Spectral Entropy as a Measure of Hypnosis in Children
Klockars, Jaakko G. M. M.D.*; Hiller, Arja M.D., Ph.D.*; Ranta, Seppo M.D., Ph.D.†; Talja, Pia M.Sc.‡; van Gils, Mark J. Ph.D.§; Taivainen, Tomi M.D., Ph.D.∥
Background: The Datex-Ohmeda S/5 Entropy Module (Datex-Ohmeda Division, Instrumentarium Corp., Helsinki, Finland), using time-frequency balanced Spectral Entropy, is a novel tool for monitoring the hypnotic state during anesthesia. The Entropy Module produces two values, State Entropy (SE) and Response Entropy (RE), and in adults, it has been shown to measure reliably the hypnotic effects of various drugs. In children, Spectral Entropy has been only preliminary studied. The authors' aim was to study Spectral Entropy as a marker of hypnotic state during general anesthesia in infants and children.
Methods: Twenty infants (aged 1 month-1 yr) and 40 children (aged 1–15 yr) were anesthetized for surgery using standardized sevoflurane-nitrous oxide-based anesthesia. The relationships between SE, RE, or Bispectral Index (BIS) and (1) a modified Observer's Assessment of Alertness/Sedation Scale, (2) non-steady state end-tidal concentration of sevoflurane, (3) steady state end-tidal concentration of sevoflurane, and (4) hemodynamic values were calculated using prediction probability, nonlinear regression, and correlation coefficients, as appropriate. The performances of SE, RE, and BIS were compared.
Results: The prediction probability values (± SEM) of SE, RE, and BIS versus the modified Observer's Assessment of Alertness/Sedation Scale in the induction phase were 0.83 ± 0.06, 0.88 ± 0.06, and 0.87 ± 0.08 for children and 0.76 ± 0.08,0.79 ± 0.08, and 0.73 ± 0.10 for infants; values in the emergence phase were 0.68 ± 0.05, 0.74 ± 0.04, and 0.64 ± 0.05 for children and 0.64 ± 0.07, 0.69 ± 0.06, and 0.72 ± 0.06 for infants, respectively. SE, RE, and BIS values were inversely proportionally related to the end-tidal concentration of sevoflurane for children, but for infants, the correlation was much less clear. No significant correlations were found between SE, RE, or BIS values and the hemodynamic values.
Conclusions: Spectral Entropy may be a useful tool for measuring the level of hypnosis in anesthetized children and seems to perform as well as BIS. In infants, the clinical usefulness of both these electroencephalogram-derived methods must be evaluated in further controlled studies.
SEVERAL methods have been introduced to reflect the hypnotic state during anesthesia induced by anesthetic or sedative drugs. These include certain physiologic signs,1
various clinical scales,2,3
and methods based on the electroencephalogram.4
In adults, electroencephalogram-derived indices, such as evoked potential features,5
the Bispectral Index,4,6,7
and the Spectral Entropy,8
have been shown to be indicators of the depth of hypnosis during anesthesia. Benefits of monitoring the level of hypnosis during anesthesia include, e.g.
, faster recovery,9–11
reduced need for anesthetics,10–12
and better individual guidance of anesthesia.7,13
The Datex-Ohmeda S/5 Entropy Module (M-Entropy; Datex-Ohmeda Division, Instrumentarium Corp., Helsinki, Finland), using Spectral Entropy, is a novel tool for monitoring the depth of hypnosis. It measures the regularity of the electroencephalogram signal according to an algorithm published previously,14
providing two values of entropy, the State Entropy (SE) and the Response Entropy (RE). The Entropy Module measures the depth of hypnosis by calculating the spectral entropy of the electroencephalographic power spectrum.15,16
SE, with a scale of 0–91, is computed over the electroencephalogram-dominant frequency range of 0.8–32 Hz, whereas RE, with a scale 0–100, is computed over the electromyographic-dominant frequency of 0.8–47 Hz. When there is no electromyographic activity present, SE and RE show the same number. The recommended range for adequate anesthesia for both Spectral Entropy parameters is from 40 to 60, and zero-values are associated with a completely suppressed electroencephalogram. Spectral Entropy has been validated to measure the depth of hypnosis in adult patients receiving inhaled or intravenous anesthesia.8,17–19
In pediatric patients, electroencephalogram-derived monitors, such as BIS,9,20–22
and auditory evoked potentials23,24
have been studied for the measurement of the depth of hypnosis during anesthesia. BIS has been validated in older children,20–22
but there are conflicting data about the validation of BIS in infants.7,9,21,25–27
Until now, Spectral Entropy has been only preliminarily studied in pediatric patients.25,28
The aim of our study was to evaluate whether Spectral Entropy reliably measures the depth of hypnosis in infants and children during sevoflurane-based general anesthesia. The level of hypnosis was assessed with a modified responsiveness category of the Observer's Assessment of Alertness/Sedation Scale (mOAA/S).2,3
We hypothesized that there should be, first, a correlation between SE or RE values and the mOAA/S and, second, a correlation between SE or RE values and the end-tidal concentration of sevoflurane. Secondarily, we also compared the performance of Spectral Entropy with BIS in children of various ages.
Materials and Methods
Patients and Study Design
After institutional ethics committee approval (Hospital for Children and Adolescents, Helsinki University, Helsinki, Finland) and written informed consent of parents were obtained, 20 infants and 40 children were studied. All patients had an American Society of Anesthesiologists physical status classification of I or II and were scheduled to undergo elective surgery during general anesthesia. The inclusion criteria were an age between 1 month and 15 yr and an elective surgical operation with an estimated duration between 0.5 and 5 h, which required general anesthesia without regional anesthetics. Children were excluded if they had a disease or medication affecting the central nervous system or if the surgery affected the head or the neck of the child.
Premedication with 0.3 mg/kg oral midazolam (maximum 15 mg) was administered approximately 45 min before the estimated induction of anesthesia. All patients received a standardized anesthetic regimen. Mask induction was performed using an open circuit at high fresh gas flow (4-8 l/min): The patients breathed 5% sevoflurane in a mixture of 70% nitrous oxide in oxygen via
a tight-fitting facemask until the loss of eyelash reflex. Thereafter, sevoflurane was decreased to the age-corrected 1 minimal alveolar concentration in oxygen.29
Simultaneously, a peripheral venous catheter was inserted, followed by 2 μg/kg intravenous fentanyl and 0.5 mg/kg rocuronium to facilitate tracheal intubation. When spontaneous breathing diminished, patients were manually ventilated to ensure normocapnia. The patients were intubated 2 min after the loss of eyelash reflex. Thereafter, controlled ventilation was started to maintain the end-tidal carbon dioxide at 35–40 mmHg (ADU S/5 and S/5 Anesthesia Monitor; Datex-Ohmeda Division, Instrumentarium Corp.). After intubation, anesthesia was maintained with 70% N2
O in O2
and sevoflurane, which was adjusted to maintain the heart rate and mean arterial blood pressure within 20% of the baseline values obtained before the anesthetic induction. In addition, 1 μg/kg fentanyl and 0.2 mg/kg rocuronium were given as needed.
At the end of surgery, 15 mg/kg intravenous acetaminophen was given to all patients, and 1 mg/kg ketoprofen was given to patients older than 6 months for postoperative pain treatment. The ulnar nerve near the wrist was stimulated with a peripheral nerve stimulator (Fisher and Paykel Healthcare, Auckland, New Zealand) using train-of-four and tetanic stimulation. The combination of 40 μg/kg intravenous glycopyrrolate and 10 μg/kg neostigmine was administered when needed (i.e., when a fade in the train-of-four stimulation was observed visually) for reversal of residual neuromuscular blockade.
At the end of surgery, the inhalational agents nitrous oxide and sevoflurane were discontinued without any tapering, and the patient was extubated when sufficient spontaneous breathing had returned. Thereafter, the patient was transferred to the postanesthesia care unit. In case of postoperative pain, 0.1 mg/kg morphine was given repeatedly when needed.
In addition to Spectral Entropy and BIS values, end-tidal sevoflurane and nitrous oxide concentrations, electrocardiogram, heart rate, noninvasive blood pressure (measured at 5-min intervals), capnogram, pulse oximetry saturation, and temperature (S/5 Anesthesia Monitor) were monitored continuously during anesthesia and collected with 5-s intervals on a laptop computer (Toshiba Satellite; Toshiba Corp., Tokyo, Japan).
The anesthesiologist in charge was blinded to Spectral Entropy and BIS monitoring throughout the study period, and an independent study nurse supervised recording of all the data. Before the anesthesia induction, a disposable adult Entropy Sensor (Datex-Ohmeda Division, Instrumentarium Corp.) and a disposable adult BIS® Sensor (Aspect Medical Systems, Newton, MA) were placed in the immediate vicinity of each other on the forehead according to the manufacturers' instructions, following as closely as possible the recommended placement in adults. The skin was carefully wiped with an alcohol swab and then allowed to dry. The Entropy Sensor was placed lower and the BIS® Sensor was placed higher on the forehead, and the distal leads of the sensors were placed on the opposite temples. Electrode impedances were considered acceptable if they were below 7.5 kΩ and 10 kΩ for Spectral Entropy and BIS, respectively. Impedances were checked at the start and the end of the recording and whenever unreliable tracking of the electroencephalogram parameters was suspected, and corrective actions were performed if the acceptable impedance levels were exceeded. Spectral Entropy and BIS indices were measured with an M-Entropy and a BIS® Module (BIS® XP, version 3.2; Aspect Medical Systems) of the S/5 Anesthesia Monitor. The sampling rate for the raw electroencephalogram was 400 Hz for Entropy and 256 Hz for BIS; the smoothing window for BIS was 15 s. Spectral Entropy and BIS indices together with all other measured parameters were collected to a laptop computer (Toshiba Satellite) using the S/5 Collect software (Datex-Ohmeda Division, Instrumentarium Corp.). Entropy and BIS measurement were started in the operating room before induction to obtain the baseline values and continued until the patient was awake in the postanesthesia care unit.
The attending anesthesiologist estimated the level of consciousness by using a modified responsiveness category of the Observer's Assessment of Alertness/Sedation Scale (OAA/S),2,3
rated from 0 to 5, where 5 = responds readily to name spoken in normal tone, 4 = lethargic response to name spoken in normal tone, 3 = responds only after name is called loudly and/or repeatedly or eyelash reflex is present, 2 = loss of eyelash reflex, 1 = no purposeful response to train-of-four stimulation or to mild painful stimulus (e.g.
, venous cannulation), and 0 = no purposeful response to 50-Hz stimulation or to painful stimulus (e.g.
, surgery). The mOAA/S estimation started before induction and was continued until the return of normal, fully awake level of consciousness in the postanesthesia care unit. The analyses of the relation between Spectral Entropy and BIS versus
the mOAA/S were divided into two periods: (1) the induction phase: the period from the beginning of the anesthetic induction until the administration of rocuronium; and (2) the emergence phase: from the termination of the surgery to the return of normal consciousness. The mOAA/S was assessed every 30 s during the induction phase and every 2 min during the emergence phase. A standardized verbal stimulus (the patient's name) was given at 30-s intervals in the operating room and postanesthesia care unit until the patient was awake (defined as eye opening, purposeful movement, or phonation as appropriate for age).
The relations between SE, RE, and BIS versus end-tidal concentrations of sevoflurane were analyzed from the start of anesthesia induction to the extubation of the patient. The end-tidal sevoflurane measurements were performed at 5-s intervals. If the end-tidal concentration had been stable, i.e., end-tidal concentration of sevoflurane within ±0.1% for 5 min, the following measurements after this steady 5 min were included in the steady state concentration of sevoflurane analysis until the concentration changed more than ±0.1%. The steady state end-tidal concentrations of sevoflurane were analyzed retrospectively. Otherwise, these 5-s interval measurements were included in the non-steady state end-tidal concentration of sevoflurane analysis.
The correlations between different electroencephalogram devices (SE, RE, and BIS) and between electroencephalogram values versus hemodynamic values (heart rate and mean arterial pressure) were analyzed from the start of anesthesia induction to the extubation of the patient.
The patients were stratified to two groups based on age: infants, 1 month-1 yr; and children, 1–15 yr. The relation between SE, RE, and BIS (ratio scale variables) versus
mOAA/S (ordinal scale variable) were analyzed with prediction probability (Pk).30
Pk values were calculated using PKMACRO software (written in Microsoft Excel; Microsoft Corp., Redmond, WA). Both per-subject and overall group relations were considered. The Pk values for different electroencephalographic indices were compared using the Wilcoxon rank sum test with Bonferroni correction, and comparisons of Pk values between the two different age groups were performed using the Mann-Whitney test.
Equation (Uncited)Image Tools
To determine the pharmacodynamic relation between the quantitative electroencephalogram variables (SE, RE, and BIS) and end-tidal sevoflurane concentration, a nonlinear regression fit was performed. This was done separately for infants and children and for steady state as well as non-steady state conditions. The following model was used for the regression fit:
where E is the recorded value of SE, RE, or BIS and C is the end-tidal sevoflurane concentration. E0
is the value at a drug concentration of 0, whereas Emax
is the maximum value for E. EC50
is the drug concentration corresponding to a half-maximal effect of SE, RE, and BIS. Gamma (γ) is a measure to describe the steepness of the drug concentration-effect relation (the Hill coefficient).31
The data were fitted to the model using a nonlinear least squares fit routine available in a commercial mathematical software package (Matlab, version 6.5, Release 13; The Mathworks Inc., Matick, MA). E0
were specified as 100 for BIS and RE and as 91 for SE.14
The results of the fit were the estimated values of EC50
and γ together with their 95% confidence intervals and plotted curves of the equations with these values, as well as 95% confidence intervals for predicted values of SE, RE, and BIS when new observations of drug concentrations would be obtained.
The correlations between the different electroencephalographic indices (SE, RE, and BIS) were calculated by mean correlation coefficients, as were also the relations between SE, RE, and BIS versus heart rate and mean arterial pressure. These comparisons were made with the Wilcoxon test (pairwise nonparametric). Correlation coefficients were calculated with SPSS V11.01 (SPSS Inc., Chicago, IL).
A P value smaller than 0.05 was considered statistically significant. Because we were mainly conducting exploratory analysis to find out the performance of Spectral Entropy in children, which has not been done before, we did not perform a power analysis.
The demographic data of the study are shown in table 1
. A representative patient case with relevant data is shown in figure 1
. In total, two infants and four children were omitted from different analyses. One infant and two child cases were omitted because of partly missing BIS data, two child cases were omitted because of partly missing mOAA/S data, and one infant case was omitted because of partly missing BIS and mOAA/S data.
The Pk values of SE, RE, and BIS versus
mOAA/S are shown in table 2
. The Pk value of RE in children was significantly higher when compared with RE in infants. Overall, the Pk values were higher among children than among infants, and the Pk values were higher during the induction phase than during the emergence phase. The box plots of SE, RE, and BIS versus
mOAA/S are shown in figure 2
The relationships between SE, RE, and BIS versus
end-tidal concentration of sevoflurane, as used in the pooled analyses for infants and children, are shown in figure 3
. The results of the analysis of the nonlinear regression relation between SE, RE, and BIS versus
non-steady state and steady state end-tidal sevoflurane concentration are presented graphically in figures 4 and 5
. The numerical values of the drug concentration corresponding to a half-maximal effect of SE, RE, and BIS (EC50
) and the steepness of the relation between the drug concentration-effect relation (the Hill coefficient, γ) for both age groups and both sevoflurane end-tidal concentrations are shown in tables 3 and 4
The mean correlation coefficients between the various electroencephalogram indices (SE, RE, and BIS) are shown in table 5
. The mean correlation coefficients between SE, RE, and BIS versus
heart rate and mean arterial pressure are shown in table 6
. There were no significant correlations between SE, RE, or BIS and the heart rate or mean arterial pressure values.
The current study shows that the Spectral Entropy indices (both SE and RE) reflected the level of hypnosis during sevoflurane-nitrous oxide anesthesia in children as judged primarily by Pk and by the pharmacodynamic regression analysis. However, these correlations were relatively low in infants. Overall, in both children and in infants, Spectral Entropy performed as well as BIS.
The Pk values were better for children than for infants and also better during induction than during emergence. The Pk values of both age groups were lower than those of the adult studies.8,17–19
This finding was most evident in infants, making the clinical usefulness of Spectral Entropy in the younger age group less obvious or even questionable. Interestingly, this is the case also with BIS.7,9,21,25–27
As shown by the box plot graph (fig. 2
), there is a significant overlap in all of the electroencephalogram-derived indices (SE, RE, and BIS) when compared with each level of hypnosis as estimated by the behavioral scale (mOAA/S). This is caused by the large interindividual variation of SE, RE, and BIS values during different stages of hypnosis, which was most evident in infants. Especially in infants, the relatively rapid wash-in of sevoflurane seems to produce a phenomenon with “on-off” behavior during the anesthetic induction.
The nonlinear regression analysis (figs. 4 and 5
) demonstrates a pharmacodynamic correlation between SE, RE, and BIS versus
both non-steady state and steady state end-tidal concentration of sevoflurane, which can be assumed from the raw plots of end-tidal sevoflurane versus
the electroencephalographic indices (fig. 3
). In children, there is a gradual relation visible, and fitting the classic Emax
model seems to be justified. However, in infants, this is much less convincing. A curve-fitting algorithm always gives some results, but the plots for the infants' data show that the practical validity of the fitted curve is poor. This is clearly seen as wide confidence intervals for the estimated values of EC50
and γ (tables 3 and 4
). In fact, the individual curves for infants are much steeper than the rather shallow curve calculated over the whole population suggests. This effect, obtained when using a pooled data approach, is well established.32,33
In this study, the relatively small amount of available data points at the level higher than 95% of the Emax
decreased the reliability of the estimated parameters EC50
and γ, as has been previously suggested.34
There are only a few studies in children using Spectral Entropy, i.e.
, only one publication25
and a congress poster have been presented.28
The published study of 23 patients by Davidson and coworkers25
consisted of infants, toddlers, and children undergoing general anesthesia based on isoflurane-nitrous oxide and supplemental peripheral nerve blockade. They found a strong correlation between BIS and Spectral Entropy for toddlers and children, but the correlation was less for infants. The Spectral Entropy indices were higher before anesthesia induction and after emergence than during anesthesia for toddlers and children, but this was not as apparent in infants. The abstract by Arnaout et al.28
presented 53 children aged 2–12 yr undergoing ear, nose, and throat surgery during general anesthesia based on sevoflurane-nitrous oxide. They found that Spectral Entropy indices and BIS were closely correlated with each other, although during deep anesthesia, the Spectral Entropy values were lower than the BIS value. Arnaout et al.
stated that Spectral Entropy indices seem to follow the depth of hypnosis in children anesthetized with sevoflurane. These findings are in line with our findings.
There was no correlation between SE, RE, or BIS versus
heart rate or mean arterial pressure (table 6
). Therefore, hemodynamic parameters do not seem to reflect the level of hypnosis, which has been shown previously also in pediatric patients.22
The BIS has been validated in older children,20–22
but there are conflicting data about the performance of BIS in infants.7,9,21,25–27
Especially, the greater the number of infants is in proportion to children in a BIS study, the poorer the correlations are among the youngest patients.7,9,21,25
Therefore, when analyzing all of these BIS studies, it is obvious that the validation results are valid for older children, but not in infants younger than 6–12 months. Our BIS results are similar to those of previous investigations, i.e.
, the performance of BIS is better in children than in infants.
Recently, a depth-of-hypnosis monitor based on auditory evoked potential (Alaris Auditory Evoked Potential Index; Danmeter A/S, Odense, Denmark)24
was preliminarily evaluated in preschool children receiving sevoflurane anesthesia.21
This method was able to differentiate reliably between awake and anesthetic states during sevoflurane facemask induction. However, there was considerable variation in individual Alaris Auditory Evoked Potential Index values as well as overlap between different clinical conditions.
The different computing times between BIS and Spectral Entropy may have some influence on the results of our study but also on the clinical usefulness of these depth-of-hypnosis devices, because the faster computing of Spectral Entropy may reflect the rapid changes of anesthetic concentrations more accurately.
Adults and pediatric patients differ not only in respect to body size, but also in respect to pharmacokinetics, cooperation, and methodologic properties. Watcha,26
and Constant et al.36
have emphasized age-related differences in the electroencephalogram and the pharmacodynamic effects of anesthetics on the electroencephalogram. There is only a limited amount of data on the effects of anesthetic agents on the electroencephalogram of children of different ages. Well-known differences in the electroencephalograms of infants and teenagers exist, e.g.
, the frequency of posterior basic rhythm; the quality of spindles, vertex waves, and K complexes; and the electroencephalogram changes during drowsiness.37
The maturation and synapse formation of the brain continues after birth, which makes the applicability of the depth-of-hypnosis devices in this age group especially challenging.
It is obvious that the differences in electroencephalographic profiles between age groups pose special requirements to all electroencephalogram-derived methods. Denman et al.20
have already highlighted the problems of validating the BIS monitor in pediatric patients. These include at least the following: the lack of an accepted standard for sedation or sleep, problems distinguishing purposeful from nonpurposeful responses, and ethical problems in the study design. Because children encounter all the same problems of inadequate general anesthesia as adults do, the indications for studying these electroencephalographic devices originally designed for adults are justified. Based on our results, the validity and clinical usefulness of Spectral Entropy and BIS in the pediatric population is not as obvious as in adults. However, the reason for this cannot be concluded from our or any of the previous studies. One can suspect physiologic differences between children and adults as a cause of the observed low performance, but methodologic problems and limitations in study design may have an effect as well.
In our study, the BIS® Sensor was placed above the Entropy Sensor, slightly higher on the forehead than recommended by the manufacturer. Similar placement of sensors was applied also by Davidson et al.25
They had compared the BIS values recorded with optimal versus
higher-than-recommended sensor positioning and concluded that there was a clinically insignificant, small but consistent bias: When the BIS® Sensor was attached above the optimal location, the readings tended to be lower. We found that BIS values could be reliably interpreted, but a true comparison of Spectral Entropy and BIS cannot be concluded from our study. In the adult Spectral Entropy sensor, two of the three electroencephalogram electrodes can be placed in the exact recommended location also in children, because only the distance and the location of the “ground” electrode are fixed in relation to the first electrode on the forehead.
The shortcoming of the current study is the lack of one generally accepted and validated scoring system that would work reliably and detect all of the various levels of sedation and anesthesia in infants and children. Unlike in adults, a relatively slow induction period enabling a precise transition from a clinical state to another is a challenging task in a patient population ranging from 1 month to 15 yr of age. Therefore, in contrast to adult studies, during induction it was not possible to correlate numerous different levels of OAA/S measurements with changing values of SE, RE, and BIS. The OAA/S was originally introduced by Chernik et al.2
The OAA/S consists of several components associated with good validity and reliability and high interrater scores in adult patients. We decided to use a modification of the responsiveness category of the OAA/S, which provides the major contribution to the total OAA/S and has been widely used as a modified OAA/S. Malviya et al.3
have presented the University of Michigan Sedation Scale to pediatric patients. They compared the University of Michigan Sedation Scale with the OAA/S and found that both scales worked reliably in pediatric patients during sedation. The second shortcoming of the current study was the use of a neuromuscular blocking agent, which may prevent the potential responses of the patients. However, to avoid the possibility of analyzing unreliable mOAA/S data, only the periods without the effect of rocuronium during induction and emergence were included. Apart from the intubation dose of rocuronium, most of our patients (17 of 20 infants and 35 of 40 children) did not receive an additional dose of the muscle relaxant. We tried to implement a clinically relevant anesthetic regimen appropriate for pediatric patients and exclude known confounding factors, such as regional anesthesia.27
Nitrous oxide is still commonly used by pediatric anesthesiologists together with sevoflurane and should not have a major effect on our results.
Although investigating the effect of muscle relaxants was not our study objective, it is interesting in the context of RE that includes also electromyographic activity from the frontal muscle. Noxious stimuli may cause electromyographic responses to some extent and produce an increase in RE values in the absence of profound neuromuscular blockade, thus possibly enabling the use of SE − RE difference as a marker of analgesia.
The impact of using steady state versus
non-steady state end-tidal concentration of sevoflurane as a variable of depth of hypnosis is unclear. Both steady state and non-steady state concentrations have been widely used in studies on the level of hypnosis.20–22,24
One might argue that steady state concentrations are more reliable; however, obtaining this kind of value is more difficult especially during constantly changing surgical stimuli, causing rapid change of the level of anesthesia and hypnosis.
Whether the immature brain and electroencephalogram, methodologic study problems, or both are behind the questionable usefulness of all the electroencephalogram-derived depth-of-hypnosis monitors in the younger age groups remains unresolved. More studies are warranted. Attention should be paid to study design in general and the infant age group.
Spectral Entropy may be a useful tool for measuring the hypnotic effects of sevoflurane combined with nitrous oxide in children's central nervous systems. Spectral Entropy seems to behave similarly to the BIS. In infants, the clinical usefulness of both of these electroencephalogram-derived methods is questionable and must be evaluated by further controlled studies.
The authors thank Minna Kymäläinen, R.N. (Clinical Specialist), and Markku Mård, R.N. (Clinical Specialist, GE Healthcare Finland, Helsinki, Finland), for their valuable contribution in the data collection.
1. Guedel AE: Inhalational Anesthesia: A Fundamental Guide. New York, Macmillan, 1937
2. Chernik DA, Gillings D, Laine H, Hendler J, Silver JM, Davidson AB, Schwam EM, Siegel JL: Validity and reliability of the Observer's Assessment of Alertness/Sedation Scale: Study with intravenous midazolam. J Clin Psychopharmacol 1990; 10:244–51
3. Malviya S, Voepel-Lewis T, Tait AR, Merkel S, Tremper K, Naughton N: Depth of sedation in children undergoing computed tomography: Validity and reliability of the University of Michigan Sedation Scale (UMSS). Br J Anaesth 2002; 88:241–5
4. Rampil IJ: A primer for EEG signal processing in anesthesia. Anesthesiology 1998; 89:980–1002
5. Thornton C, Sharpe RM: Evoked responses in anaesthesia. Br J Anaesth 1998; 81:771–81
6. Drummond JC: Monitoring depth of anesthesia. Anesthesiology 2000; 93:876–82
7. Johansen JW, Sebel PS: Development and clinical application of electroencephalographic bispectrum monitoring. Anesthesiology 2000; 93:1336–44
8. Vakkuri A, Yli-Hankala A, Talja P, Mustola S, Tolvanen-Laakso H, Sampson T, Viertiö-Oja H: Time-frequency balanced spectral entropy as a measure of anesthetic drug effect in central nervous system during sevoflurane, propofol, and thiopental anesthesia. Acta Anaesthesiol Scand 2004; 48:145–53
9. Bannister CF, Brosius KK, Sigl JC, Meyer BJ, Sebel PS: The effect of bispectral index monitoring on anesthetic use and recovery in children anesthetized with sevoflurane in nitrous oxide. Anesth Analg 2001; 92:877–81
10. Kreuer S, Biedler A, Larsen R, Altmann S, Wilhelm W: Narcotrend monitoring allows faster emergence and a reduction of drug consumption in propofol-remifentanil anesthesia. Anesthesiology 2003; 99:34–41
11. White PF, Hong Ma, Tang J, Wender RH, Sloninsky A, Kariger R: Does the use of electroencephalographic Bispectral Index or Auditory Evoked Potential Index monitoring facilitate recovery after desflurane anesthesia in the ambulatory setting? Anesthesiology 2004; 100:811–7
12. Song D, Girish PJ, White PF: Titration of volatile anesthetics using Bispectral Index Facilitates recovery after ambulatory anesthesia. Anesthesiology 1997; 87:842–8
13. Bonhomme V, Hans P: Monitoring depth of anaesthesia: Is it worth the effort? Eur J Anaesthesiol 2004; 21:423–8
14. Viertiö-Oja H, Maja V, Särkelä M, Talja P, Tenkanen N, Tolvanen-Laakso H, Paloheimo M, Vakkuri A, Yli-Hankala A, Meriläinen P: Description of the Entropy algorithm as applied in the Datex-Ohmeda S/5 Entropy Module. Acta Anaesthesiol Scand 2004; 48:154–61
15. Shannon CE: A mathematical theory of communication. Bell System Techn J 1948; 27:623–56
16. Johnson RW, Shore JE: Which is the better entropy expression for speech processing: -S log S or log S? IEEE Acoust Speech Signal Proc 1984; 32:129–37
17. Ellerkman RK, Lierman V-M, Alves TM, Wenningmann I, Kreuer S, Wilhelm W, Roepcke H, Hoeft A, Bruhn J: Spectral entropy and Bispectral Index as measures of the electroencephalographic effects of sevoflurane. Anesthesiology 2004; 101:1275–82
18. Vanluchene ALG, Vereecke H, Thas O, Mortier EP, Shafer SL, Struys MMF: Spectral entropy as an electroencephalographic measure of anesthetic drug effect. Anesthesiology 2004; 101:34–42
19. Schmidt GN, Bischoff P, Standl T, Hellstern A, Teuber O, Schulte am Esch J: Comparative evaluation of the Datex-Ohmeda S/5 Entropy Module and the Bispectral Index Monitor during propofol-remifentanil anesthesia. Anesthesiology 2004; 101:1283–90
20. Denman WT, Swanson EL, Rosow W, Ezbicki K, Connors PD, Rosow CE: Pediatric evaluation of the Bispectral Index (BIS) Monitor and correlation of BIS with end-tidal sevoflurane concentration in infants and children. Anesth Analg 2000; 90:872–7
21. Davidson JA, McCann ME, Devavaram P, Auble S, Sullivan LJ, Gillis JM, Laussen PC: The differences in the bispectral index between infants and children during emergence from anesthesia after circumcision surgery. Anesth Analg 2001; 93:326–30
22. McCann ME, Bacsik J, Davidson A, Auble S, Sullivan L, Laussen P: The correlation of bispectral index with end-tidal sevoflurane concentration and haemodynamic parameters in preschoolers. Paediatr Anaesth 2002; 12:519–25
23. O'Kelly SW, Smith DC, Pilkington SN: The auditory evoked potentials and paediatric anaesthesia. Br J Anaesth 1995; 75:428–30
24. Weber F, Bein T, Hobbhahn J, Taeger K: Evaluation of the Alaris Auditory Evoked Potential Index as an indicator of anesthetic depth in preschool children during induction of anesthesia with sevoflurane and remifentanil. Anesthesiology 2004; 101:294–8
25. Davidson AJ, Kim MJ, Sangolt GK: Entropy and bispectral index during anaesthesia in children. Anaesth Intensive Care 2004; 32:485–93
26. Watcha MF: Investigations of the Bispectral Index Monitor in pediatric anesthesia: First things first (editorial). Anesth Analg 2001; 92:805–7
27. Ganesh A, Watcha MF: Bispectral index monitoring in pediatric anesthesia. Curr Opin Anaesthesiol 2004; 17:229–34
28. Arnaout L, Constant I, Lefoll C, Richard P, Murat I: A comparative study between Bispectral Index and Spectral Entropy Indices in children anesthetised with sevoflurane (abstract). Anesthesiology 2004; 101:A1496
29. Lerman J, Sikich N, Kleinman S, Yentis S: The pharmacology of sevoflurane in infants and children. Anesthesiology 1994; 80:814–24
30. Smith WD, Dutton RC, Smith N: Measuring the performance of anesthetic depth indicators. Anesthesiology 1996; 84:38–51
31. Hill A: The possible effect of the aggregation of the molecules of hemoglobin on its dissociation curves. J Physiol 1910; 40:IV–VII
32. Sani O, Shafer SL: MAC Attack (editorial view)? Anesthesiology 2003; 99:1249–50
33. Lu W, Ramsay JG, Bailey JM: Reliability of pharmacodynamic analysis by logistic regression. Anesthesiology 2003; 99:1255–62
34. Dutta S, Matsumoto Y, Ebling WF: Is it possible to estimate the parameters of the sigmoid Emax model with truncated data typical of clinical studies? J Pharm Sci 1996; 85:232–9
35. Constant I: BIS use in pediatric anesthesia: where are we? Can J Anaesth 2004; 51:411–6
36. Constant I, Dubois M-C, Piat V, Moutard M-L, McCue M, Murat I: Changes in electroencephalogram and autonomic cardiovascular activity during induction of anesthesia with sevoflurane compared with halothane in children. Anesthesiology 1999; 91:1604–15
37. Niedermeyer E: Maturation of the EEG: Development of waking and sleep patterns, Electroencephalography: Basic Principles, Clinical Applications and Related Fields, 2nd edition. Edited by Niedermeyer E, Lopes da Silva F. Munich-Baltimore, Urban & Schwarzenberg, 1987, pp 133–58
This article has been cited 7 time(s).
British Journal of AnaesthesiaEEG variables as measures of arousal during propofol anaesthesia for general surgery in children: rational selection and age dependenceBritish Journal of Anaesthesia
AnaesthesistAwareness - A problem in paediatric anaesthesia?Anaesthesist
Anesthesia and AnalgesiaThe Narcotrend Index Indicates Age-Related Changes During Propofol Induction in ChildrenAnesthesia and Analgesia
Anaesthesia, Pain, Intensive Care and Emergency: A.P.I.C.E
Depth of anaesthesia monitoring in children
Anaesthesia, Pain, Intensive Care and Emergency: A.P.I.C.E, ():
British Journal of AnaesthesiaMeasurement of surgical stress in anaesthetized childrenBritish Journal of Anaesthesia
European Journal of Anaesthesiology (EJA)The performance of Bispectral Index in children during equi‐MAC halothane vs. sevoflurane anaesthesiaEuropean Journal of Anaesthesiology (EJA)
© 2006 American Society of Anesthesiologists, Inc.
Publication of an advertisement in Anesthesiology Online does not constitute endorsement by the American Society of Anesthesiologists, Inc. or Lippincott Williams & Wilkins, Inc. of the product or service being advertised.