Endotracheal Intubation in Patients with Cervical Spine Immobilization: A Comparison of Macintosh and Airtraq Laryngoscopes
Maharaj, Chrisen H. B.Sc., M.B., F.C.A.R.C.S.I., D.P.M.*; Buckley, Elma M.B., F.C.A.R.C.S.I.†; Harte, Brian H. M.B., F.F.A.R.C.S.I.‡; Laffey, John G. M.D., M.A., B.Sc., F.F.A.R.C.S.I.§
Background: The Airtraq laryngoscope (Prodol Ltd., Vizcaya, Spain) is a novel single-use tracheal intubation device. The authors compared ease of intubation with the Airtraq and Macintosh laryngoscopes in patients with cervical spine immobilization in a randomized, controlled clinical trial.
Methods: Forty consenting patients presenting for surgery requiring tracheal intubation were randomly assigned to undergo intubation using a Macintosh (n = 20) or Airtraq (n = 20) laryngoscope. All patients were intubated by one of four anesthesiologists experienced in the use of both laryngoscopes.
Results: No significant differences in demographic or airway variables were observed between the groups. All but one patient, in the Macintosh group, were successfully intubated on the first attempt. The Airtraq reduced the duration of intubation attempts (mean ± SD: 13.2 ± 5.5 vs. 20.3 ± 12.2 s), the need for additional maneuvers, and the intubation difficulty scale score (0.1 ± 0.5 vs. 2.7 ± 2.5). Tracheal intubation with the Airtraq caused fewer alterations in blood pressure and heart rate.
Conclusions: These findings demonstrate the utility of the Airtraq laryngoscope for tracheal intubation in patients with cervical spine immobilization.
FAILURE to adequately immobilize the neck during tracheal intubation in patients with cervical spine injuries can result in a devastating neurologic outcome.1
A widely used approach to neck immobilization during tracheal intubation is manual in-line axial stabilization (MIAS).2
Although the evidence base supporting the use of MIAS is limited, this approach has been demonstrated to reduce cervical spine mobility in anatomical studies,3
and did not result in additional neurologic injury in a case series of oral tracheal intubation in cervical spine–injured adults.4
However, a key concern is the fact that, with cervical spine immobilization, it is more difficult to visualize the larynx using conventional laryngoscopy.5–7
Failure to successfully intubate the trachea and secure the airway remains a leading cause of morbidity and mortality, in the operative8–10
and emergency settings.11,12
The Airtraq (Prodol Ltd., Vizcaya, Spain) is a new indirect laryngoscope that has been developed to facilitate tracheal intubation in patients with normal or difficult airways (fig. 1
). As a result of the exaggerated curvature of the blade and an internal arrangement of optical components, a view of the glottis is provided with minimal need for airway optimization maneuvers. The blade of the Airtraq consists of two side-by-side channels. One channel acts as a conduit through which an endotracheal tube (ETT) can be passed, whereas the other channel contains a series of lenses, prisms, and mirrors that transfers the image from the illuminated tip to a proximal viewfinder. A high-quality wide-angle view of the glottis and surrounding structures and the tip of the ETT is provided. The ETT is clearly visualized passing through the vocal cords unlike, for example, the lighted stylet, fiberoptic bronchoscope, or LMA CTrach
™ (The Laryngeal Mask Company, Singapore).13
We have previously demonstrated that the Airtraq performs comparably to the Macintosh laryngoscope when used by experienced anesthesiologists in easy laryngoscopy scenarios in the manikin.14
Of interest, the Airtraq seems to have advantages compared with the Macintosh laryngoscope in simulated difficult intubation scenarios, including reduced cervical spine mobility, in the manikin14
and when used by both inexperienced15
users. More recently, we have demonstrated that the Airtraq seems to reduce intubation difficulty in patients at low risk for difficult laryngoscopy.17
The purpose of this study was to evaluate the usefulness of this new device for use by experienced anesthesiologists in patients with neck immobilization using MIAS. We hypothesized that, in comparison with the Macintosh, the Airtraq would be associated with lower laryngoscopy times, lower intubation difficulty scale (IDS) scores, and less hemodynamic stimulation after intubation.
Materials and Methods
After obtaining approval by the Galway University Hospitals Research Ethics Committee (Galway, Ireland) and written informed patient consent, we studied 40 patients with American Society of Anesthesiologists physical status I–III, aged 18 yr or older, scheduled to undergo surgical procedures necessitating tracheal intubation, in a randomized, single-blind, controlled clinical trial. Patients were excluded if risk factors for gastric aspiration and/or difficult intubation (Mallampati class III or IV, thyromental distance less than 6 cm, interincisor distance less than 4.0 cm) were present, or where there was a history of relevant drug allergy. All data were collected by an independent unblinded observer.
Patients were randomly assigned to two groups by sealed envelopes, and patients were blinded to their group assignment. All patients received a standardized general anesthetic. Standard monitoring, including electrocardiography, noninvasive blood pressure, oxygen saturation measured by pulse oximetry, end-tidal carbon dioxide, and volatile anesthetic levels, were used in all patients. Before induction of anesthesia, all patients were given fentanyl (1–1.5 μg/kg) intravenously. A sleep dose of propofol (2–3 mg/kg) was titrated to induce anesthesia. After induction of anesthesia, all patients were manually ventilated with sevoflurane (2.0–2.5%) in oxygen and nitrous oxide, and atracurium (0.35 mg/kg) was administered. After the onset of neuromuscular blockade, the pillow was removed, and the neck was immobilized using MIAS applied by an experienced individual holding the sides of the neck and the mastoid processes, thus preventing flexion/extension or rotational movement of the head and neck.
The trachea was then intubated by one of four anesthesiologists (C.H.M., E.B., B.H.H., J.G.L.) experienced in the use of both laryngoscopes. In patients randomly assigned to undergo intubation with the Airtraq, the blade was inserted into the mouth in the midline, over the center of the tongue (fig. 2A
). After the device was passed over the back of the tongue, the view from the viewfinder was used to position the tip in the vallecula (fig. 2B
). The view of the glottis could then be optimized by lifting the epiglottis by elevating the blade into the vallecula (fig. 2C
). When the view of the glottis had been optimized, the ETT was passed through the vocal cords and held in place, and the device was removed.
Thereafter, in all patients, the lungs were mechanically ventilated for the duration of the procedure, and anesthesia was maintained with sevoflurane (1.25–1.75%) in a mixture of nitrous and oxygen in a 2:1 ratio. No other medications were administered and no other procedures were performed during the 5-min data collection period after tracheal intubation. Further management was left to the discretion of the anesthesiologist providing care for the patient.
The primary endpoints were the duration of the tracheal intubation procedure and the IDS score.18
The duration of the intubation attempt was defined as the time taken from insertion of the blade between the teeth until the ETT was placed through the vocal cords, as evidenced by visual confirmation by the anesthesiologist. However, in patients in whom the ETT was not directly visualized passing through the vocal cords, the intubation attempt was not considered complete until the ETT was connected to the anesthetic circuit and evidence was obtained of the presence of carbon dioxide in the exhaled breath. The IDS score, developed by Adnet et al.
is a quantitative scale of multiple indices of intubation difficulty that can more objectively compare the complexity of tracheal intubations (appendix
A secondary endpoint was the rate of successful placement of the ETT in the trachea. A failed intubation attempt was defined as an attempt in which the trachea was not intubated or an attempt that required more than 120 s to perform. Additional endpoints included the number of intubation attempts and the number of optimization maneuvers required (use of a bougie, BURP, second assistant) to aid tracheal intubation, and the Cormack and Lehane grade at laryngoscopy.19
We based our sample size estimation on the IDS score. An IDS score of 0 represents ideal intubating conditions, and increasing scores represent progressively more difficult intubating conditions. Based on initial pilot studies, we projected an IDS score of 1 or greater in 70% patients with the Macintosh laryngoscope. We considered that a clinically important reduction in the number of patients with an IDS score greater than 0 in these low-risk patients would be a 30% reduction, i.e., an IDS score of 1 or greater in 40% of patients. Based on these figures, using α = 0.05 and β = 0.2, for an experimental design incorporating two equally sized groups, we estimated that 20 patients would be required per group. We therefore aimed to enroll 20 patients per group.
Data for duration of intubation attempts were analyzed using the t test. Data for the success of tracheal intubation attempts were analyzed using the Fisher exact test. Data for the IDS score, the number of intubation attempts, and the number of optimization maneuvers were analyzed using the Mann–Whitney rank sum test. The comparisons of hemodynamic data within groups were analyzed using one-way repeated-measures analysis of variance. For these analyses, the preintubation data were chosen as baseline data, rather than the preinduction values. Between-group comparisons were made using an unpaired t test. Continuous data are presented as mean ± SD, ordinal data are presented as median (interquartile range), and categorical data are presented as number and frequency. The α level for all analyses was set as P < 0.05.
A total of 40 patients were entered into the study. Twenty patients were randomly assigned to undergo tracheal intubation with the Macintosh laryngoscope, whereas 20 underwent tracheal intubation with the Airtraq laryngoscope. There were no significant differences in demographic or baseline airway parameters between the groups (table 1
). There were no between-group differences with regard to anesthetic management, with similar mean doses of propofol and fentanyl, and end-tidal sevoflurane after tracheal intubation in both groups (table 2
All patients were successfully intubated on the first attempt with the Airtraq laryngoscope, whereas three attempts were required in one patient with the Macintosh laryngoscope (table 3
). The Airtraq significantly reduced median IDS score (table 3
), and improved Cormack and Lehane glottic view obtained at laryngoscopy (fig. 4
), compared with the Macintosh group. Fourteen patients in the Macintosh group had an IDS score of 1 or greater, compared with one in the Airtraq group. In the Macintosh group, four patients had an IDS score of 5 or greater, indicating moderate to severe intubation difficulty (fig. 3
). The duration of intubation attempts were significantly shorter with the Airtraq (table 3
). Fewer maneuvers were required in the Airtraq group to improve the glottic exposure, compared with the Macintosh group (table 3
). There was no between-group difference in the incidence of complications, with two patients experiencing bruising to the lip with the Macintosh, compared with no injuries seen with the Airtraq, or in the lowest oxygen saturation during intubation attempts (table 3
). There was no incidence of dental or other airway trauma with either laryngoscope.
Tracheal intubation with the Macintosh resulted in a significant increase in heart rate and mean arterial blood pressure, compared with preintubation values, in contrast to the Airtraq. There were significant between-group differences in heart rate and mean blood pressure at all time points after intubation (figs. 5 and 6
Although the frequency of airway management aggravating preexisting cervical spine injury is unclear, it is likely to be relatively low.4
However, cervical spine injury resulting in quadriplegia has been previously reported in association with airway management in a patient in whom cervical spine immobilization was not performed.1
The evidence base supporting the use of MIAS is surprisingly limited. In anatomical studies, after complete C4–C5 ligamentous injury, MIAS did reduce segmental angular rotation and distraction, although it did increase subluxation, compared with nonimmobilization.3
In a case series of 150 patients with traumatic cervical spine injuries with well-preserved neurologic function, oral tracheal intubation with MIAS, whether performed after induction of general anesthesia or with the patient awake, did not result in any neurologic complications.4
It is therefore an accepted practice in many institutions to remove rigid collars and use MIAS for cervical immobilization during tracheal intubation in patients with suspected or proven cervical spine injury.
A key concern remains the fact that glottic views obtained during direct laryngoscopy with cervical spine immobilization are consistently poorer, compared with nonimmobilized controls.5,6
MIAS prevents head extension and neck flexion, which are necessary for optimal alignment of the three airway axes and exposure of the vocal cords using direct laryngoscopic techniques. Avoidance of use of a pillow further limits optimization of intubating position. Alternative approaches to neck immobilization, such as the use of a rigid collar, tape, and sandbags, may result in an increased incidence of grade 3 and 4 laryngoscopic views (up to 64%) with conventional laryngoscopy owing to the combination of decreased interincisor distance and cervical spine immobility.7
The adequacy of the laryngeal view obtained is a major factor in determining the difficulty of intubation.6
Consequently, maneuvers to stabilize the neck in patients at risk of cervical spinal injury may result in failure to secure the airway, which may result in substantial morbidity and even mortality in this patient group. These issues highlight the need to develop alternative approaches to securing the airway in patients at risk of cervical spine injury.
Our group has previously already assessed performance of the Airtraq in manikins when used by anesthesiologists,14
relatively inexperienced medical personnel,15
and novice users16
and in patients at low risk for difficult intubation.17
These studies demonstrated potential advantages, in both easy and simulated difficult laryngoscopy scenarios. Specifically, we have demonstrated improved intubating conditions in a patient simulator with reduced cervical range of movement.14
We therefore wished to compare the utility of the Airtraq with that of the Macintosh laryngoscope in patients after cervical spine immobilization, in this randomized, controlled clinical trial. Previous studies have used this model of simulated cervical spine to assess optimal intubation technique in similar patient populations.20,21
All intubations in this study were performed by one of four experienced anesthesiologists in patients deemed on clinical assessment to be at low risk for difficult tracheal intubation. Each anesthesiologist had performed more than 500 intubations with the Macintosh laryngoscope and at least 50 intubations with the Airtraq in manikins, and 50 intubations with the Airtraq in patients, before this study.
Our study demonstrated that, in comparison with the Macintosh laryngoscope, the Airtraq provided superior intubating conditions in patients undergoing cervical immobilization with MIAS. The Airtraq resulted in reduced duration of intubation attempts and reduced IDS scores. The Cormack and Lehane grading system, although originally designed to compare glottic views at direct laryngoscopy,19
provided a useful comparison of the direct and indirect laryngoscopic views achieved in this study. Nineteen patients intubated with the Airtraq had a grade I Cormack and Lehane glottic view, compared with six patients in the Macintosh group. Fewer patients required additional maneuvers to improve glottic exposure with the Airtraq device. However, the limitations of this latter measurement are acknowledged.
The Airtraq resulted in less stimulation of heart rate and blood pressure after tracheal intubation in comparison with the Macintosh laryngoscope. This finding probably reflects the fact that the Airtraq provides a view of the glottis without a need to align the oral, pharyngeal, and tracheal axes, and therefore requires less force to be applied during laryngoscopy. The hemodynamic findings for direct laryngoscopy in our study were similar to those described previously.17,22–24
The potential of the Airtraq to produce less stimulation of heart rate may be particularly advantageous in clinical situations such as coronary artery disease or arrhythmias.
An important potential advantage of the Airtraq is that it is a single-use device. This removes concerns regarding the potential for multiuse intubation devices to facilitate transmission of prions, which are thought to be responsible for causing variant Creutzfeldt–Jakob disease.25,26
These concerns arise from the difficulties in ensuring that all proteinaceous material has been removed from reusable laryngoscope blades during cleaning and sterilization.27,28
In recognition of these concerns, the guidelines of the Association of Anesthetists of Great Britain and Ireland state that “single use intubation aids” should be used where possible.29
The relatively low cost of this device in comparison with reusable indirect laryngoscopes allows for this device to be provided at multiple locations within the hospital in which a difficult intubation may be encountered, such as in the emergency room and the obstetric, radiologic, and ambulatory surgical suites. Our study demonstrates that the Airtraq is at least as effective as the reusable Macintosh bladed laryngoscope, attesting to its safety in this regard. Potential disadvantages of disposable device such as the Airtraq do exist, including increased cost, the potential for breakage, and issues surrounding the quality of image obtained. However, in our experience with the Airtraq, we have not encountered problems with breakage or with image quality to date.
A number of important limitations exist regarding this study. In particular, we acknowledge that the potential for bias exists, because it is impossible to blind the anesthesiologist to the device being used. Furthermore, certain measurements used in this study, such as laryngoscopic grading, are by their nature subjective. However, there was good agreement between subjective indices of difficulty of intubation and more objective measures, such as the IDS score.18
In addition, the possibility remains that factors other than the difficulty of laryngoscopy may have contributed to our hemodynamic findings. However, these findings are not explained by between-group differences in the anesthetic regimen, because there were no differences in the anesthetic agents used in each group. Furthermore, our hemodynamic data for intubation with the Macintosh laryngoscope are consistent with previously reported data,17,22–24
whereas the time required for tracheal intubation with the Macintosh laryngoscope was shorter than previously described. This study was conducted by experienced users of each device. The results may differ when the procedures are performed by less experienced users. Finally, the Airtraq was compared with the Macintosh laryngoscope in this study, because this remains the accepted standard. Similar results to that seen with the Airtraq have been demonstrated with other indirect laryngoscopes, such as the Bullard (Circon Corporation, Stamford, CT),30
GlideScope (Saturn Biomedical Systems, Inc., Burnaby, BC, Canada),31
and WuScope (Pentax Precision Instruments, Orangeburg, NY).6
Further comparative studies are needed to determine the relative efficacies of these devices.
In conclusion, the Airtraq laryngoscope offers a new approach to tracheal intubation of patients who require cervical spine immobilization. The Airtraq reduced the difficulty of tracheal intubation and the degree of hemodynamic stimulation compared with the Macintosh laryngoscope in these patients. These findings demonstrate the efficacy of the Airtraq in a clinically relevant context and add to the evolving body of knowledge regarding this potentially useful device. Further studies in the clinical context, particularly in predicted difficult intubation scenarios, are necessary to confirm and extend these positive findings.
Appendix: Intubation Difficulty Scale Score The IDS score∥ is the sum of the following seven variables:
N1: Number of intubation attempts greater than 1
N2: Number of operators greater than 1
N3: Number of alternative intubation techniques used
N4: Glottic exposure (Cormack and Lehane grade minus 1)
N5: Lifting force required during laryngoscopy (0 = normal; 1 = increased)
N6: Necessity for external laryngeal pressure (0 = not applied; 1 = applied)
N7: Position of the vocal cords at intubation (0 = abduction/not visualized; 1 = adduction)
1. Hastings RH, Kelley SD: Neurologic deterioration associated with airway management in a cervical spine-injured patient. Anesthesiology 1993; 78:580–3
2. Stene JK, Grande CM: General anesthesia: Management considerations in the trauma patient. Crit Care Clin 1990; 6:73–84
3. Lennarson PJ, Smith DW, Sawin PD, Todd MM, Sato Y, Traynelis VC: Cervical spinal motion during intubation: Efficacy of stabilization maneuvers in the setting of complete segmental instability. J Neurosurg 2001; 94:265–70
4. Suderman VS, Crosby ET, Lui A: Elective oral tracheal intubation in cervical spine-injured adults. Can J Anaesth 1991; 38:785–9
5. Nolan JP, Wilson ME: Orotracheal intubation in patients with potential cervical spine injuries: An indication for the gum elastic bougie. Anaesthesia 1993; 48:630–3
6. Smith CE, Pinchak AB, Sidhu TS, Radesic BP, Pinchak AC, Hagen JF: Evaluation of tracheal intubation difficulty in patients with cervical spine immobilization: Fiberoptic (WuScope) versus conventional laryngoscopy. Anesthesiology 1999; 91:1253–9
7. Heath KJ: The effect of laryngoscopy of different cervical spine immobilisation techniques. Anaesthesia 1994; 49:843–5
8. Caplan RA, Posner KL, Ward RJ, Cheney FW: Adverse respiratory events in anesthesia: A closed claims analysis. Anesthesiology 1990; 72:828–33
9. Cheney FW: The American Society of Anesthesiologists Closed Claims Project: What have we learned, how has it affected practice, and how will it affect practice in the future? Anesthesiology 1999; 91:552–6
10. Peterson GN, Domino KB, Caplan RA, Posner KL, Lee LA, Cheney FW: Management of the difficult airway: A closed claims analysis. Anesthesiology 2005; 103:33–9
11. Mort TC: Esophageal intubation with indirect clinical tests during emergency tracheal intubation: A report on patient morbidity. J Clin Anesth 2005; 17:255–62
12. Mort TC: Emergency tracheal intubation: Complications associated with repeated laryngoscopic attempts. Anesth Analg 2004; 99:607–13
13. Liu EH, Goy RW, Chen FG: The LMA CTrach, a new laryngeal mask airway for endotracheal intubation under vision: Evaluation in 100 patients. Br J Anaesth 2006; 96:396–400
14. Maharaj CH, Higgins B, Harte BH, Laffey JG: Evaluation of ease of intubation with the Airtraq® or Macintosh laryngoscope by anaesthetists in easy and simulated difficult laryngoscopy: A manikin study. Anaesthesia 2006; 61:469–77
15. Maharaj CH, Ni Chonghaile M, Higgins B, Harte BH, Laffey JG: Tracheal intubation by inexperienced medical residents using the Airtraq® and Macintosh laryngoscope: A manikin study. Am J Emerg Med 2006; 24:769–74
16. Maharaj CH, Costello J, Higgins B, Harte BH, Laffey JG: Learning and performance of tracheal intubation by novice personnel: A comparison of the Airtraq® and Macintosh laryngoscope. Anaesthesia 2006; 61:671–7
17. Maharaj CH, O'Croinin D, Curley G, Harte BH, Laffey JG: A comparison of tracheal intubation using the Airtraq or the Macintosh laryngoscope in routine airway management: A randomised, controlled clinical trial. Anaesthesia 2006; 61:1093–9
18. Adnet F, Borron SW, Racine SX, Clemessy JL, Fournier JL, Plaisance P, Lapandry C: The intubation difficulty scale (IDS): Proposal and evaluation of a new score characterizing the complexity of endotracheal intubation. Anesthesiology 1997; 87:1290–7
19. Cormack RS, Lehane J: Difficult tracheal intubation in obstetrics. Anaesthesia 1984; 39:1105–11
20. Bilgin H, Bozkurt M: Tracheal intubation using the ILMA, C-Trach or McCoy laryngoscope in patients with simulated cervical spine injury. Anaesthesia 2006; 61:685–91
21. Kihara S, Yaguchi Y, Taguchi N, Brimacombe JR, Watanabe S: The StyletScope is a better intubation tool than a conventional stylet during simulated cervical spine immobilization. Can J Anaesth 2005; 52:105–10
22. Tan PH, Yang LC, Shih HC, Lin CR, Lan KC, Chen CS: Combined use of esmolol and nicardipine to blunt the haemodynamic changes following laryngoscopy and tracheal intubation. Anaesthesia 2002; 57:1207–12
23. Nathan N, Vandroux D, Benrhaiem M, Marquet P, Preux PM, Feiss P: Low alfentanil target-concentrations improve hemodynamic and intubating conditions during induction with sevoflurane. Can J Anaesth 2004; 51:382–7
24. Montes FR, Giraldo JC, Betancur LA, Rincon JD, Rincon IE, Vanegas MV, Charris H: Endotracheal intubation with a lightwand or a laryngoscope results in similar hemodynamic variations in patients with coronary artery disease. Can J Anaesth 2003; 50:824–8
25. Lowe PR, Engelhardt T: Prion-related diseases and anaesthesia (letter). Anaesthesia 2001; 56:485
26. Will RG, Ironside JW, Zeidler M, Cousens SN, Estibeiro K, Alperovitch A, Poser S, Pocchiari M, Hofman A, Smith PG: A new variant of Creutzfeldt-Jakob disease in the UK. Lancet 1996; 347:921–5
27. Miller DM, Youkhana I, Karunaratne WU, Pearce A: Presence of protein deposits on “cleaned” re-usable anaesthetic equipment. Anaesthesia 2001; 56:1069–72
28. Hirsch N, Beckett A, Collinge J, Scaravilli F, Tabrizi S, Berry S: Lymphocyte contamination of laryngoscope blades: A possible vector for transmission of variant Creutzfeldt-Jakob disease. Anaesthesia 2005; 60:664–7
29. Infection Control in Anaesthesia. London, The Association of Anaesthetists of Great Britain and Ireland, 2002
30. Wackett A, Anderson K, Thode H: Bullard laryngoscopy by naive operators in the cervical spine immobilized patient. J Emerg Med 2005; 29:253–7
31. Agro F, Barzoi G, Montecchia F: Tracheal intubation using a Macintosh laryngoscope or a GlideScope in 15 patients with cervical spine immobilization. Br J Anaesth 2003; 90:705–6
∥ Intubation difficulty scale score from Adnet et al.18
; reproduced with permission. Cited Here...
This article has been cited 38 time(s).
Bmc AnesthesiologyComparison of the glidescope, CMAC, storz DCI with the Macintosh laryngoscope during simulated difficult laryngoscopy: a manikin studyBmc Anesthesiology
Bmc AnesthesiologyA systematic review of the role of videolaryngoscopy in successful orotracheal intubationBmc Anesthesiology
Annales Francaises D Anesthesie Et De ReanimationA comparison of Airtraq (TM) laryngoscope and standard direct laryngoscopy in adult patients with immobilized cervical spineAnnales Francaises D Anesthesie Et De Reanimation
TrialsSWIVIT - Swiss video-intubation trial evaluating video-laryngoscopes in a simulated difficult airway scenario: study protocol for a multicenter prospective randomized controlled trial in SwitzerlandTrials
Annales Francaises D Anesthesie Et De ReanimationTwo consecutive intubations using neonatal Airtraq (R) in an infant with difficult airwayAnnales Francaises D Anesthesie Et De Reanimation
AnaesthesiaTracheal intubation in patients with rigid collar immobilisation of the cervical spine: a comparison of Airtraq (R) and LMA CTrach (TM) devicesAnaesthesia
AnaesthesiaEvaluation of the GlideScope (R) for tracheal intubation in patients with cervical spine immobilisation by a semi-rigid collarAnaesthesia
Obesity SurgeryVideo-Assisted Versus Conventional Tracheal Intubation in Morbidly Obese PatientsObesity Surgery
AnaesthesiaDetermination of the optimal stylet strategy for the C-MAC (R) videolaryngoscopeAnaesthesia
British Journal of AnaesthesiaTracheal intubation in patients with cervical spine immobilization: a comparison of the Airwayscope (R), LMA CTrach (R), and the Macintosh laryngoscopesBritish Journal of Anaesthesia
British Journal of AnaesthesiaComparison of the Glidescope (R), the Pentax AWS (R), and the Truview EVO2 (R) with the Macintosh laryngoscope in experienced anaesthetists: a manikin studyBritish Journal of Anaesthesia
Anaesthesia and Intensive Care
Video and optic laryngoscopy assisted tracheal intubation - the new era
Anaesthesia and Intensive Care, 37(2):
Prehospital Emergency CareComparison of Use of the the Airtraq with Direct Laryngoscopy by Paramedics in the Simulated AirwayPrehospital Emergency Care
British Journal of AnaesthesiaRandomized controlled trial of the Pentax AWS (R), Glidescope (R), and Macintosh laryngoscopes in predicted difficult intubationBritish Journal of Anaesthesia
Anales Del Sistema Sanitario De Navarra
Airtraq (R) optical laryngoscopy
Anales Del Sistema Sanitario De Navarra, 32(1):
Acta Anaesthesiologica ScandinavicaPerformance of the Airtraq((TM)) laryngoscope after failed conventional tracheal intubation: a case seriesActa Anaesthesiologica Scandinavica
AnaesthesiaA comparison of cervical spine movement during laryngoscopy using the Airtraq (R) or Macintosh laryngoscopesAnaesthesia
AnaesthesiaComparison of the Glidescope (R), the McGrath (R), the Airtraq (R) and the Macintosh laryngoscopes in simulated difficult airwaysAnaesthesia
AnaesthesistAirway management in a patient with immobilized cervical spineAnaesthesist
Journal of Clinical AnesthesiaUse of the Airtraq as the primary technique to manage anticipated difficult airway: a report of three casesJournal of Clinical Anesthesia
AnaesthesiaTracheal intubation using the Airtraq (R): a comparison with the lightwandAnaesthesia
AnaesthesiaLaryngoscopy: time to change our viewAnaesthesia
Canadian Journal of Anaesthesia-Journal Canadien D Anesthesie
Use of the nasotracheal Airtraq (R) to assist difficult nasal fibreoptic intubation
Canadian Journal of Anaesthesia-Journal Canadien D Anesthesie, 55():
British Journal of AnaesthesiaTracheal intubation of morbidly obese patients: a randomized trial comparing performance of Macintosh and Airtraq (TM) laryngoscopesBritish Journal of Anaesthesia
Canadian Journal of Anaesthesia-Journal Canadien D AnesthesieAirway topical anesthesia using the Airtraq (R) laryngoscope in patients with difficult airwaysCanadian Journal of Anaesthesia-Journal Canadien D Anesthesie
AnaesthesiaEvaluation of the Airtraq (R) and Macintosh laryngoscopes in patients at increased risk for difficult tracheal intubationAnaesthesia
British Journal of AnaesthesiaComparison of Macintosh, Truview EVO2 (R), Glidescope (R), and Airwayscope (R) laryngoscope use in patients with cervical spine immobilizationBritish Journal of Anaesthesia
British Journal of Anaesthesia
Difficult tracheal intubation
British Journal of Anaesthesia, 104(2):
AnaesthesistIndirect laryngoscopy. Alternatives to securing the airwayAnaesthesist
AnaesthesiaComparison of the Glidescope((R)) and Airtraq((R)) optical laryngoscopes in patients undergoing direct microlaryngoscopyAnaesthesia
Current Opinion in AnesthesiologyReview of video laryngoscopy and rigid fiberoptic laryngoscopyCurrent Opinion in Anesthesiology
Current Opinion in Critical CareAirway management and initial resuscitation of the trauma patientCurrent Opinion in Critical Care
European Journal of Anaesthesiology (EJA)Learning curves of the Glidescope, the McGrath and the Airtraq laryngoscopes: a manikin studyEuropean Journal of Anaesthesiology (EJA)
Journal of Neurosurgical AnesthesiologyNeuroanesthesiology Review—2007Journal of Neurosurgical Anesthesiology
© 2007 American Society of Anesthesiologists, Inc.
Publication of an advertisement in Anesthesiology Online does not constitute endorsement by the American Society of Anesthesiologists, Inc. or Lippincott Williams & Wilkins, Inc. of the product or service being advertised.
- Articles in PubMed by Chrisen H. Maharaj, B.Sc., M.B., F.C.A.R.C.S.I., D.P.M.
- This article in PubMed
- Articles in Google Scholar by Chrisen H. Maharaj, B.Sc., M.B., F.C.A.R.C.S.I., D.P.M.
- Other articles in this journal by Chrisen H. Maharaj, B.Sc., M.B., F.C.A.R.C.S.I., D.P.M.