2006 Lippincott Williams & Wilkins, Inc.
Colleague's E-mail is Invalid
Your Name: (optional)
Separate multiple e-mails with a (;).
Thought you might appreciate this item(s) I saw at Innovations:Technology and Techniques in Cardiothoracic and Vascular Surgery.
Send a copy to your email
Your message has been successfully sent to your colleague.
Some error has occurred while processing your request. Please try after some time.
An Existing Folder
A New Folder
The item(s) has been successfully added to "".
Login with your LWW Journals username and password.
Username or Email:
Enter and submit the email address you registered with. An email with instructions to reset your password will be sent to that address.
Link to reset your password has been sent to specified email address.
What does "Remember me" mean?
By checking this box, you'll stay logged in until you logout. You'll get easier access to your articles, collections,
media, and all your other content, even if you close your browser or shut down your
To protect your most sensitive data and activities (like changing your password),
we'll ask you to re-enter your password when you access these services.
What if I'm on a computer that I share with others?
If you're using a public computer or you share this computer with others, we recommend
that you uncheck the "Remember me" box.
Save my selection