Evaluation of a Cerebral Oximeter as a Monitor of Cerebral Ischemia during Carotid Endarterectomy
Samra, Satwant K. M.D.*; Dy, Eric A.†; Welch, Kathy M.S., M.P.H.‡; Dorje, Pema M.D.§; Zelenock, Gerald B. M.D.∥; Stanley, James C. M.D.∥
Background: Stroke is an important contributor to perioperative morbidity and mortality associated with carotid endarterectomy (CEA). This investigation was designed to compare the performance of the INVOS-3100 cerebral oximeter to neurologic function, as a means of detecting cerebral ischemia induced by carotid cross-clamping, in patients undergoing carotid endarterectomy with cervical plexus block.
: Ninety-nine patients undergoing 100 CEAs with regional anesthesia (deep or superficial cervical plexus block) were studied. Bilateral regional cerebrovascular oxygen saturation (rSO2
) was monitored using the INVOS-3100 cerebral oximeter. Patients were retrospectively assigned to one of two groups: those in whom a change in mental status or contralateral motor deficit was noted after internal carotid clamping (neurologic symptoms; n = 10) and those who did not show any neurologic change (no neurologic symptoms; n = 90). Data from 94 operations (neurologic symptoms = 10 and no neurologic symptoms = 84) were adequate for statistical analyses for group comparisons. A relative decrease in ipsilateral rSO2
after carotid occlusion (calculated as a percentage of preocclusion value) during all operations (n = 100) was also calculated to determine the critical level of rSO2
decrease associated with a change in neurologic function.
: The mean (± SD) decrease in rSO2
after carotid occlusion in the neurologic symptoms group (from 63.2 ± 8.4% to 51.0 ± 11.6%) was significantly greater (P = 0.0002) than in the no neurologic symptoms group (from 65.8 ± 8.5% to 61.0 ± 9.3%). Logistic regression analysis used to determine if a change in rSO2
, calculated as a percentage of preclamp value, could be used to predict change in neurologic function was highly significant (likelihood ratio chi-square = 13.7;P
= 0.0002). A 20% decrease in rSO2
reading from the preclamp baseline, as a predictor of neurologic compromise, resulted in a sensitivity of 80% and specificity of 82.2%. The false-positive rate using this cutoff point was 66.7%, and the false-negative rate was 2.6%, providing a positive predictive value of 33.3% and a negative predictive value of 97.4%.
Conclusion: Monitoring rSO2 with INVOS-3100 to detect cerebral ischemia during CEA has a high negative predictive value, but the positive predictive value is low.
RECENT multicenter trials 1–3
have clearly established that carotid endarterectomy (CEA) is beneficial in symptomatic as well as asymptomatic patients with high-grade and moderate carotid stenoses. Nevertheless, perioperative stroke occurs in 2–3% of patients undergoing CEA, 3–5
usually caused by either cerebral ischemia or embolism during surgery. Routine insertion of a shunt to prevent or minimize cerebral ischemia may increase the likelihood of stroke caused by embolism. It is therefore important to identify patients who are at risk for developing cerebral ischemia during carotid occlusion and who are likely to benefit from shunting, before insertion of a shunt. Despite several existing monitoring methods, the assessment of cerebral ischemia during CEA has been unreliable 5,6
in predicting perioperative stroke. New monitoring modalities need to be evaluated.
Cerebral oximetry, based on the principles of near-infrared spectroscopy as first described by Jobsis, 7
is a noninvasive technique to monitor cerebral oxygenation. Although this technique was described nearly two decades ago, the instrumentation to make it a clinically useful monitor is still being perfected. Early clinical investigations with this technology used prototype instruments. Recent research has been devoted to the development of an instrument that is more simple to use. This has resulted in the introduction of a commercially available cerebral oximeter, INVOS 3100 (Somanetics Corp., Troy, MI). Several validation studies using this cerebral oximeter to monitor brain oxygenation during conditions of hypoxemia, hypercapnia, and hypocapnia in awake volunteers have been published. 8,9
Concerns regarding the inability of this monitor to eliminate contamination from extracranial tissues were raised by early studies. 10,11
More recent clinical investigations 12,13
have suggested that INVOS-3100 primarily tracks changes in intracranial circulation. However, the ability of this monitor to detect acute cerebral ischemia in humans has not been fully investigated. The majority of clinical investigations with this cerebral oximeter have been conducted in patients during general anesthesia during CEA and deep hypothermic circulatory arrest. Performance of this device as a monitor of cerebral ischemia has been compared with other means of assessing cerebral ischemia such as electroencephalography, 14
somatosensory evoked potentials (SSEPs), 15
transcranial Doppler, and measurements of jugular bulb venous oxygen saturation. 16,17
All of these monitors provide indirect evidence of cerebral ischemia and have been used to guide decision for shunt insertion, but their dependability in predicting neurologic outcome has not been systematically investigated.
The present study was designed to evaluate the performance of the INVOS-3100 cerebral oximeter to detect development of cerebral ischemia during carotid artery cross-clamping, as evidenced by changes in mental status or development of motor deficits in awake patients undergoing CEA with regional anesthesia.
Ninety-nine adults (51 men, 48 women) between the ages of 43 and 90 yr who underwent 100 CEAs over a 4-yr time period were studied. All patients gave informed consent to participate in the study according to the guidelines approved by the Institutional Review Board of the University of Michigan Health System. All patients were scheduled to undergo CEA with regional anesthesia by one of three attending vascular surgeons. All patients had high-grade carotid artery stenoses (> 70%) documented by preoperative carotid duplex sonography.
An ipsilateral cervical plexus block (superficial or deep) was performed using 0.375% bupivacaine. A radial artery was cannulated for continuous monitoring of blood pressure. Electrocardiogram (lead V5
), peripheral hemoglobin saturation, and regional cerebrovascular saturation (rSO2
) were continuously monitored during operation. Two cerebral oximeters (model INVOS-3100) were used for simultaneous, bilateral rSO2
monitoring. Details of the principles of cerebral oxymetry used by INVOS-3100 have been described in a previous report. 18
This oximeter monitors only changes in cerebral oxygen saturation from an unknown baseline, recording those as numerical values of rSO2
index. INVOS-3100 is thus suitable as a trend monitor only.
The cerebral oximeter sensors were applied to the forehead, one on either side of the midline, such that the light transmitters were placed at least 3 cm from the midline. The sensors were covered with an adhesive cover to shield them from ambient light. Adhesive tape was used to hold the sensors firmly in place and assure contact with skin throughout the operation. The numerical rSO2 readings recorded at 1-min intervals were stored on computer disks for offline data analysis at a later time. The signal average time for each numerical value was 4 s.
Minimal sedation with 1–2 mg midazolam, administered intravenously, was used before performing the cervical plexus block in a preoperative holding room approximately 30 min before the skin incision. After all monitoring was established, surgery proceeded with an additional 25–50 μg fentanyl administered intravenously if the patient appeared apprehensive and requested more sedation. No intravenous sedatives or analgesics were administered for at least 10 min before or during the period of carotid artery occlusion. During the time of occlusion, neurologic function was assessed at 5-min intervals by determining the patient’s ability to respond to verbal commands and exhibit normal motor strength in the contralateral upper extremity. Inability to respond appropriately to verbal commands, unconsciousness, slurring of speech, or development of motor weakness was used as criteria for insertion of shunt. Duration of carotid cross-clamp, development of change in neurologic function, as well as the need for and time of shunt insertion and removal were recorded. It should be emphasized that no change in clinical management occurred based on the rSO2 readings. The surgeons were not aware of the rSO2 values during the operation.
Regional Cerebrovascular Oxygen Saturation Data Analysis
The duration of the CEA procedure was divided into three phases: (1) preclamp: from skin incision to carotid occlusion; (2) cross-clamp: from carotid artery clamp application to clamp release; and (3) postclamp: from clamp removal to skin closure. Mean values of rSO2 for preclamp, cross-clamp, and postclamp phases for each hemisphere in each patient were calculated from the readings recorded at 1-min intervals. After completion of the study, patients were assigned to one of two groups: those who did not show a change in neurologic function (no neurologic symptoms, n = 90), and those who did (neurologic symptoms, n = 10). Intersubject variability in rSO2 index values is well known and was noticed in this study. To facilitate comparison of rSO2 changes after carotid cross-clamp among all patients, and to determine the magnitude of rSO2 change that was associated with a change in neurologic function, the rSO2 data were normalized by calculating a percentage change in rSO2 reading during cross-clamp periods in each patient according to the formula:
Percent change = [(mean rSO2 reading preclamp − minimum rSO2 reading cross-clamp)/mean rSO2 reading preclamp] × 100
Thus, a change of rSO2 reading from a mean preclamp value of 70% saturation to a minimal (after cross-clamp) of 63% saturation according to this formula would represent a 10% decrease in rSO2 reading.
The phase of operation (preclamp, cross-clamp, and postclamp), side of measurement (ipsilateral and contralateral), and group designation (neurologic symptoms vs. no neurologic symptoms) effect on mean rSO2 values were studied using three-way repeated-measures analysis of variance. Numerical data for rSO2 from the 10 CEAs in which a change in neurologic function developed (neurologic symptoms group) and the 84 operations in which no change in neurologic function occurred (no neurologic symptoms group) were compared. The numerical data for contralateral hemispheres from six operations in the no neurologic symptoms group were incomplete or not retrievable because of defective data disks. Raw data (rather than normalized percentage change) for rSO2 values were used for the analysis of variance. Phase and side were included in the model as within-subjects effects and group as a between-subjects effect. Because of unequal variances that were observed for the neurologic symptoms and no neurologic symptom groups, the repeated-measures analysis of variance incorporated this feature. All post hoc comparisons were performed using the Tukey-Kramer adjustment for multiple comparisons. A Wilcoxon rank sum nonparametric test was used to compare the two groups for the duration of cross-clamp, because it could not reasonably be assumed to be normally distributed.
Normalized data from ipsilateral hemispheres from all operations (n = 100) were subjected to logistic regression analysis to determine if it was possible to estimate the probability of change in neurologic function, based on a decrease in rSO2 during cross-clamp relative to the mean rSO2 reading during the preclamp phase. Sensitivity and specificity, false-positive rate and false-negative rate, as well as the positive and negative predictive value of a 20% relative decrease in rSO2 as a predictor of cerebral ischemia were calculated. The SAS statistical software package (Proprietary software release 6.12, SAS Institute, Inc., Cary NC) was used for statistical analyses.
Carotid endarterectomy was successfully completed with regional anesthesia in all patients. No neurologic change was observed in 90 operations, whereas neurologic changes occurred in 10 operations after carotid clamping. Neurologic change resolved after insertion of an intravascular shunt in seven of these patients. In the three remaining patients who developed a neurologic change, the surgeon anticipated a very short clamp time and elected to complete the operation without use of a shunt. Data from all 10 patients with neurologic changes, including the three latter patients, were included in the neurologic symptoms group for statistical analyses.
Ninety-seven patients left the hospital without a clinically detectable, new neurologic deficit. One patient who had no intraoperative problems (no neurologic symptoms group), developed significant hemodynamic instability in the postoperative period, necessitating insertion of a pulmonary artery catheter, and subsequently died. Another patient, an 80-yr-old woman (neurologic symptoms group) developed hemiplegia. She presented with symptomatic high-grade stenosis of right internal carotid artery and evidence of generalized cerebral atrophy on magnetic resonance imaging. Bilateral rSO2
readings were within normal limits during the preclamp period but after occlusion of carotid artery, ipsilateral rSO2
value decreased by 29.3% (from 55 to 39) within 2 min (fig. 1
). The patient became aphasic, unresponsive, and apneic, resulting in intravascular shunt being quickly inserted. One minute later, the patient responded to verbal commands, and the rSO2
increased to mid-40s and remained at that level as the CEA was completed. She developed severe hypotension after administration of protamine, which progressed to cardiac electromechanical dissociation necessitating endotracheal intubation and cardiopulmonary resuscitation as the skin incision was being closed. On awakening, the patient had decreased responsiveness to verbal commands and developed left hemiplegia. A computed tomography scan revealed an ischemic infarct in the right hemisphere in the watershed distribution between ipsilateral anterior and middle cerebral as well as between the middle and posterior cerebral arteries.
The numerical values (mean ± SD) for duration of cross-clamp and rSO2
for 94 patients, with bilateral monitoring in the three phases of operation are shown in table 1
. The duration of cross-clamp time was longer in neurologic symptoms group, but this difference was not statistically significant. There was no significant difference between the two groups when either ipsilateral or contralateral values of rSO2
during the three phases were compared. Results of pairwise comparisons of interest from this analysis were revealing (table 2
). There were no significant changes in rSO2
in the contralateral hemispheres in either group. However, on the ipsilateral side in both groups there was a significant decrease in rSO2
during the cross-clamp phase when compared with the preclamp and postclamp phases. The decrease in rSO2
from the preclamp to cross-clamp period on the ipsilateral side was significantly greater (12.2%vs.
= 0.0001) in the neurologic symptoms group. The change from preclamp to cross-clamp period (2.2% decrease vs.
1.7% increase) on the contralateral side between the two groups was also significant (P
The relative decrease (percent change from mean preclamp value of rSO2
) observed after carotid occlusion in 100 operations is shown in figure 2
. A logistic regression analysis was used to determine if a change in rSO2
(relative decrease) after carotid occlusion could be used to predict development of neurologic deterioration. It was decided to determine a reasonable cutoff point and calculate the sensitivity and specificity of this cutoff point. This logistic regression was highly significant (likelihood ratio chi-square = 13.7, 1 df
= 0.0002). The sensitivity and specificity of several cutoff points were analyzed based on the logistic regression model probabilities. The model probability for the best cutoff point was then translated back in to the corresponding decrease in rSO2
. A cutoff point of 20% relative decrease in rSO2
gave the best sensitivity and specificity results. This cutoff value resulted in a sensitivity of 80% (8 of 10 patients who developed a neurologic change had rSO2
decrease > 20%, whereas two did not). The specificity of this cutoff point was 82.2% (74 of the 90 patients who did not develop a neurologic change had < 20% relative decrease of rSO2
, whereas in 16 patients rSO2
decrease was > 20%). The false-positive rate for this cutoff was 66.7% (16 of the 24 patients predicted to develop neurologic dysfunction based on the cutoff did not develop one) and a false-negative rate of 2.6% (2 of 76 patients who were predicted to not develop neurologic dysfunction actually did develop one). The positive–negative predictive values of a 20% decrease in rSO2
(compared with preclamp value) were calculated as 97.4% negative predictive value (74 of the 76 patients with < 20% decrease did not show neurologic deterioration) and a 33.3% positive predictive value (only 8 of 24 patients with > 20% decrease developed a neurologic change).
This study was designed to evaluate the performance of the INVOS-3100 cerebral oximeter to detect development of cerebral ischemia. Clinical signs of cerebral ischemia developed during 10 operations and were temporally related to occlusion of internal carotid artery. An intravascular shunt was used in seven of these patients, whereas surgery was completed without shunt in three in whom a brief period of occlusion was anticipated. No postoperative, new neurologic deficit developed in any of these three patients, demonstrating that a short period of cerebral ischemia may be tolerated without permanent sequelae.
The intersubject variability of rSO2
values in the preclamp period and variable decreases in rSO2
observed after cross-clamp in the current study is in agreement with that reported in previous studies. 18,19
The change in rSO2
after cross-clamp was significantly greater in patients who developed neurologic symptoms when compared with those who did not. This finding is in agreement with a recent study comparing cerebral oximetry with changes in amplitude of SSEPs during CEA in 29 patients. 15
Cho et al.15
reported that the decrease in rSO2
was greater than 10 units in patients who showed a significant decrease in SSEP amplitude and ranged between −2 and −6.1 units in those who did not have a significant change in SSEPs. They concluded that a decrease of > 10 units or an rSO2
reading < 50 (as measured by INVOS-3100) is indicative of cerebral ischemia. The magnitude of the mean decrease in rSO2
in the current study (4.8 units in patients without neurologic dysfunction and 12.2 units in those with neurologic dysfunction) was similar.
The intersubject variability in rSO2 index could be addressed by normalizing the data and recording the decrease in rSO2 after cross-clamp as a percentage of preclamp value. Such normalization will not only account for intersubject variability but will also allow comparisons between different investigations.
Data from the current investigation revealed that the use of rSO2
monitoring and percentage change in rSO2
after cross-clamp to predict development of neurologic deficit has significant limitations. In this study, observed rSO2
values were similar in the two groups in all three phases of operation (table 1
). On post hoc
analysis, the difference in ipsilateral rSO2
decrease after carotid occlusion was significantly greater in patients in the neurologic symptoms group. However, within subjects, a cutoff point of 20% decrease from preclamp value, which provided the best sensitivity (80%) and specificity (82.2%) in awake patients, had a negative predictive value of 97.4% and a positive predictive value of only 33.3%. There are several possible explanations for false-negative and false-positive results in our study: (1) the sensors of cerebral oximeter were applied to the hairless scalp overlying the frontal lobe, whereas the most vulnerable area of brain for ischemia–embolism is in the distribution of middle cerebral artery. The cerebral blood flow changes are heterogenous in nature, and ischemia may develop in parietal lobes (distribution of middle cerebral artery) without a simultaneous change in rSO2
monitored over the region of frontal lobes. (2) The neurologic examination was crude and limited (for practical reasons) and might have missed evidence of cerebral ischemia. (3) There was variable contribution of extracranial circulation to monitored rSO2
A comparison of rSO2
monitoring with other currently accepted monitors of cerebral ischemia is desirable but difficult to achieve because an intervention (insertion of shunt) is generally made whenever monitoring suggests development of cerebral ischemia in anesthetized patients. Only a few studies have determined sensitivity and specificity of different monitoring modalities using development of stroke as primary outcome variable. Blume et al.20
have evaluated the significance of electroencephalography changes in 176 patients undergoing CEA with general anesthesia, in whom a shunt was not inserted even if electroencephalography changes were observed. They had no false-negative results but had a 90.9% false-positive rate (only 2 of 22 patients predicted to require a shunt actually had a stroke). Lam et al.21
compared the sensitivity and specificity of electroencephalography and SSEPs as a monitor of cerebral ischemia in 67 patients during CEAs with general anesthesia, without use of a shunt, even if electroencephalography or SSEP changes were observed. They concluded that the relative sensitivity and specificity for electroencephalography and SSEPs in detecting postoperative stroke (without any intervention) was 50% and 92% for electroencephalography and 100% and 94% for SSEPs, respectively. However, our findings cannot be directly compared with these studies because different end points (temporary ischemia vs. stroke) were used.
When compared with intraoperative electroencephalography and SSEP monitoring, rSO2
monitoring has the advantages of being easy to use and less expensive. One major limitation in accepting rSO2
as a replacement for electroencephalography and SSEP monitoring is that changes in electroencephalography and SSEPs have been shown to correlate directly with changes in cerebral blood flow, 22–24
whereas no such data for rSO2
are currently available. The evidence that rSO2
corresponds with changes in cerebral blood flow is indirect, provided by studies correlating changes in electroencephalography, SSEPs, and transcranial Doppler, with those in rSO2
The results of the current investigation, which suggest a 20% relative decrease in rSO2
as a predictor of cerebral ischemia, are comparable to those recently reported in a study by Roberts et al.25
They monitored 45 patients undergoing 50 CEAs with regional anesthesia. In their study, four patients who required a shunt had a > 27% decrease in rSO2
after carotid cross-clamp, and in the remaining 41 patients the rSO2
decrease varied between 0% and 23%. They suggested that a decrease of ≥ 27% in rSO2
should suggest the need for a shunt, and a decrease < 27% should not require a shunt. In other words, they did not have any false-positive or false-negative results. In our study, using a 20% decrease as the cutoff point, we had a false-negative rate of 2.6% and a false-positive rate of 66.7%. A possible explanation for this difference is that the duration of cross-clamp was shorter in their study. Patients may tolerate cerebral ischemia of short duration without clinical development of neurologic symptoms. Three patients in our study who developed neurologic symptoms and completed surgery without a shunt had short clamp times and did not have any permanent sequelae, clearly demonstrating that it is a combination of both the magnitude and duration of ischemia that leads to neurologic deficit.
One of the limitations of currently available technology is that the sensors are applicable only to the skin devoid of hair follicles. Development of sensors that can be applied to the scalp overlying the area of distribution of middle cerebral artery and monitoring from multiple sensors may further improve the performance of this cerebral oximeter as a monitor of cerebral ischemia.
In conclusion, this investigation suggests that a change in rSO2 after carotid cross-clamp during CEA can be used as a trend monitor for predicting development of cerebral ischemia only to a limited extent. It is not possible to specify an absolute rSO2 reading as the critical value below which cerebral ischemia may develop. A relative decrease of ≥ 20% (from the preclamp rSO2 values) after carotid occlusion has a high negative predictive value, i.e., if rSO2 does not decrease, ischemia is unlikely, but a low positive predictive value, i.e., a decrease in rSO2, may not always indicate cerebral ischemia. When used in this manner, rSO2 monitoring has a sensitivity and specificity similar to electroencephalography and SSEP monitoring with a low (2.6%) false-negative rate but high (66.7%) false-positive rate. Development of oximeter sensors that can be applied to the scalp over the parietal region may further enhance the clinical utility of this monitoring technique.
1. North American Symptomatic Carotid Endarterectomy Trial Collaborators: Beneficial effect of carotid endarterectomy in symptomatic patients with high-grade carotid stenosis. N Engl J Med 1991; 325: 445–53
2. Barnett HJM, Taylor W, Eliasziw M, Fox AJ, Haynes RB, Rankin RN, Glagett GP, Hachinski VC, Sackett DL, Thorpe KE, Math M, Meldrum HE: Benefit of carotid endarterectomy in patients with symptomatic moderate or severe stenosis. N Engl J Med 1998; 339: 1415–25
3. Executive Committee for the Asymptomatic Carotid Arthrosclerosis Study: Endarterectomy for asymptomatic carotid artery stenosis. JAMA 1995; 273: 1421–8
4. Berman SS, Bernhard M, Erly WK, McIntyre KE, Erodes LS, Hunter GC: Critical carotid artery stenosis: Diagnosis, timing of surgery, and outcome. J Vasc Surg 1994; 20: 499–510
5. Gewertz BL, McCaffrey MT: Intraoperative monitoring during carotid endarterectomy. Curr Probl Surg 1987; 24: 475–532
6. Horsch S, DeVleeschawer P, Ktandis K: Intraoperative assessment of cerebral ischemia during carotid surgery. J Cardiovasc Surg 1990; 31: 599–602
7. Jobsis FF: Noninvasive, infrared monitoring of cerebral and myocardial oxygen sufficiency and circulatory parameters. Science 1977; 198: 1264–7
8. Pollard V, Prough DS, DeMelo E, Deyo DG, Uchida T, Stoddart F: Validation in volunteers of a hear-infrared spectroscope for monitoring brain oxygenation in vivo. Anesth Analg 1996; 82: 269–77
9. Henson LC, Calalang C, Temp JA, Ward DS: Accuracy of a cerebral oximeter in healthy volunteers under conditions of isocapnic hypoxia. A nesthesiology 1998; 88: 58–65
10. Germon TJ, Kane NM, Manara AR, Nelson RJ: Near infrared spectroscopy in adults: Effects of extracranial ischemia on estimation of cerebral oxygenation. Br J Anesth 1994; 73: 503–6
11. Germon TJ, Young AER, Manara AR, Nelson RJ: Extracerebral absorption of near infrared light influences the detection of increased cerebral oxygenation monitored by infrared spectroscopy. J Neurol Neurosurg Psych 1995; 58: 477–99
12. Hongo K, Kobayeshi S, Okudora H, Hokama M, Nakagawa F: Noninvasive cerebral optical spectroscopy: Depth resolved by measurements of cerebral hemodynamics using Indocyanine green. Neurol Res April 1995: 17:89–93
13. Grubhofer G, Lassnigg A, Manlik F, Marx E, Trubel W, Hiesmayr M: The contribution of extracranial blood oxygenation on near-infrared spectroscopy during carotid thromboendarterectomy. Anaesthesia 1997; 52: 116–20
14. de Letter JA, Sie HT, Thomas BM, Moll FL, Algra A, Eikelbloom BC, Ackerstaff RG: Near-infrared reflected spectroscopy and electroencephalography during carotid endarterectomy—In search of a new shunt criterion. Neurol Res 1998; 20: S23–7
15. Cho H, Nemoto EM, Yonas H, Blazer J, Sclabassi RJ: Cerebral monitoring by means of oximetry and somatosensory evoked potentials during carotid endarterectomy. J Neurosurg 1998; 89: 533–8
16. Williams IM, Picton A, Farrel A, Mead GE, Mortimer AJ, McColloum CN: Light reflective cerebral oximetry and jugular bulb venous oxygen saturation during carotid endarterectomy. Br J Surg 1994; 81: 1292–5
17. Duncan LA, Ruckley CV, Wildsmith JAW: Cerebral oximetry: A useful monitor during carotid artery surgery. Anesthesia 1995; 50: 1041–5
18. Samra SK, Dorje P, Zelenock GB, Stanley JC: Cerebral oximetry in patients undergoing carotid endarterectomy under regional anesthesia. Stroke 1996; 27: 49–55
19. Beese U, Langer H, Lang W, Dinkel M: Comparison of near-infrared spectroscopy and somatosensory evoked potentials for the detection of cerebral ischemia during carotid endarterectomy. Stroke 1998; 29: 2032–7
20. Blume WT, Ferguson GG, McNeill DK: Significance of EEG changes at carotid endarterectomy. Stroke 1986; 17: 891–7
21. Lam AM, Manninen PH, Ferguson GG, Nantau W: Monitoring electrophysiologic function during carotid endarterectomy: A comparison of somatosensory evoked potentials and conventional electroencephalogram. A nesthesiology 1991; 75: 15–21
22. Sundt T, Sharbrough FW, Piepgras DG, Kearns TP, Messick JMJ, O’Fallon WM: Correlation of cerebral blood flow and electroencephalographic changes during carotid endarterectomy—With results of surgery and hemodynamics of cerebral ischemia. Mayo Clinic Proc 1981; 56: 533–43
23. Messick JM, Casement B, Sharbrough FW, Milde LN, Michenfelder JD, Sundt TM Jr: Correlation of regional cerebral blood flow (rCBF) with EEG changes during isoflurane anesthesia for carotid endarterectomy: Critical rCBF. A nesthesiology 1987; 66: 344–9
24. Symon L, Hargardine J, Zawirski M, Branston NM: Central conduction time as an index of ischemia in subarachnoid hemorrhage. J Neurol Sci 1979; 44: 95–103
25. Roberts KW, Crnkovic AP, Linneman LJ: Near infrared spectroscopy detects critical cerebral hypoxia during carotid endarterectomy in awake patients [abstract]. A nesthesiology 1998; 9 (3A):A933
This article has been cited 86 time(s).
Journal of Shoulder and Elbow SurgeryCerebral desaturation events during shoulder arthroscopy in the beach chair position: patient risk factors and neurocognitive effectsJournal of Shoulder and Elbow Surgery
The differences in cerebral oxygenation among patients undergoing thyroidectomy in different positions and comparison of their effects on nausea-vomiting
Canadian Journal of Anesthesia-Journal Canadien D AnesthesieCerebral monitors versus regional anesthesia to detect cerebral ischemia in patients undergoing carotid endarterectomy: a meta-analysisCanadian Journal of Anesthesia-Journal Canadien D Anesthesie
Journal of Critical CareCerebral desaturation events in the intensive care unit following cardiac surgeryJournal of Critical Care
Journal of Cardiothoracic and Vascular AnesthesiaPro: Cerebral Oximetry Should be a Routine Monitor During Cardiac SurgeryJournal of Cardiothoracic and Vascular Anesthesia
World NeurosurgeryIntraoperative Dual Monitoring During Carotid Endarterectomy Using Motor Evoked Potentials and Near-Infrared SpectroscopyWorld Neurosurgery
Expert Review of Medical DevicesCerebral oxygenation monitoring: near-infrared spectroscopyExpert Review of Medical Devices
Neurologic ClinicsAnesthetic issues and perioperative blood pressure management in patients who have cerebrovascular diseases undergoing surgical proceduresNeurologic Clinics
Journal of International Medical Research
The effect of gynaecological laparoscopic surgery on cerebral oxygenation
Journal of International Medical Research, 34(5):
Surgical NeurologyDetection of hemodynamic transient ischemic attack during hemodialysis with near-infrared monitoring in a patient with internal carotid artery occlusionSurgical Neurology
Journal of Oral and Maxillofacial SurgeryNitroglycerin- and nicardipine-induced hypotension does not affect cerebral oxygen saturation and postoperative cognitive function in patients undergoing orthognathic surgeryJournal of Oral and Maxillofacial Surgery
Anesthesia and AnalgesiaIncreased Oxygen Administration Improves Cerebral Oxygenation in Patients Undergoing Awake Carotid SurgeryAnesthesia and Analgesia
South African Journal of Surgery
Cerebral monitoring during carotid endarterectomy - a comparison between electroencephalography, transcranial cerebral oximetry and carotid stump pressure
South African Journal of Surgery, 45(2):
British Journal of AnaesthesiaReduced cerebral oxygen saturation measured by absolute cerebral oximetry during thoracic surgery correlates with postoperative complicationsBritish Journal of Anaesthesia
Anesthesia and AnalgesiaIs bilateral monitoring of cerebral oxygen saturation necessary during neonatal aortic arch reconstruction?Anesthesia and Analgesia
International Journal of Clinical PracticeThe effect of acute normovolemic haemodilution on cerebral oxygenationInternational Journal of Clinical Practice
Journal of Bone and Joint Surgery-American VolumeNear-Infrared Spectroscopy in Lower Extremity TraumaJournal of Bone and Joint Surgery-American Volume
Annals of Thoracic SurgeryHigh flow rates during modified ultrafiltration decrease cerebral blood flow velocity and venous oxygen saturation in infantsAnnals of Thoracic Surgery
Journal of Clinical AnesthesiaThe effect of hemodynamic changes induced by propofol induction on cerebrat oxygenation in young and elderly patientsJournal of Clinical Anesthesia
Annals of Thoracic SurgeryRandomized controlled trial of pericardial blood processing with a cell-saving device on neurologic markers in elderly patients undergoing coronary artery bypass graft surgeryAnnals of Thoracic Surgery
Neurocritical CareBilateral near infrared spectroscopy in space-occupying middle cerebral artery strokeNeurocritical Care
European Journal of Vascular and Endovascular SurgeryNear infrared spectroscopy monitoring during carotid endarterectomy: Which threshold value is critical?European Journal of Vascular and Endovascular Surgery
Anesthesia and AnalgesiaContinuous monitoring of cerebral oxygen saturation in elderly patients undergoing major abdominal surgery minimizes brain exposure to potential hypoxiaAnesthesia and Analgesia
Journal Des Maladies Vasculaires
Cerebral protection in carotid surgery: current views
Journal Des Maladies Vasculaires, 27(1):
Acta Anaesthesiologica Scandinavica
Use of cerebral oximetry to detect and manage cerebral desaturation with a rapidly expanding neck hematoma
Acta Anaesthesiologica Scandinavica, 46(5):
Pediatric AnesthesiaCerebral oxygenation monitoring using near infrared spectroscopy during controlled hypotensionPediatric Anesthesia
British Journal of AnaesthesiaComparison of changes in jugular venous bulb oxygen saturation and cerebral oxygen saturation during variations of haemoglobin concentration under propofol and sevoflurane anaesthesiaBritish Journal of Anaesthesia
Journal of Cardiothoracic and Vascular AnesthesiaCon: All cardiac surgical patients should not have intraoperative cerebral oxygenation monitoringJournal of Cardiothoracic and Vascular Anesthesia
European Journal of Cardio-Thoracic SurgeryCerebral regional oxygenation during aortic coarctation repair in pediatric populationEuropean Journal of Cardio-Thoracic Surgery
Journal of Vascular SurgeryCerebral oximetry does not correlate with electroencephalography and somatosensory evoked potentials in determining the need for shunting during carotid endarterectomyJournal of Vascular Surgery
Journal of Clinical AnesthesiaA clinical evaluation of near-infrared cerebral oximetry in the awake patient to monitor cerebral perfusion during carotid endarterectomyJournal of Clinical Anesthesia
Pediatric PulmonologyCerebral oximetry improves detection of sickle cell patients at risk for nocturnal cerebral hypoxiaPediatric Pulmonology
New technology for noninvasive brain monitoring: continuous cerebral oximetry
Minerva Anestesiologica, 72():
Journal of Clinical NeuroscienceThe effectiveness of regional cerebral oxygen saturation monitoring using near-infrared spectroscopy in carotid endarterectomyJournal of Clinical Neuroscience
Prolonged retraction on the normal common carotid artery induced lethal stroke after cervical spine surgery
Annals of Vascular SurgerySignificant correlation between cerebral oximetry and carotid stump pressure during carotid endarterectomyAnnals of Vascular Surgery
Journal of Cardiothoracic and Vascular AnesthesiaPro: Near-infrared spectroscopy should be used for all cardiopulmonary bypassJournal of Cardiothoracic and Vascular Anesthesia
Canadian Journal of Neurological Sciences
Carotid endarterectomy: A review
Canadian Journal of Neurological Sciences, 31(1):
Journal of Thoracic and Cardiovascular SurgeryRegional cerebral saturation monitoring with near-infrared spectroscopy during selective antegrade cerebral perfusion: Diagnostic performance and relationship to postoperative strokeJournal of Thoracic and Cardiovascular Surgery
Journal of Cardiothoracic and Vascular AnesthesiaPro: All cardiac surgical patients should have intraoperative cerebral oxygenation monitoringJournal of Cardiothoracic and Vascular Anesthesia
Annals of Vascular SurgeryBispectral Index Changes in Carotid SurgeryAnnals of Vascular Surgery
Vascular and Endovascular Surgery
Correlation of Cerebral Oximetry With Internal Carotid Artery Stump Pressures in Carotid Endarterectomy
Vascular and Endovascular Surgery, 44(4):
ResuscitationNear-infrared spectroscopy: a tool to monitor cerebral hemodynamic and metabolic changes after cardiac arrest in ratsResuscitation
Journal of Biomedical OpticsDevelopment and validation of a multiwavelength spatial domain near-infrared oximeter to detect cerebral hypoxia-ischemiaJournal of Biomedical Optics
Neurosurgery Clinics of North AmericaPerioperative uses of transcranial perfusion monitoring (Reprinted from Neurologic Clinics, November, 2006)Neurosurgery Clinics of North America
Pediatric AnesthesiaMonitoring cerebral oxygenation in a pediatric patient undergoing surgery for vascular ringPediatric Anesthesia
Acta Anaesthesiologica ScandinavicaThe effect of esmolol-induced controlled hypotension in combination with acute normovolemic hemodilution on cerebral oxygenationActa Anaesthesiologica Scandinavica
Surgical NeurologyCognitive impairment associated with intraoperative and postoperative hypoperfusion without neurologic deficits in a patient undergoing carotid endarterectomySurgical Neurology
Non-invasive monitoring of haemoglobin concentration in paediatric surgical patients using near-infrared spectroscopy
Experimental Biology and Medicine
New insight into the mechanism of cardiovascular dysfunction in the elderly: Transfer function analysis
Experimental Biology and Medicine, 230(8):
British Journal of AnaesthesiaCarotid endarterectonlyBritish Journal of Anaesthesia
Journal of Cardiothoracic SurgeryBaseline cerebral oximetry values in cardiac and vascular surgery patients: a prospective observational studyJournal of Cardiothoracic Surgery
British Journal of AnaesthesiaSignificant decrease of cerebral oxygen saturation during single-lung ventilation measured using absolute oximetryBritish Journal of Anaesthesia
Canadian Journal of Neurological Sciences
Comparison of monitoring techniques for intraoperative cerebral ischemia
Canadian Journal of Neurological Sciences, 31(3):
Anesthesia and AnalgesiaAn evaluation of bilateral monitoring of cerebral oxygen saturation during pediatric cardiac surgeryAnesthesia and Analgesia
Anesthesia and AnalgesiaThe effect of desflurane and sevoflurane on cerebral oximetry under steady-state conditionsAnesthesia and Analgesia
Annals of Emergency MedicineEvaluation of Cerebral Oxygenation During Procedural Sedation in Children Using Near Infrared SpectroscopyAnnals of Emergency Medicine
Cardiovascular and Interventional RadiologyMassive hemorrhage from internal carotid artery pseudoaneurysm successfully treated by transcatheter arterial embolization with assessment of regional cerebral oxygenationCardiovascular and Interventional Radiology
Perfusion-UkInfluence of hematocrit and pump prime on cerebral oxygen saturation in on-pump coronary revascularizationPerfusion-Uk
European Respiratory JournalSome factors affecting cerebral tissue saturation during obstructive sleep apnoeaEuropean Respiratory Journal
Journal of Neurosurgery
Evaluation of extracranial-intracranial arterial bypass function by using near-infrared spectroscopy
Journal of Neurosurgery, 99(2):
Anesthesia and Analgesia
Regional cerebral oxygen saturation is a sensitive marker of cerebral hypoperfusion during orthotopic liver transplantation
Anesthesia and Analgesia, 99(2):
Journal of Cerebral Blood Flow and MetabolismCerebral hyperperfusion after carotid endarterectomy is associated with preoperative hemodynamic impairment and intraoperative cerebral ischemiaJournal of Cerebral Blood Flow and Metabolism
Neurocritical CareMonitoring with the Somanetics INVOS 5100C After Aneurysmal Subarachnoid HemorrhageNeurocritical Care
Changes in cerebral oxygenation in children undergoing surgical repair of ventricular septal defects
Anesthesia and AnalgesiaThe Influence of Inspired Oxygen Fraction and End-Tidal Carbon Dioxide on Post-Cross-Clamp Cerebral Oxygenation During Carotid Endarterectomy Under General AnesthesiaAnesthesia and Analgesia
Effect of intra-operative end-tidal carbon dioxide partial pressure on tissue oxygenation
Indiana Law Journal
A quantitative methodology for determining the need for exposure-prompted medical monitoring
Indiana Law Journal, 79(1):
European Journal of Vascular and Endovascular SurgeryLocal haemodynamic changes during carotid endarterectomy - The influence on cerebral oxygenationEuropean Journal of Vascular and Endovascular Surgery
Invos (R) Cerebral Oximeter compared with the transcranial Doppler for monitoring adequacy of cerebral perfusion in patients undergoing carotid endarterectomy
International Angiology, 25(4):
Journal of Cardiothoracic and Vascular AnesthesiaCerebral oximetry guides treatment during Blalock-Taussig shunt procedureJournal of Cardiothoracic and Vascular Anesthesia
Current Opinion in AnesthesiologyCerebral monitoring during cardiovascular surgeryCurrent Opinion in Anesthesiology
European Journal of Anaesthesiology (EJA)Cerebral oximetry monitoring during unexpected cardiopulmonary arrest and tension pneumothoraxEuropean Journal of Anaesthesiology (EJA)
European Journal of Anaesthesiology (EJA)Determinants of cerebral oximetry in patients undergoing off-pump coronary artery bypass grafting: an observational studyEuropean Journal of Anaesthesiology (EJA)
European Journal of Anaesthesiology (EJA)Regional cerebral saturation versus transcranial Doppler during carotid endarterectomy under regional anaesthesiaEuropean Journal of Anaesthesiology (EJA)
European Journal of Anaesthesiology (EJA)The use of near‐infrared spectroscopy (NIRS) in surgical clipping of giant cerebral aneurysmEuropean Journal of Anaesthesiology (EJA)
Journal of Cardiovascular MedicineCerebral oximetry during carotid clamping: is blood pressure raising necessary?Journal of Cardiovascular Medicine
Journal of Neurosurgical AnesthesiologyNeurological Evaluation by Intraoperative Wake-up During Carotid Endarterectomy Under General AnesthesiaJournal of Neurosurgical Anesthesiology
The NeurologistIntraoperative NeuromonitoringThe Neurologist
Cerebral oximetry; near-infrared spectroscopy.
© 2000 American Society of Anesthesiologists, Inc.
Publication of an advertisement in Anesthesiology Online does not constitute endorsement by the American Society of Anesthesiologists, Inc. or Lippincott Williams & Wilkins, Inc. of the product or service being advertised.
Highlight selected keywords in the article text.