Sphenoid sinus technique
Nasal endoscopy is performed with a 0° endoscope and a straight introducing cannula or ‘sinus guide’ is passed medial to the middle turbinate with its final position just medial to the anterior tip of the superior turbinate. The endoscope is then removed from the nasal cavity and reinserted inferior to the instrument. A lighted guide wire is passed through the sinus guide and advanced until it successfully traverses the sphenoid ostium. With the endoscope positioned below the sinus guide, inadvertent superior or inferior guide wire advancement can be easily detected. With successful access of the sphenoid sinus achieved through the sphenoid ostium, there is a lack of resistance on the guide wire until it begins to curl within the sphenoid sinus. The lighted guide wire transilluminates the sinus and results in a diffuse ‘glow’ of the sphenoid sinus, best seen when the light source to the endoscope is momentarily turned off. Cross-table fluoroscopy can also confirm correct placement of the guide wire for any of the peripheral sinuses.
A balloon dilating catheter is passed over the guide wire under endoscopic visualization until the distal aspect of the balloon just enters the sinus. The balloon is expanded using the inflation device under endoscopic visualization to the recommended pressure for approximately 5 s. The balloon is deflated and withdrawn. With all sinuses, irrigation can be performed and the dilated sinus can be visualized with the appropriately angled endoscope.
Frontal sinus technique
A 0° endoscope is typically used for the procedure, although the internal frontal ostium itself is best visualized with more angled endoscopes. Access to the frontal sinus is best achieved if the tip of the 70° sinus guide is insinuated between the uncinate process and the face of the ethmoid bulla in the parasagittal plane. The distal tip should be visible near the upper third of the ethmoid bulla and not passed into the frontal recess in order to allow the guide wire a broader area to search. The lighted guide wire is passed under endoscopic visualization until it finds entry into the frontal sinus. This is confirmed with the transillumination of the forehead skin with the focal bright light coming from within the sinus. If the guide wire does not pass into the frontal sinus, it is retracted back into the sinus guide, the sinus guide is repositioned, and attempted cannulation is repeated. Once the frontal sinus is successfully accessed, balloon advancement and dilation are performed and may be repeated more proximally or distally within the frontal sinus outflow tract depending on the anatomy and length of balloon chosen.
Maxillary sinus technique
The maxillary sinus can be the most challenging sinus to treat with balloon dilation because the uncinate process remains in place as the maxillary sinus is treated. Under visualization with a 0° or 30° endoscope, a 110° or 90° sinus guide is passed into the middle meatus with the tip pointing inferiorly. It is passed posteriorly and inferiorly within the middle meatus before being rotated laterally to insinuate the tip between the uncinate process and ethmoid bulla. The uncinate process is gently retracted anteriorly using the tip of the sinus guide. The tip of the sinus guide is then again rotated until it is anatomically in line with the position of the natural maxillary sinus ostium. This approximates the 8 o’clock position on the patient's right side and the 4 o’clock on the left side. This correlates with a CT analysis by Sikand [10▪▪], in which the angle of the natural ostium averaged 17.5° from the horizontal plane. It is important to keep the tip of the sinus guide visible, so it does not inadvertently penetrate the posterior or anterior fontanelle.
The lighted guide wire is passed into the sinus; transillumination of the lateral and inferior maxilla confirms the guide wire is within the maxillary sinus. It is important to insure that the tip is not confined to the medial aspect of the maxilla, which could be the case if the tip were incorrectly positioned in a Haller cell or in the submucosal plane. It is important to conceptually note that transillumination simply confirms that the tip of the guide wire is in the maxillary sinus. Transillumination anteriorly and inferiorly can also occur if the guide wire is passed into the maxillary sinus inappropriately via an accessory ostium or any procedurally created antrostomy. This is the importance of endoscopic visualization during positioning of the sinus guide – it must be passed to the correct position and used to retract the uncinate process anteriorly without itself penetrating the medial maxillary sinus wall. Incorrectly placed antrostomies have been described in a cadaver series with resident surgeons [11▪]. However, the technique described above should lead to successful cannulation of the natural ostium. If there is doubt, the uncinate process can be partially removed to confirm proper guide wire and subsequent balloon catheter placement.
The balloon catheter is passed over the guide wire fully into the sinus, then partially withdrawn so that it spans the entire outflow tract of the maxillary sinus. This is achieved by rotating and repositioning the entire unit (sinus guide and balloon catheter) medially and posteriorly. Dilation is performed as detailed above.
All of the above-mentioned techniques employ a transnasal endoscopic approach; Entellus manufactures an additional system that utilizes a sublabial, transantral technique. The procedure begins with a sublabial approach under local anesthesia by making an incision within the canine fossa. A trocar with a cutting tip penetrates the anterior maxillary wall to access the antrum. Through the trocar, a small flexible or rigid endoscope can be passed into the maxillary sinus for visualization. A balloon dilating catheter is passed through the trocar, into the maxillary sinus, across the maxillary ostium, and into the ethmoid infundibulum. The balloon is expanded, deflated, and the system is removed. Irrigation is performed as indicated.
As with any new device, safety and feasibility are initial concerns. Patient tolerability is an additional concern specific to the consideration of in-office surgery. BCD tools have been proven from the beginning to be well tolerated and effective in dilating peripheral sinus outflow tracts in patients undergoing surgery in the operating room [4–7,8▪,9]. Complication rates associated with BCD compare favorably with those reported in the literature for traditional FESS [4–7,8▪,9,12–16,17▪▪–19▪▪] (Table 1).
More recent studies have also shown that similar safety, feasibility, tolerability, and technical success can be achieved in the office setting with these tools utilized under local anesthesia. Albritton et al. [17▪▪] reported on 37 patients, in whom a total of 59 sinuses were dilated in-office for medically refractory CRS. There was a single nonserious adverse event reported when a patient inadvertently swallowed a pledget used for local topical anesthesia. Ninety-one percent of attempted sinus dilations were successfully achieved. Only 3.4% of patients felt the procedure was intolerable and only 5.6% of patients reported intense pain during the procedure, specifically associated with balloon inflation. In a multicenter study, Karanfilov et al. [18▪▪] reported 552 sinus dilations in 203 patients (mean 2.7 sinuses/patient) again with only a single adverse event reported. One patient experienced periorbital swelling that resolved without sequelae within 1 week after the procedure. Technical success in dilating sinuses was achieved in 93.3% of maxillary, 90.5% of sphenoid, and 93.7% of frontal sinuses. Only 3% found the procedure intolerable with local anesthesia and nearly 70% resumed normal activities within 48 h after the procedure. Finally, Cutler et al. reported on 71 patients who underwent transantral dilation of the maxillary and ethmoid infundibulum, 19 of whom underwent the procedure in the office setting under local anesthesia. They achieved 100% technical success with only one adverse event. A patient developed subcutaneous emphysema after using continuous positive airway pressure following the procedure. This resolved spontaneously without consequence. Procedures were well tolerated with a patient reported mean pain score of 2.7/10 and 88% resumed normal activities within 48 h [17▪▪–19▪▪]. The majority of patients report that the most significant pain during BCD is associated with the inflation of the balloon. Educating patients in advance and communicating with them throughout the procedure may help minimize this. A summary of pain and tolerability is seen in Table 2.
In each of these studies, anesthesia was not standardized and was based on physician preference. In addition, each of the above three studies reported statistically significant and clinically meaningful improvement in quality-of-life measures similar to prior studies performed in an operating room setting. This suggests that outcomes are not dependent on the site of service but depend on the appropriate application of the technology to carefully selected patients with CRS.
Although there are numerous reports of safe and successful outcomes related to BCD technology utilized in CRS patients, the large majority of them are not randomized and blinded trials and therefore fall short of the ideal clinical design. Batra et al. and Burton et al. have criticized the current data largely based on study design and the lack of randomized, prospective trials [21,22]. Plaza et al. [20▪] published a randomized, blinded, prospective study reporting an increase in frontal sinus outflow tract patency in patients in whom BCD was used to treat the frontal sinus in a ‘hybrid’ fashion compared to traditional FESS techniques.
In the current healthcare environment, cost considerations become paramount. Physicians and patients alike seek new avenues for cost reduction and containment, however, not at the expense of sacrificing outcomes. Prickett et al. [23▪▪] presented a cost analysis of office-based vs. operating room rhinologic procedures in 29 matched pairs. Current Procedural Terminology (CPT) codes included in the comparison were 31231, 31237, 31254, 31255, 31256, 31267, 31276, and 31287. Using direct total charges, allowed charges, and scaled charges, they found that office procedures could be performed for $3300–$5000 less than matched procedures performed in the operating room. There is still opportunity for significant cost savings when performing BCD in-office even when factoring in the cost of the disposable tools. The authors also reported that all 29 in-office procedures, which included primary and revision surgery, were performed without complications.
In-office surgery has numerous potentially attractive features – avoidance of general anesthesia, rapid recovery and return to normal activities, and significant potential cost savings. However, if procedures in this setting carry higher risk, are unacceptably painful, and are less effective, then offering this option to patients would seem unethical. From the current body of evidence, it appears that in-office FESS utilizing balloon sinus dilation tools is safe, well tolerated, and with outcomes similar to those achieved in an operating room setting.
Conflicts of interest
Disclosures: C.T. Melroy: Acclarent, Inc. – Consultant and speaker; Olympus – Consultant.
Michael J. Sillers: Acclarent – Scientific Advisory Board; GE Navigation – Scientific Advisory Board; Xoran Technologies – Speakers Bureau.
REFERENCES AND RECOMMENDED READING
Papers of particular interest, published within the annual period of review, have been highlighted as:
- ▪ of special interest
- ▪▪ of outstanding interest
Additional references related to this topic can also be found in the Current World Literature section in this issue (pp. 87–88).
1. Pleis J, Lucas J. Summary health statistics for U.S. adults: National Health Interview Survey, 2007. Vital Health Stat 2009; 240:1–159.
2. Gliklich R, Metson R. The health impact of chronic sinusitis in patients seeking otolaryngologic care. Otolaryngol Head Neck Surg 1995; 113:104–109.
3. Smith T, Kern R, Palmer J, et al. Medial therapy vs. surgery for chronic rhinosinusitis: a prospective, multiinstitutional study. Int Forum Allergy Rhinol 2011; 1:235–241.
4. Bolger W, Brown C, Church C, et al. Safety and outcomes of balloon catheter sinusotomy: a multicenter 24 week analysis in 115 patients. Otolaryngol Head Neck Surg 2007; 137:10–20.
5. Levine H, Sertich A II, Hoisington D, et al. A multicenter registry of balloon catheter sinusotomy outcomes for 1036 patients. Ann Otol Rhinol Laryngol 2008; 117:263–270.
6. Weiss R, Church C, Kuhn F, et al. Long-term outcomes analysis of balloon catheter sinusotomy: two year follow-up. Otolaryngol Head Neck Surg 2008; 139:S38–S46.
7. Stankiewicz J, Truitt T, Atkins J. One-year results: transantral balloon dilation of the ethmoid infundibulum. Ear Nose Throat 2010; 89:72–77.
8▪. Stankiewicz J, Truitt T, Winegar B, et al. Two year results: antral dilation of the ethmoid infundibulum. Int Forum Allergy Rhinol 2012; 2:199–206.
This confirms the continued durability of symptom improvement and work productivity previously noted at 1 year.
9. Stankiewicz J, Tami T, Truitt T, et al. Impact of chronic rhinosinusitis on work productivity through one-year follow-up after balloon dilation of the ethmoid infundibulum. Int Forum Allergy Rhinol 2011; 1:38–45.
10▪▪. Sikand A. Computed tomography-based exploration of the infundibular anatomy for maxillary sinus balloon dilation. Ann Otol Rhinol Laryngol 2011; 120:656–662.
The angle with which the sinus guide is positioned and wire trajectory are important for successful dilation of the natural maxillary sinus ostium. A CT basis for recommendations is presented.
11▪. Brenner P, Abadie W, Weitzel E, et al. Unexpected consequences of transantral balloon dilation of the maxillary ostium. Int Forum Allergy Rhinol 2011; 1:466–470.
False passages were created in all maxillary sinuses but procedures were performed with resident physicians. Operator experience, detailed technique, and level of supervision were not outlined.
12. May M, Levine H, Mester S, Shaitkin B. Complications of endoscopic sinus surgery: analysis of 2108 patient-incidence and prevention. Laryngoscope 1994; 104:1080–1083.
13. Rombout J, de Vries N. Complications in sinus surgery and new classification proposal. Am J Rhinol 2001; 15:363–370.
14. Kutluhan A, Bozdemir K, Cetin H, et al. Endoscopic balloon dilation sinuplasty including ethmoidal air cells in chronic rhinosinusitis. Ann Otol Rhinol Laryngol 2009; 118:881–886.
15. Ramadan H, McLaughlin K, Josephson G, et al. Balloon catheter sinuplasty in young children. Am J Rhinol Allergy 2010; 24:54–56.
16. Stankiewicz J, Lal D, Conner M, Welch K. Complications in endoscopic sinus surgery for chronic rhinosinusitis: a 25-year experience. Laryngoscope 2011; 121:2684–2701.
17▪▪. Albritton F, Casiano R, Sillers M. Fesability of in-office endoscopic sinus surgery with balloon sinus dilation. Am J Rhinol Allergy 2012; 26:243–248.
This article reports the safety, technical success, and clinically significant improvement in patients with CRS undergoing frontal, maxillary, and sphenoid sinus surgery with BCD tools. The technical success in accessing the target sinus was greater than 90%.
18▪▪. Karanfilov B, Silvers S, Pasha R, et al.
Office-based balloon sinus dilation: a prospective multicenter study of 203 patients. (in press).
This is a large, multicenter, prospective study further reporting the safety, tolerability, and clinically significant improvement in QOL in patients with CRS undergoing in-office surgery with BCD tools under local anesthesia. It is a ‘real world’ experience using varied anesthetic techniques with similar patient acceptance. These results should be reproducible.
19▪▪. Cutler J, Truitt T, Atkins J, et al. First clinic experience: patient selection and outcomes for ostial dilation for chronic rhinosinusitis. Int Forum Allergy Rhinol 2011; 1:460–465.
Patients can be successfully treated by transantral dilation with high patient acceptance. A total of 88% of patients return to normal activities within 48 h of the procedure.
20▪. Plaza G, Eisenberg G, Montoio J, et al. Balloon dilation of the frontal recess: a randomized clinical trial. Ann Otol Rhinol Laryngol 2011; 120:511–518.
This is the first, randomized, blinded, prospective study utilizing BCD tools. Frontal recess patency is durable and may be superior to patency seen with traditional FESS.
21. Batra P, Ryan M, Sindawi, Marple B. Balloon catheter technology in rhinology: reviewing the evidence. Laryngoscope 2011; 121:226–232.
22. Burton M, Bhattacharyya N, Rosenfeld R. Extracts from the Cochrane Library: functional endoscopic balloon dilation of sinus ostia for chronic rhinosinusitis. Otolaryngol Head Neck Surg 2011; 145:371–374.
23▪▪. Prickett K, Wise S, DelGaudio J. Cost analysis of office-based and operating room procedures in rhinology. Int Forum Allergy Rhinol 2012; 2:207–211.
Keywords:© 2013 Lippincott Williams & Wilkins, Inc.
balloon catheter dilation; in-office surgery; patient tolerability; safety; sphenoid, frontal, and maxillary sinuses