Respiratory Variations in Pulse Oximetry Plethysmographic Waveform Amplitude to Predict Fluid Responsiveness in the Operating Room
Cannesson, Maxime M.D.*; Attof, Yassin M.D.*; Rosamel, Pascal M.D.†; Desebbe, Olivier M.D.‡; Joseph, Pierre M.D.†; Metton, Olivier M.D.§; Bastien, Olivier M.D., Ph.D.∥; Lehot, Jean-Jacques M.D., Ph.D.#
Background: Respiratory variations in pulse oximetry plethysmographic waveform amplitude (ΔPOP) are related to respiratory variations in pulse pressure (ΔPP) and are sensitive to changes in preload. The authors hypothesized that ΔPOP can predict fluid responsiveness in mechanically ventilated patients during general anesthesia.
Methods: Twenty-five patients referred for cardiac surgery were studied after induction of general anesthesia. Hemodynamic data (cardiac index, central venous pressure, pulmonary capillary wedge pressure, ΔPP, and ΔPOP) were recorded before and after volume expansion (500 ml hetastarch, 6%). Fluid responsiveness was defined as an increase in cardiac index of 15% or greater.
Results: Volume expansion induced changes in cardiac index (2.0 ± 0.4 to 2.3 ± 0.5 mmHg; P < 0.05), ΔPP (11 ± 7 to 6 ± 5%; P < 0.05), and ΔPOP (12 ± 9 to 7 ± 5%; P < 0.05). ΔPOP and ΔPP were higher in responders than in nonresponders (17 ± 8 vs. 6 ± 4 and 14 ± 7 vs. 6 ± 4%, respectively; P < 0.05 for both). A ΔPOP greater than 13% before volume expansion allowed discrimination between responders and nonresponders with 80% sensitivity and 90% specificity. There was a significant relation between ΔPOP before volume expansion and percent change in cardiac index after volume expansion (r = 0.62; P < 0.05).
Conclusions: ΔPOP can predict fluid responsiveness noninvasively in mechanically ventilated patients during general anesthesia. This index has potential clinical applications.
RECENTLY published studies have shown that intraoperative optimization of cardiac output using volume expansion decreases postoperative morbidity and hospital stay.1
On the other hand, if inappropriate, volume expansion may have deleterious effects. Therefore, preload dependence and fluid responsiveness assessments are of major importance during surgery. Static indicators of fluid responsiveness such as central venous pressure (CVP), pulmonary capillary wedge pressure (PCWP), or left ventricular end diastolic area index are invasive or not easily available and have been shown to be poor predictors of fluid responsiveness.2–9
Dynamic indicators, relying on the respiratory variations in stroke volume or its surrogates in mechanically ventilated patients, have been shown to be superior to static indicators for prediction of fluid responsiveness.2–5,7,9–12
However, they are either invasive (respiratory variations in arterial pulse pressure (ΔPP),2
stroke volume variations5,12
) with their associated complications,13,14
technically challenging (respiratory variations in pulse Doppler aortic flow velocity,3
inferior vena cava diameter11
), or not widely available (esophageal Doppler15
Respiratory variations in the pulse oximetry plethysmographic waveform amplitude (ΔPOP) have now been extensively studied in mechanically ventilated patients.16,17
They are strongly related to the respiratory variations in arterial pulse pressure (ΔPP)18–20
and are sensitive to changes in ventricular preload.21
Recently published studies have shown promising results regarding the ability of ΔPOP to predict fluid responsiveness in the intensive care unit22
and in the operating room.20
However, the POP waveform must be recorded and analyzed in specific conditions to be interpretable for fluid responsiveness prediction.
The hypotheses tested in the current study were that ΔPOP can predict fluid responsiveness in mechanically ventilated patients during general anesthesia in the operating room and that ΔPOP is as sensitive and specific as ΔPP to predict fluid responsiveness.
Materials and Methods
The protocol was approved by the institutional review board for human subjects of our institution (Comité Consultatif de Protection des Personnes dans la Recherche Biomédicale Lyon B). All patients gave informed and written consent. We studied 25 consecutive patients undergoing coronary artery bypass grafting. Patients with cardiac arrhythmias and intracardiac shunt were excluded.
This group consisted of 18 men and 7 women aged between 56 and 85 yr (mean age, 69 ± 7 yr). Eighteen patients received β-blockers preoperatively. Induction of anesthesia was performed with propofol (3–5 mg/kg) and sufentanil (0.5–1.0 μg/kg), and orotracheal intubation was facilitated with pancuronium (0.1–0.15 mg/kg). After induction of anesthesia, a 8-cm, 5-French tipped catheter (Arrow International Inc., Reading, PA) was inserted in the left or right radial artery, and a triple-lumen, 16-cm, 8.5-French central venous catheter (Arrow International Inc.) and a 7.5-French pulmonary artery catheter (Swan-Ganz catheter; Baxter Edwards, Lifescience, LLC, Irvine, CA) were inserted in the right internal jugular vein. Pressure transducers (Medex Medical Ltd., Rossendale, Lancashire, United Kingdom) were placed on the midaxillary line and fixed to the operation table to keep the transducer at the atrial level all along the study protocol. All transducers were zeroed to atmospheric pressure. Correct position of the pulmonary artery catheter in West zone 3 was assessed using the method of Teboul et al.23
Cardiac output was measured by thermodilution, using the average of five successive measurements obtained by injection of 10 ml dextrose at room temperature randomly during respiratory cycle. Cardiac index and stroke volume index were calculated using the same formula: cardiac index = cardiac output/body surface area. A pulse oximeter probe (Oxymax; Tyco Healthcare Group LP, Pleasanton, CA) was attached to the index of either right or left hand and was wrapped to prevent outside light from interfering with the signal. Anesthesia was maintained with continuous infusions of propofol (5–8 mg · kg−1
) and sufentanil (0.7–1.0 μg · kg−1
) to keep a Bispectral Index (Aspect 1000; Aspect Medical Systems Inc., Natick, MA) between 40 and 50. All patients were ventilated in a volume-controlled mode with a tidal volume of 8–10 ml/kg body weight at a frequency of 12–15 cycles/min. Positive end-expiratory pressure was set between 0 and 2 cm H2
O according to the attending physician.
Data Recording and Analysis
Arterial pressure and POP waveforms were recorded from a bedside monitor (Intellivue MP70; Philips Medical Systems, Suresnes, France) to a personal computer using data acquisition software (TrendfaceSolo 1.1; Ixellence GmbH, Wildau, Germany) and were analyzed by an observer blinded to the other hemodynamic data.
Respiratory Variations in Pulse Pressure Analysis.
Pulse pressure (PP) was defined as the difference between systolic and diastolic pressure. Maximal (PPmax) and minimal (PPmin) values were determined over the same respiratory cycle. ΔPP was then calculated as described by its authors2
: ΔPP = (PPmax − PPmin)/[(PPmax + PPmin)/2]. The measurements were repeated on three consecutive respiratory cycles and averaged for statistical analysis.
Respiratory Variations in POP Waveform Amplitude Analysis.
The plethysmographic gain factor was held constant during POP waveform recording so that the POP waveform amplitude did not depend on automatic gain adjustment. The signal quality was considered as optimal when the perfusion index displayed by the monitor was greater than 1.0, as recommended by the manufacturer. POP waveform amplitude was measured on a beat-to-beat basis as the vertical distance between peak and preceding valley trough in the waveform and was expressed as pixels. Maximal POP (POPmax) and minimal POP (POPmin) were determined over the same respiratory cycle (fig. 1
). ΔPOP was then calculated as previously described18,21
: ΔPOP = (POPmax − POPmin)/[(POPmax + POPmin)/2]. The measurements were repeated on three consecutive respiratory cycles and averaged for statistical analysis.
Other Hemodynamic Measurements.
At each step of the protocol, the following parameters were recorded: systolic arterial pressure, mean arterial pressure, diastolic arterial pressure, heart rate, end-expiratory CVP, end-expiratory PCWP, oxygen saturation measured by pulse oximeter, stroke volume index, cardiac index (CI), and systemic vascular resistance index.
All patients were studied immediately after induction of anesthesia and after a 5-min period of hemodynamic stability with no changes in anesthetic protocol and no volume expansion. A baseline set of hemodynamic measurements was then performed and followed by an intravenous volume expansion consisting in 500 ml hetastarch, 6%, given over 10 min. Hemodynamic measurements were performed within 3 min after volume expansion.
All data are presented as mean ± SD. Changes in hemodynamic parameters induced by volume expansion were assessed using a nonparametric Mann–Whitney U test or Wilcoxon rank sum test when appropriate. Patients were divided into two groups according to the percent increase in CI after volume expansion: Responders were defined as patients presenting an increase of 15% or more2
in CI, and nonresponders were defined as patients presenting an increase of less than 15% in CI. Receiver operating characteristic curves were generated for CI, CVP, PCWP, ΔPP, and ΔPOP, varying the discriminating threshold of each parameter, and areas under the receiver operating characteristic curves were calculated and compared24
(MedCalc 126.96.36.199; MedCalc Software, Mariakerke, Belgium). Considering previously published results,2
power analysis showed that 25 patients were necessary to detect differences of 0.15 between ΔPOP and ΔPP areas under the receiver operating characteristic curves (5% type I error rate, 80% power, two-tailed test). The Spearman rank method was used to test correlation. A P
value less than 0.05 was considered as statistically significant. All statistic analyses were performed using SPSS 13.0 for Windows (SPSS, Chicago, IL).
No patients received vasoactive drugs. Pulse oximeter plethysmography waveform was analyzable in all patients. Perfusion index was always greater than 1.0, consistent with an optimal POP waveform signal. PPmax and POPmax appeared during inspiration in all patients. Correlation between ΔPP and ΔPOP over the 50 measurements was strong (r
= 0.90; P
< 0.01) (fig. 2
). Agreement between ΔPP and ΔPOP (Bland-Altman analysis) was 1.1 ± 3.2% (fig. 2
). We also observed significant relations between ΔPP and ΔPOP before volume expansion (r
= 0.95; P
< 0.01) and after volume expansion (r
= 0.73; P
< 0.01). Agreement between ΔPP and ΔPOP before volume expansion was 1.3 ± 2.8%. Agreement between ΔPP and ΔPOP after volume expansion was 1.1 ± 3.4%.
Changes in Hemodynamic Parameters after Volume Expansion
Hemodynamic data at baseline and after volume expansion are shown in table 1
. As expected, volume expansion induced significant increases in CI (from 2.0 ± 0.4 to 2.3 ± 0.5 l · min−1
< 0.001), mean arterial pressure (from 60.0 ± 9.1 to 68.1 ± 10.2 mmHg; P
< 0.001), CVP (from 9.8 ± 5.2 to 12.0 ± 3.2 mmHg; P
< 0.001), and PCWP (from 12.2 ± 4.9 to 15.6 ± 4.4 mmHg; P
< 0.001). At the same time, we observed significant decreases in both ΔPP (from 11.1 ± 7.3% to 6.2 ± 4.6%; P
< 0.01) and ΔPOP (from 12.3 ± 8.7% to 7.2 ± 4.8%; P
ΔPOP to Predict Fluid Responsiveness
Fifteen patients were responders and 10 patients were nonresponders to volume expansion. Their hemodynamic data are shown in table 2
. ΔPP and ΔPOP were significantly higher in responders than in nonresponders (14.5 ± 6.9 vs.
5.8 ± 4.3 and 16.7 ± 8.2% vs.
5.8 ± 4.0%; P
< 0.01 for both) (fig. 3
), whereas differences in CVP (10.5 ± 5.9 mmHg in responders vs.
8.5 ± 3.8 mmHg in nonresponders; P
= 0.33), PCWP (12.3 ± 4.3 mmHg in responders vs.
12.1 ± 5.9 mmHg in nonresponders; P
= 0.94), and CI (1.98 ± 0.43 ml · min−1
in responders vs.
1.95 ± 0.29 ml · min−1
in nonresponders; P
= 0.84) did not reach statistical significance between these two groups. The areas under the receiver operating characteristic curves (± SE) were as follows: 0.847 ± 0.084 for ΔPP, 0.847 ± 0.081 for ΔPOP, 0.570 ± 0.115 for CVP, 0.510 ± 0.122 for PCWP, and 0.520 ± 0.118 for CI (fig. 4
). The area for ΔPOP was significantly higher than the area for CVP, PCWP, and mean arterial pressure (P
< 0.05 for both). The difference in area under the curve between ΔPP and ΔPOP did not reach significance (P
= 0.91). The threshold ΔPP value of 11% allowed discrimination between responders and nonresponders with a sensitivity of 80% and a specificity of 90%. The threshold ΔPOP value of 13% allowed discrimination between responders and nonresponders with a sensitivity of 93% and a specificity of 90%.
ΔPOP to Quantify Response to Volume Expansion
There was a statistically significant positive linear correlation between ΔPOP at baseline and percent changes in CI induced by volume expansion (ΔCI) (r
= 0.62; P
< 0.01) as well as between ΔPP and ΔCI (r
= 0.56; P
< 0.01) (fig. 5
), indicating that the higher ΔPOP and ΔPP are at baseline, the higher ΔCI is. We observed no statistically significant relation between CVP at baseline and ΔCI (r
= −0.28; P
= 0.17) and between PCWP at baseline and ΔCI (r
= −0.06; P
= 0.79) (fig. 5
This study shows that ΔPOP can predict response to volume expansion and can quantify the effects of volume expansion on hemodynamic parameters in the operating room.
Over the past 10 years, fluid responsiveness assessment has been extensively studied in mechanically ventilated patients.2–12,15
It is now well accepted that dynamic parameters (relying on the cardiopulmonary interactions in patients under positive pressure ventilation) are better predictors of response to volume expansion than static indicators. Respiratory variations in left ventricular stroke volume or its surrogates are predictive of response to volume expansion.7
It has been demonstrated that the respiratory variations in arterial PP are better predictors of fluid responsiveness than the respiratory variations in systolic arterial pressure.
Several studies are now suggesting that the pulse oximeter waveform contains much other information that has been underexploited despite its potential useful clinical applications.16–19,21,22,25,26
One of these informations is the respiratory variations component of the waveform. The respiratory variations in the POP waveform amplitude (ΔPOP) are closely related to the respiratory variations in the arterial pulse pressure (ΔPP)18,19,27
and are sensitive to changes in ventricular preload.21
Moreover, some recent studies suggest that ΔPOP may be an accurate predictor of fluid responsiveness.20,22
Solus-Biguenet et al.20
investigated the ability of various noninvasive indices to predict fluid responsiveness during major hepatic surgery. They found that ΔPOP was significantly higher in responders compared with nonresponders to volume expansion and that ΔPOP was able to predict fluid responsiveness. More recently, a promising study from Natalini et al.22
demonstrated that ΔPOP is able to predict response to volume expansion in patients with circulatory failure related to sepsis in the intensive care setting. They found no difference between ΔPOP and ΔPP to compare fluid responsiveness in these patients. These two previously published studies are clearly encouraging, and our results are consistent with theirs. In the current study, we found better sensitivity and specificity regarding the ability of ΔPOP to predict fluid responsiveness. However, our protocol used slightly different techniques for plethysmographic waveform acquisition and analysis and was performed in a different surgical patient population.28
Pulse oximeters are part of the routine monitoring in patients under mechanical ventilation.29
The signal displayed by the pulse oximeter is proportional to light absorption between the nail and the anterior face of the finger. During systole, the amount of hemoglobin present in the fingertip is increased and, consequently, light absorption is decreased. An inverse phenomenon is observed during diastole. Therefore, the POP waveform depends on the arterial pulse.29
The pulse density signal is not really the pulse density change but a time-averaged mean-adjusted signal wherein the actual mean density is held constant but the changes in density reported for calibration purposes for the calculation of oxygen saturation. The raw plethysmographic signal is much more variable. Density will be a function of tissue (nonchanging signal) and blood (changing signal), and changing density will be a function of changing blood. Finally, the blood density change is determined by both perfusion pressure and vasomotor tone.25
POP waveform also depends on outside light absorption. In our study, attention was paid to all of these parameters. The pulse oximeter was wrapped to prevent outside light from interfering with the signal, gain was maintained constant during POP waveform recording, and the perfusion index was always greater than 1.0. Moreover, one can postulate that vascular tone is constant throughout a single respiratory cycle and that it does not impact the minimal and maximal pulse oximeter waveform amplitudes during the same respiratory cycle. Another important point is that the POP waveform is unitless. However, because ΔPOP assesses relative changes in POP waveform amplitude, it can be used whatever the unit.18–21
We chose to record the POP waveform at the finger, and we consistently did it over the 25 studied patients. This is of major importance because it is now known that these variations depend on the site of measurements.17
Shelley et al.17
recently demonstrated that respiratory variations in POP waveform can be up to 10 times stronger in the region of the head when compared with the finger. Consequently, mixing the site of measurements may induce bias. Further studies evaluating the ear, forehead, and finger signal for fluid responsiveness prediction are required to answer this question.
We used a specific pulse oximeter model for this study. Whether our finding can be exported to other models cannot be answered. However, other authors using different models found similar results.17,19,20,22
Further studies exploring this issue are warranted.
Finally, ΔPOP has to be calculated off-line because no monitor allows online estimation. Future studies are planned to elaborate and to test software for automated calculation of ΔPOP.
As other indices relying on the respiratory variations of left ventricular stroke volume or its surrogate, ΔPOP cannot be calculated in patients with cardiac arrhythmia. Similarly, ΔPOP interpretation in patients with right ventricular failure must be cautious.
The threshold value of 13% for prediction of fluid responsiveness has to be interpreted with caution. As for ΔPP and for any other indices of fluid responsiveness, threshold value may vary among studies and settings. To quote Solus-Biguenet et al.20
: “ΔPP values ranging from 8 to 13% may constitute an inconclusive or ‘gray zone’30
where its predictive value is uncertain.”
Finally, ΔPOP can only be performed in mechanically ventilated patients during general anesthesia. However, this situation reflects the most common situation encountered by the anesthesiologists in their daily practice.
In conclusion, ΔPOP seems to be a noninvasive and widely available index of fluid responsiveness in mechanically ventilated patients during general anesthesia. ΔPOP has potential clinical applications.
The authors thank all the physicians and nurse anesthetists from the Department of Anesthesiology (Louis Pradel Hospital, Hospices Civils de Lyon, Lyon, France) for their help and support during this study.
1. Grocott MPW, Mythen MG, Gan TJ: Perioperative fluid management and clinical outcomes in adults. Anesth Analg 2005; 100:1093–106
2. Michard F, Boussat S, Chemla D, Anguel N, Mercat A, Lecarpentier Y, Richard C, Pinsky MR, Teboul JL: Relation between respiratory changes in arterial pulse pressure and fluid responsiveness in septic patients with acute circulatory failure. Am J Respir Crit Care Med 2000; 162:134–8
3. Feissel M, Michard F, Mangin I, Ruyer O, Faller JP, Teboul JL: Respiratory changes in aortic blood velocity as an indicator of fluid responsiveness in ventilated patients with septic shock. Chest 2001; 119:867–73
4. Tavernier B, Makhotine O, Lebuffe G, Dupont J, Scherpereel P: Systolic pressure variation as a guide to fluid therapy in patients with sepsis-induced hypotension. Anesthesiology 1998; 89:1313–21
5. Rex S, Brose S, Metzelder S, Huneke R, Schalte G, Autschbach R, Rossaint R, Buhre W: Prediction of fluid responsiveness in patients during cardiac surgery. Br J Anaesth 2004; 93:782–8
6. Wiesenack C, Fiegl C, Keyser A, Prasser C, Keyl C: Assessment of fluid responsiveness in mechanically ventilated cardiac surgical patients. Eur J Anaesthesiol 2005; 22:658–65
7. Michard F: Changes in arterial pressure during mechanical ventilation. Anesthesiology 2005; 103:419–28
8. Bendjelid K, Romand JA: Fluid responsiveness in mechanically ventilated patients: A review of indices used in intensive care. Intensive Care Med 2003; 29:352–60
9. Pinsky MR, Teboul JL: Assessment of indices of preload and volume responsiveness. Curr Opin Crit Care 2005; 11:235–9
10. Feissel M, Badie J, Merlani PG, Faller JP, Bendjelid K: Pre-ejection period variations predict the fluid responsiveness of septic ventilated patients. Crit Care Med 2005; 33:2534–9
11. Feissel M, Michard F, Faller JP, Teboul JL: The respiratory variation in inferior vena cava diameter as a guide to fluid therapy. Intensive Care Med 2004; 30:1834–7
12. Reuter DA, Felbinger TW, Schmidt C, Kilger E, Goedje O, Lamm P, Goetz AE: Stroke volume variation for assessment of cardiac responsiveness to volume loading in mechanically ventilated patients after cardiac surgery. Intensive Care Med 2002; 28:392–8
13. Bedford RF, Wollman H: Complications of percutaneous radial-artery cannulation: An objective prospective study in man. Anesthesiology 1973; 38:228–36
14. Jones RM, Hill AB, Nahrwold ML, Bolles RE: The effect of method of radial artery cannulation on postcannulation blood flow and thrombus formation. Anesthesiology 1981; 55:76–8
15. Slama M, Masson H, Teboul JL, Arnould ML, Nait-Kaoudjt R, Colas B, Peltier M, Tribouilloy C, Susic D, Frohlich E, Andréjak M: Monitoring of respiratory variations of aortic blood flow velocity using oesophageal Doppler. Intensive Care Med 2004; 30:1182–7
16. Shelley KH, Awad AA, Stout RG, Silverman DG: The use of joint time frequency analysis to quantify the effect of ventilation on the pulse oximeter waveform. J Clin Monit Comput 2006; 20:81–7
17. Shelley KH, Jablonka DH, Awad AA, Stout RG, Rezkanna H, Silverman DG: What is the best site for measuring the effect of ventilation on the pulse oximeter waveform? Anesth Analg 2006; 103:372–7
18. Cannesson M, Besnard C, Durand PG, Bohe J, Jacques D: Relation between respiratory variations in pulse oximetry plethysmographic waveform amplitude and arterial pulse pressure in ventilated patients. Crit Care 2005; 9:R562–8
19. Natalini G, Rosano A, Franschetti ME, Fachetti P, Bernardini A: Variations in arterial blood pressure and photoplethysmography during mechanical ventilation. Anesth Analg 2006; 103:1182–8
20. Solus-Biguenet H, Fleyfel M, Tavernier B, Kipnis E, Onimus J, Robin E, Lebuffe G, Decoene C, Pruvot FR, Vallet B: Non-invasive prediction of fluid responsiveness during major hepatic surgery. Br J Anaesth 2006; 97:808–16
21. Cannesson M, Desebbe O, Hachemi M, Jacques D, Bastien O, Lehot JJ: Respiratory variations in pulse oximeter waveform amplitude are influenced by venous return in mechanically ventilated patients under general anaesthesia. Eur J Anaesthesiol 2007; 24:245–51
22. Natalini G, Rosano A, Taranto M, Faggian B, Vittorielli E, Bernardini A: Arterial versus plethysmographic dynamic indices to test responsiveness for testing fluid administration in hypotensive patients: A clinical trial. Anesth Analg 2006; 103:1478–84
23. Teboul JL, Besbes M, Andrivet P, Axler O, Douguet D, Zelter M, Lemaire F, Brun-Buisson C: A bedside index assessing the reliability of pulmonary occlusion pressure measurements during mechanical ventilation with positive end-expiratory pressure. J Crit Care 1992; 7:22–9
24. Hanley JA, McNeil BJ: A method of comparing the areas under receiver operating characteristic curves derived from the same cases. Radiology 1983; 148:839–43
25. Shelley KH, Murray WB, Chang D: Arterial-pulse oximetry loops: A new method of monitoring vascular tone. J Clin Monit 1997; 13:223–8
26. Shelley KH, Dickstein M, Shulman SM: The detection of peripheral venous pulsation using the pulse oximeter as a plethysmograph. J Clin Monit 1993; 9:283–7
27. Monnet X, Lamia B, Teboul JL: Pulse oximeter as a sensor of fluid responsiveness: Do we have our finger on the best solution? Crit Care 2005; 9:429–30
28. Cannesson M, Desebbe O, Lehot JJ: Fluid responsiveness using non-invasive predictors during major hepatic surgery. Br J Anaesth 2007; 98:272–4
29. Dorlas JC, Nijboer JA: Photo-electric plethysmography as a monitoring device in anaesthesia: Application and interpretation. Br J Anaesth 1985; 57:524–30
30. Feinstein AR: The inadequacy of binary models for the clinical reality of three-zone diagnostic decisions. J Clin Epidemiol 1990; 43:109–13
This article has been cited 36 time(s).
Biomedical Signal Processing and ControlPulse pressure variation tracking using sequential Monte Carlo methodsBiomedical Signal Processing and Control
Acta PaediatricaPerfusion index and its dynamic changes in preterm neonates with patent ductus arteriosusActa Paediatrica
Anesthesia and AnalgesiaDoes the pleth variability index indicate the respiratory-induced variation in the plethysmogram and arterial pressure waveforms?Anesthesia and Analgesia
AnaesthesistArterial pressure curve and fluid statusAnaesthesist
Anesthesia and AnalgesiaThe ability of a novel algorithm for automatic estimation of the respiratory variations, in arterial pulse pressure to monitor fluid responsiveness in the operating roomAnesthesia and Analgesia
Acta Anaesthesiologica ScandinavicaPleth variability index predicts hypotension during anesthesia inductionActa Anaesthesiologica Scandinavica
AnaesthesistPerioperative fluid and volume management - Goal-directed therapy necessary!Anaesthesist
American Journal of Respiratory and Critical Care MedicineUpdate in critical care 2007American Journal of Respiratory and Critical Care Medicine
American Journal of Emergency MedicineRelation between pulse oximetry plethysmographic waveform amplitude induced by passive leg raising and cardiac index in spontaneously breathing subjectsAmerican Journal of Emergency Medicine
Intensive Care MedicineComment on "Plethysmographic dynamic indices predict fluid responsiveness in septic ventilated patients" by Feissel et alIntensive Care Medicine
British Journal of AnaesthesiaPleth variability index to monitor the respiratory variations in the pulse oximeter plethysmographic waveform amplitude and predict fluid responsiveness in the operating theatreBritish Journal of Anaesthesia
Anesthesia and AnalgesiaThe Ability of Stroke Volume Variations Obtained with Vigileo/FloTrac System to Monitor Fluid Responsiveness in Mechanically Ventilated PatientsAnesthesia and Analgesia
Journal of Cardiothoracic and Vascular AnesthesiaArterial Pressure Variation and Goal-Directed Fluid TherapyJournal of Cardiothoracic and Vascular Anesthesia
Anasthesiologie & Intensivmedizin
Optimisation of intraoperative fluid management - an unresolved problem
Anasthesiologie & Intensivmedizin, 50():
Anaesthesia, Pain, Intensive Care and Emergency: A.P.I.C.E
Use of pulse oximeter waveform a non invasive functional haemodynamic monitoring technique
Anaesthesia, Pain, Intensive Care and Emergency: A.P.I.C.E, ():
Acta Anaesthesiologica ScandinavicaPulse pressure variation and stroke volume variation during different loading conditions in a paediatric animal modelActa Anaesthesiologica Scandinavica
Anesthesia and AnalgesiaThe Ability of Pleth Variability Index to Predict the Hemodynamic Effects of Positive End-Expiratory Pressure in Mechanically Ventilated Patients Under General AnesthesiaAnesthesia and Analgesia
Critical CareAbility of pleth variability index to detect hemodynamic changes induced by passive leg raising in spontaneously breathing volunteersCritical Care
Anasthesiologie & Intensivmedizin
Perioperative haemodynamic monitoring
Anasthesiologie & Intensivmedizin, 51():
British Journal of AnaesthesiaPleth variability index is a weak predictor of fluid responsiveness in patients receiving norepinephrineBritish Journal of Anaesthesia
2012 Annual International Conference of the Ieee Engineering in Medicine and Biology Society (Embc)
Modulation of finger photoplethysmographic traces during forced respiration: venous blood in motion?
2012 Annual International Conference of the Ieee Engineering in Medicine and Biology Society (Embc), ():
Journal of Critical CarePlethysmographic variation index predicts fluid responsiveness in ventilated patients in the early phase of septic shock in the emergency department: A pilot studyJournal of Critical Care
Journal of Clinical Monitoring and ComputingNew analyses of standard physiological signals: the old pipes give the sweetest smokeJournal of Clinical Monitoring and Computing
Current Opinion in AnesthesiologyPulse oximetry and photoplethysmographic waveform analysis of the esophagus and bowelCurrent Opinion in Anesthesiology
Current Opinion in Critical CareNoninvasive assessment of arterial pressureCurrent Opinion in Critical Care
Current Opinion in Critical CareVolume responsivenessCurrent Opinion in Critical Care
AnesthesiologyAt the Threshold of Noninvasive Functional Hemodynamic MonitoringAnesthesiology
Current Opinion in AnesthesiologyUsing ventilation-induced plethysmographic variations to optimize patient fluid statusCurrent Opinion in Anesthesiology
Current Opinion in Critical CareThe pulse oximetry plethysmographic curve revisitedCurrent Opinion in Critical Care
© 2007 American Society of Anesthesiologists, Inc.
Publication of an advertisement in Anesthesiology Online does not constitute endorsement by the American Society of Anesthesiologists, Inc. or Lippincott Williams & Wilkins, Inc. of the product or service being advertised.