Enter your Email address:
Wolters Kluwer Health may email you for journal alerts and information, but is committed
to maintaining your privacy and will not share your personal information without
You currently have no recent searches
Moran, Robert W.; Stewart, Andrew M.; Spurdle, Anastasia
Unitec, Auckland, New Zealand.
Ultrasound imaging is a non-invasive tool that can be used to generate measurements of muscle size, therefore providing a method of direct assessment of muscle atrophy and hypertrophy in response to various pain states or rehabilitative stimuli. Despite considerable clinical research interest in the anterior abdominal wall, especially transversus abdominis (TA), there has been little published research investigating the reliability of measurement of this muscle.
PURPOSE: To determine reliability of B-mode ultrasound for the measurement of TA muscle thickness in volunteers with a history of low back pain.
METHODS: Eight subjects (four male and four female) ranging in age from 21 to 45 yr (mean ± SD = 29.9 ± 7.6 yr) were recruited. An experienced registered sonographer performed five serial measurements of TA thickness for each subject in the supine position during two sessions at an interval of seven days.
RESULTS: The mean coefficients of variation (%) were 5.7 (90% CI=4.7 to 7.4) and 5.5 (4.2 to 6.6) for session one and two respectively. The mean intraclass correlation coefficients (ICC) across the five trials for all subjects were 0.99 (0.97 to 1.0) and 0.99 (0.97 to 1.0) for session one and session two respectively. The range of ICCs (and confidence intervals) across the five trials for all subjects was 0.94 to 0.98 (CI range 0.75 to 0.99) for session one and 0.82 to 0.98 (CI range 0.4 to 1) for session two. The mean (SD) thickness of TA taken at the first and second measurement sessions was 0.37 cm (0.11) and 0.40 cm (0.10) respectively. The mean typical error between the two measurement sessions was 8.1% (5.7 to 15.1). The mean intraclass correlation coefficient for measurement of TA thickness between sessions was 0.93 (0.74 to 0.99).
CONCLUSION: The use of B-mode ultrasound to determine TA thickness as an outcome measure appears to be a highly reliable measurement technique. Both intra and inter-session reliability is excellent, however, the wider confidence interval associated with inter-session reliability indicates more caution should be applied in interpreting the results of multiple session B-mode ultrasound in intervention studies.
©2007The American College of Sports Medicine
Colleague's E-mail is Invalid
Your Name: (optional)
Separate multiple e-mails with a (;).
Thought you might appreciate this item(s) I saw at Medicine & Science in Sports & Exercise.
Send a copy to your email
Your message has been successfully sent to your colleague.
Some error has occurred while processing your request. Please try after some time.
An Existing Folder
A New Folder
The item(s) has been successfully added to "".
Login with your LWW Journals username and password.
Username or Email:
Enter and submit the email address you registered with. An email with instructions to reset your password will be sent to that address.
Link to reset your password has been sent to specified email address.
What does "Remember me" mean?
By checking this box, you'll stay logged in until you logout. You'll get easier access to your articles, collections,
media, and all your other content, even if you close your browser or shut down your
To protect your most sensitive data and activities (like changing your password),
we'll ask you to re-enter your password when you access these services.
What if I'm on a computer that I share with others?
If you're using a public computer or you share this computer with others, we recommend
that you uncheck the "Remember me" box.
Save my selection
Article Level Metrics